- Research article
- Open Access
Effect and potential risks of using multilevel cement-augmented pedicle screw fixation in osteoporotic spine with lumbar degenerative disease
BMC Musculoskeletal Disorders volume 21, Article number: 274 (2020)
The increase of augmented level and bone cement dose are accompanied by the rising incidence of cement leakage (CL) of cement-augmented pedicle screw instrumentation (CAPSI). But the effect and potential risks of the application of CAPSI to osteoporotic lumbar degenerative disease (LDD) have not been studied in the case of multilevel fixation. This study aimed to investigate the effectiveness and potential complications of using multilevel CAPSI for patients with osteoporotic LDD.
A total of 93 patients with multilevel LDD were divided into the CAPSI group (46 subjects) and the conventional pedicle screw (CPS) group (47 subjects), including 75 cases for three levels and 18 cases for four levels. Relevant data were compared between two groups, including baseline data, clinical results, and complications.
In the CAPSI group, a total of 336 augmented screws was placed bilaterally. The CL was observed in 116 screws (34.52%). Three cemented screws (0.89%) were found loosened during the follow-up and the overall fusion rate was 93.47%. For perioperative complications, two patients (4.35%) experienced pulmonary cement embolism (PCE), one patient augmented vertebral fracture, and three patients (6.52%) wound infection. And in the CPS group, thirty-three screws (8.46%) suffered loosening in cranial and caudal vertebra with a fusion rate of 91.49%. The operation time and hospital stay of CAPSI group were longer than the CPS group, but CAPSI group has a lower screw loosening percentage (P<0. 05). And in terms of blood loss, perioperative complications, fusion rate, and VAS and ODI scores at the follow-up times, there were no significant differences between the two groups.
Patients with osteoporotic LDD underwent multilevel CPS fixation have a higher rate of screw loosening in the cranial and caudal vertebra. The application of cemented pedicle screws for multilevel LDD can achieve better stability and less screw loosening, but it also accompanied by longer operating time, higher incidence of CL, PCE and wound infections. Selective cement augmentation of cranial and caudal pedicle screws may be a worthy strategy to decrease the complications.
Pedicle screw fixation is widely used in degenerative lumbosacral disease, secondary kyphosis or deformity, and thoracolumbar fractures due to a variety of indications, such as bony fusion promotion, deformity correction, and fixation for vertebral fractures [1,2,3]. Although it is the gold standard for the treatment of degenerative and traumatic spinal diseases , the purchase strength of it in low-quality bone still reduced significantly [4, 5]. Instrument failure, such as screw loosening and back-out, associated with nonunion, pseudoarthrosis, and progressive kyphosis, is the most common complication of posterior internal fixation in osteoporosis patients [6, 7].
BMD plays a pivotal role in the stability of the pedicle screw , which can be also affected by other factors include age, smoking, diabetes, different spinal diseases, fusion methods, fixation segments, sacral fixation, repeated screw placement, spinopelvic parameters, and geometry shape of the screw [9,10,11,12]. Studies have shown that the rate of screw loosening is 1–15% in non-osteoporotic patients, but up to 10–60% in osteoporotic patients [9, 10] and even 50–62.8% in multilevel fixation [11,12,13]. Since both osteoporosis and multilevel lumbar degenerative disease (LDD) are common disorders in elderly patients , improving the attachment of internal instruments for patients with multilevel osteoporotic LDD has become an essential mission for spine surgeons.
Both fenestrated and solid screws with cement augmentation have been proven effective in improving the pullout strength of screws and reducing the risk of fixation failure in biomechanical studies and clinical trials [14, 15]. However, CL remains a common complication of CAPSI, and the leakage rate is 11.6–82.4% [16, 17]. CL may cause severe complications such as nerve injury, vascular damage, pulmonary cement embolism (PCE), cardiac embolism, and anaphylactic shock [17, 18]. Multilevel cement-augmented pedicle screws in osteoporotic bone are considered to lead to a higher incidence of CL and cement-related complications. However, no study has selectively analyzed the effect and potential risks of multilevel CAPSI in osteoporotic spine with LDD. Therefore, this study aimed to compare the clinical and radiological results of osteoporotic patients with multilevel LDD (three or four levels) who were treated with CAPSI with those who were treated with CPS.
From February 2010 to February 2017, all patients visiting our hospital for CAPSI or CPS were recruited. The inclusion criteria were as follows: (1) patients diagnosed with LDD by symptoms, signs, and imaging examinations who were not responsive to at least 3 months of conservative treatment, (2) those with posterior interbody fusion using consecutive three-level (L2–L5 or L3–S1) or four-level (L1–L5 or L2–S1) pedicle screw implantation, and (3) those who had undergone lumbar vertebral bone mineral density (BMD) measurement using dual-energy X-ray absorptiometry and with T ≤ − 2.5 SD. The exclusion criteria were as follows: (1) patients with vertebral fracture, tumor, or infection, (2) those with a history of lumbar surgery, (3) those with partial lumbar screws that are augmented in the CAPSI group, or (4) those who had incomplete imaging data at follow-up.
Ninety-three patients, including 75 three level cases (L2–L5: 21 cases, L3–S1: 54 cases) and 18 four level cases (L1–L5 2 cases, L2–S1 16 cases) with an average follow-up of 35.61 ± 19.56 (range: 24–108) months, were divided into the CAPSI group (46 patients) and the CPS group (47 patients). In the CAPSI group, all patients had received cement-augmented screws at the lumbar vertebra, but only 8 patients received augmented screws and 27 received conventional pedicle screws at S1 because of the lack of technique for sacral pedicle screw with cement augmentation, which would be used later. The CAPSI group had 7 men and 39 women with a mean age of 70.65 ± 7.20 (range: 52–86) years, T scores of − 3.18 ± 0.94 (range: − 2.5 to − 4.9) SD, and follow-up of 31.87 ± 15.49 (range: 24–108) months. Twenty-five patients had lumbar spinal stenosis with spinal instability, 9 had lumbar spondylolisthesis, and 12 had lumbar spinal stenosis with degenerative scoliosis. Solid screws were used in 16 patients and fenestrated screws were used in 30 patients. The CAP group had 6 males and 41 females with a mean age of 67.91 ± 7.62 (range: 55–86) years, T scores of − 2.89 ± 0.49 (range: − 2.5 to − 5.0) SD, and follow-up of 35.53 ± 21.54(range: 24–96) months. Thirty patients had lumbar spinal stenosis with spinal instability, 8 had lumbar spondylolisthesis, and 9 had lumbar spinal stenosis with degenerative scoliosis. A flowchart of this study is shown in Fig. 1.
All patients underwent standard posterior lumbar interbody fusion (PLIF) and/or transforaminal lumbar interbody fusion (TLIF) under general anesthesia. The decision for cement augmentation was made by an experienced surgeon after evaluating the patients’ BMD and the biomechanical stability of the implanted pedicle screw. The operation for installation of a traditional pedicle screw was performed using a standard technique according to the literature [4, 10]. The procedure for placement of fenestrated pedicle screw with cement and solid pedicle screw with pre-augmented trajectory has been described in our previous study . In general, pedicle screws with a diameter and length of 6.5 mm and 45 mm, respectively, were used in the lumbar spine, and those with a diameter and length of 6.5 mm and 40 mm, respectively, were used in the sacrum. For every lumbar pedicle screw and sacral pedicle screw, approximately 1.5–4 mL and 1.5–2.5 mL of polymethylmethacrylate (PMMA) was injected, respectively.
At the early stage after surgery, the patients were required to perform lower limb exercises on the bed. The drainage tube was removed when the amount of drainage was < 50 mL/d. All patients began to walk with protection on the waist 3 or 4 days after the operation. Patients were required to wear a waist protector for the first month after surgery. Routine postoperative radiography of the thorax was performed in augmented patients, and patients complaining of discomfort in the heart or lung received additional thoracic CT. All patients received anti-osteoporosis treatment—calcium carbonate, vitamin D3, and bisphosphonate—throughout the treatment period.
The visual analog scale (VAS) and the Oswestry Disability Index (ODI) at different follow-up periods were recorded to assess clinical outcomes. Procedure duration, intraoperative blood loss, and duration of hospitalization of the two groups were abstracted from medical records. Major postoperative complications were also compared between the two groups, including nerve injury, dural tearing, wound infection, and revision surgery.
Radiologic evaluation included spinopelvic parameters, screw loosening rate, and lumbar fusion status. Two experienced orthopedists were assigned to evaluate the radiologic parameters independently. All data were measured twice by two researchers, and the mean values were used for analysis. The degree of lumbar lordosis (LL), pelvic tilt (PT), and pelvic incidence (PI) were assessed as previously described . Screw loosening was defined as a halo sign showing a radiolucent line of ≥1 mm around the screw in X-ray or CT images on one or both sides after the surgery [11, 21]. Solid fusion was determined according to the criterion described by Dakhil et al.  as a bridging interbody bone occurring in the operated segment and an angle of < 5° on dynamic X-ray images at the fixed level.
The data were analyzed using the SPSS 19.0 software (IBM, Inc., Armonk, NY, USA). The difference of the continuous variables between the two groups was compared by using the independent sample t-test (age, BMD, BMI, VAS and ODI scores, follow-up, operation time, blood loss, and hospital stay and spinopelvic parameters). The chi-square test was used to compare the classification variables (gender, diabetes, hypertension, smoking, diagnosis, fusion methods, bone graft, fusion level, lumbosacral fixation, different patterns of S1 pedicle screw placement, lumbosacral fixation, complications, screw loosening, nonunion patients). A P < 0.05 was considered statistically significant.
There were no statistical differences between these two groups in terms of sex, age, BMD, BMI, diabetes, hypertension, smoking, diagnosis, fusion methods, bone graft, fusion level, lumbosacral fixation, different patterns of S1 pedicle screw placement, lumbosacral fixation, preoperative VAS score, preoperative ODI score, follow-up, preoperative LL, preoperative PT, and preoperative PI (Table 1).
Clinical effects and complications
There were 336 cement-augmented pedicle trajectories used, with a mean of 7.30 ± 1.4 (range: 6–10) for the instrumented screw and 1.85 ± 0.65 mL (range: 1–4) for PMMA per pedicle screw. CL was observed in 44 patients (95.65%) and 116 screws (34.52%), and CL most often occurred in the perivertebral venous system; leakage via segmental veins and basivertebral veins was seen in 109 screws (93.97%). Perioperative complications included the following: PCE (2 patients, of whom 1 was symptomatic and the other was asymptomatic; Fig. 2), hypotension (2 patients), superficial skin infection that required surgical debridement (2 patients), deep infection but the instrumentation was not removed (1 patient), dural tear and discharged after surgical suture repair (1 patient), augmented vertebral fracture (1 patient; Fig. 3), radiating leg pain (6 patients), and discomfort at the last follow-up (2 patients). There were no serious complications caused by the leakage of bone cement, such as nerve damage, cardiac embolism, or death. Three cemented screws (0.89%) (Fig. 4) and 24 S1 conventional screws (44.44%) were found to be loose at the follow-up. Of the 3 patients who had nonunion, 2 required revision surgery with a fusion rate of 93.47%. During follow-up, adjacent fractures were observed in 6 patients and vertebral fractures were observed in 8 patients. Symptomatic adjacent disc degeneration was observed in 3 patients, of which 2 required revision surgery. A typical case of the CAPSI group is shown in Fig. 5.
In the CPS group, perioperative complications included the following: hypotension (1 patient), dural tear (1 patient), and severe radiating leg pain due to pedicle screw misplacement (8 patients, of which 3 still complained about discomfort at the last follow-up). Seventeen patients (36.17%) and 33 screws (8.46%) showed loosening in the cranial or caudal vertebra (Fig. 6). Of the 4 patients who experienced nonunion, 1 required revision surgery with a fusion rate of 91.49%. Adjacent fractures were observed in 3 patients and 3 vertebrae during follow-up, and symptomatic adjacent disc degeneration was observed in 3 patients—all required revision surgery. A typical case of the CPS group is shown in Fig. 7.
Compared with the date of preoperation, a larger postoperation LL was found in both the CAPSI and CPS groups. However, there were no statistical differences in preoperative and postoperative PT and PI both groups. Longer operation time, longer hospital stay, and less screw loosening were found in the CAPSI group than in the CPS group (P < 0.05). In terms of blood loss, perioperative complications, fusion rate, VAS and ODI scores at follow-up, postoperative LL, postoperative PT, and postoperative PI, no significant difference was detected between the two groups (Table 2 and Fig. 8).
Pedicle screw fixation in osteoporotic spine is gradually becoming common because of the aging population. Therefore, we will have to face the increasing incidences of pedicle screw loosening and instrumentation-associated complications . According to the literature, 17% of revision surgeries are associated with pedicle screw failure . Thus, CAPSI has been widely used for patients with poor bone quality due to its reliable biomechanical stability [15, 24,25,26]. But the treatment of multilevel LDD with cement-augmented pedicle screws is more complicated and challenging compared with 1 or 2-level LDD because of the increase of CL and mechanical overload . Many studies either assessed only the results of short segment augmentation or assessed the results of long and short segment augmentation together, and most data are not related to osteoporotic LDD. So far there is currently little data about the effectiveness and potential risks of CAPSI treating the multilevel LDD.
Although the application of augmented pedicle screws for multilevel LDD can achieve better stability and less screw loosening, it also carries a high risk of CL, PCE, and wound infections. The CL rate of the present study was 34.52%. Several studies indicated that CAPSI has a high risk of asymptomatic cement leakage and serious complications were also reported in the literature. Methods to reduce CL in LDD include: (1) Using high viscosity cement or toothpaste viscosity cement . (2) Reducing the volume of cement and the number of augmented screws [29, 30]. Experimental studies have indicated that screw stability does not significantly improve when the volume of PMMA exceeded 2.8 ml each screw . Therefore, the volumes of bone cement should be optimized according to the severity of osteoporosis. (3) Planning the trajectory and the size of pedicle screws by preoperative CT to achieve the accurate placement of pedicle screws. Our previous study has found that a smaller distance between the tip of screw and the midline of vertebra is closely related to the epidural CL . (4) We suggest injecting cement with small doses and slow speed (the cement was injected by every 0.1 ml increment in this study) or creating a small cavity in the vertebral body prior to the cement injection . (5) The injection of cement should be confirmed consecutively by fluoroscopy.
In the present study, PCE was detected in two cases (4.35%) that one patient was symptomatic. Clinical research has shown the incidence rate of pulmonary embolism ranged from 3.5 to 23% after percutaneous vertebroplasty and kyphoplasty [17, 31]. Most of the pulmonary embolism cases are asymptomatic while the incidence of symptomatic is about 1.4% ~ 4.1% . Severe PCE is rare but fatal [17, 32]. The optimal treatment strategy for PCE remains controversial. Several researches have suggested that for asymptomatic PCE, it is feasible to adopt clinical observation and regular follow-up without anticoagulants. And for symptomatic PCE, long-term anticoagulation therapies, including initial heparin during hospitalization and oral coumarin therapy, can prevent cardiovascular complications. Surgery for removing cement embolus is also an alternative strategy for patients with severe embolism .
Wound infection may lead to nonunion and instrumentation removal. The infection rate of CAPSI group was very high (6.52%, 3/46). We believe it was related to advanced age, long operation time, large intraoperative bleeding, and other comorbidities, such as diabetes. The average age of these patients was 78 years old, average operation time 326.67 min, average blood loss 1433.33 ml, and two of them have diabetes. Few clinical studies reported augmented screw loosening, but it can’t mask the fact that it is one of the non-ignorable complications in CAPSI. Two patients (three augmented screws) were found loosening in our study, one of the screws is caused by tumble and the others poor distribution of bone cement around them. Due to the worse balance control, lower extremity pain, and musculoskeletal degeneration, LDD patients become more prone to fall, even after spinal surgery . When bone cement was poorly-dispersed around the pedicle screw, the biomechanical stability were dramatically lower than those well distributed one [14, 23]. In addition, according to our experience, when a solid screw with PMMA augmentation is implanted late and cement became solid, the sclerotic wall of the cement does not allow this screw to interdigitate into the PMMA, and thus the screw may deviate from augmented trajectory.
We attempted to inject PMMA and then placed screws immediately for multilevel augmentation from the beginning, but it brought a risk that screws cannot interdigitate into sclerotic cement (Fig.3). Later we found using fenestrated screws leads to better control of PMMA injection (especially when more than two vertebra need to be augmented). Thus, we propose using fenestrated screws for multilevel augmentation, especially for inexperienced surgeons. But we tend to insert solid screws in S1 to reduce the leakage caused by fenestrated screws penetrating the presacral bone cortex . Otherwise, we usually use augmented screws on two vertebral bodies at a time rather than three bodies, which minimizes the difficulty of operation and facilitates C-arm fluoroscopy.
Screw loosening of multilevel fixation often occurs in the cranial and caudal segment and far more frequently in the caudal (S1 in particular). Consistent with previous studies [9, 27], twenty-four S1 conventional screws (44.44%) were found loosening in the CAPSI group, and thirty-three screws (8.46%) in CPS group, including seven screws at cranial level and twenty-six at caudal level (eight in L5, eighteen in S1). Wu et al.  analyzed 658 screws of 126 patients and found that 25 patients experienced screw loosening, 18 patients (72%) had loosened screws in caudal vertebra while 7 (28%) in the cranial vertebra. Li et al.  also found that the failure of internal fixation usually develops at the caudal level in cases of multilevel pedicle screw fixation. Therefore, we recommend increasing properly the bone cement dose in cranial and caudal screw to reduce the need for sacroiliac screw fixation and other pelvic fixation .s.
Since the application of pedicle screws with cement on all segments will increase the operating time, CL, PCE, and incision infection, only cementing the selected strategic vertebrae, such as cranial or caudal pedicle screw alone, could be a worthy strategy. Based on the clinical and radiological data of 31 patients treated with cement injections only for the selected strategic vertebrae and augmentation for all segments, Erdem  found the rate of PCE and operative time were markedly higher in all segments cement injections group, but there is no significant difference in screw loosening and clinical efficacy between two groups. Therefore, selective cement augmentation for both cranial and caudal pedicle screws could be a potential optimal procedure to decrease the side complications of CAPSI.
There are several limitations of our study. First, since we have not analyzed other risk factors of the screw loosening and fusion, such as the strength of paravertebral muscles, single or double cage, the length/diameter of screw and the repeated placement of screws during operation, as well as the specific time of screw loosening and asymptomatic adjacent segment degeneration. Second, as a retrospective study, it is difficult for us to control the uniformity of the patients and there is a degree of selective bias in this study, such as the inclusion of cases, the grouping of patients, and the selection of surgical methods. In addition, the actual rate of loosening may be lower than those reported in this paper, because the patients with screw loosening are more likely to receive follow-up, while patients with good curative effect may be reluctant to return to the hospital.
Patients with osteoporotic LDD underwent multilevel CPS fixation have a high rate of screw loosening in the cranial and caudal level. The application of augmented pedicle screws on multilevel LDD can achieve better stability and less screw loosening, but it also accompanied by longer operating time, higher incidence of CL, PCE and wound infections. Selective cement-augmenting cranial and caudal pedicle screws may be a worthy strategy to decrease the complications.
Availability of data and materials
The datasets used in the current study are not publicly available due to privacy protection but are available from the corresponding author on reasonable request.
Cement-augmented pedicle screw instrumentation
Lumbar degenerative disease
Conventional pedicle screw
Pulmonary cement embolism
bone mineral density
Body mass index
Visual analogue scale
Oswestry disability index
Transforaminal lumbar interbody fusion
Polyether ether ketone
Moussazadeh N, Rubin DG, McLaughlin L, Lis E, Bilsky MH, Laufer I. Short-segment percutaneous pedicle screw fixation with cement augmentation for tumor-induced spinal instability. Spine J 2015;15:1609–1617. https://doi.org/https://doi.org/10.1016/j.spinee.2015.03.037.
Rihn JA, Hilibrand AS, Zhao W, Lurie JD, Vaccaro AR, Albert TJ, et al. Effectiveness of surgery for lumbar stenosis and degenerative spondylolisthesis in the octogenarian population: analysis of the spine patient outcomes research trial (SPORT) data. J Bone Joint Surg Am 2015;97:177–185. https://doi.org/https://doi.org/10.2106/JBJS.N.00313.
Rong Z, Zhang F, Xiao J, Wang Z, Luo F, Zhang Z, et al. Application of cement-injectable Cannulated pedicle screw in treatment of osteoporotic thoracolumbar vertebral compression fracture (AO type a): a retrospective study of 28 cases. World Neurosurg. 2018;120:e247-247e258. https://doi.org/https://doi.org/10.1016/j.wneu.2018.08.045.
Weiser L, Huber G, Sellenschloh K, Viezens L, Püschel K, Morlock MM, et al. Insufficient stability of pedicle screws in osteoporotic vertebrae: biomechanical correlation of bone mineral density and pedicle screw fixation strength. Eur Spine J 2017;26:2891–2897. https://doi.org/https://doi.org/10.1007/s00586-017-5091-x.
Bostelmann R, Keiler A, Steiger HJ, Scholz A, Cornelius JF, Schmoelz W. Erratum to: effect of augmentation techniques on the failure of pedicle screws under cranio-caudal cyclic loading. Eur Spine J 2015;24:2098. https://doi.org/https://doi.org/10.1007/s00586-015-4001-3.
Weiser L, Dreimann M, Huber G, Sellenschloh K, Püschel K, Morlock MM, et al. Cement augmentation versus extended dorsal instrumentation in the treatment of osteoporotic vertebral fractures: a biomechanical comparison. Bone Joint J 2016;98-B:1099–1105. https://doi.org/https://doi.org/10.1302/0301-620X.98B8.37413.
Park CK, Park CK, Lee DC, Lee DG. A new technique of bone cement augmentation via the disc space for percutaneous pedicle screw fixation. J Neurosurg Spine. 2016;24:16–19. https://doi.org/https://doi.org/10.3171/2015.4.SPINE141115.
Oberkircher L, Krüger A, Hörth D, Hack J, Ruchholtz S, Fleege C, et al. Anterior cement augmentation of adjacent levels after vertebral body replacement leads to superior stability of the corpectomy cage under cyclic loading-a biomechanical investigation. Spine J 2018;18:525–531. https://doi.org/https://doi.org/10.1016/j.spinee.2017.10.068.
Wu JC, Huang WC, Tsai HW, Ko CC, Wu CL, Tu TH, et al. Pedicle screw loosening in dynamic stabilization: incidence, risk, and outcome in 126 patients. Neurosurg Focus 2011;31:E9. https://doi.org/https://doi.org/10.3171/2011.7.FOCUS11125.
Galbusera F, Volkheimer D, Reitmaier S, Berger-Roscher N, Kienle A, Wilke HJ. Pedicle screw loosening: a clinically relevant complication. Eur Spine J 2015;24:1005–1016. https://doi.org/https://doi.org/10.1007/s00586-015-3768-6.
Kim JB, Park SW, Lee YS, Nam TK, Park YS, Kim YB. The effects of Spinopelvic parameters and Paraspinal muscle degeneration on S1 screw loosening. J Korean Neurosurg Soc 2015;58:357–362. https://doi.org/https://doi.org/10.3340/jkns.2015.58.4.357.
Röllinghoff M, Schlüter-Brust K, Groos D, Sobottke R, Michael JW, Eysel P, et al. Mid-range outcomes in 64 consecutive cases of multilevel fusion for degenerative diseases of the lumbar spine. Orthop Rev (Pavia). 2010;2:e3 https://doi.org/10.4081/or.2010.e3.
El Saman A, Meier S, Sander A, Kelm A, Marzi I, Laurer H. Reduced loosening rate and loss of correction following posterior stabilization with or without PMMA augmentation of pedicle screws in vertebral fractures in the elderly. Eur J Trauma Emerg Surg 2013;39:455–460. https://doi.org/https://doi.org/10.1007/s00068-013-0310-6.
Choma TJ, Frevert WF, Carson WL, Waters NP, Pfeiffer FM. Biomechanical analysis of pedicle screws in osteoporotic bone with bioactive cement augmentation using simulated in vivo multicomponent loading. Spine (Phila Pa 1976). 2011;36:454–462. https://doi.org/https://doi.org/10.1097/BRS.0b013e3181d449ec.
Tan JS, Bailey CS, Dvorak MF, Fisher CG, Cripton PA, Oxland TR. Cement augmentation of vertebral screws enhances the interface strength between interbody device and vertebral body. Spine (Phila Pa 1976) 2007;32:334–341. https://doi.org/https://doi.org/10.1097/01.brs.0000253645.24141.21.
Martín-Fernández M, López-Herradón A, Piñera AR, Tomé-Bermejo F, Duart JM, Vlad MD, et al. Potential risks of using cement-augmented screws for spinal fusion in patients with low bone quality. Spine J 2017;17:1192–1199. https://doi.org/https://doi.org/10.1016/j.spinee.2017.04.029.
Janssen I, Ryang YM, Gempt J, Bette S, Gerhardt J, Kirschke JS, et al. Risk of cement leakage and pulmonary embolism by bone cement-augmented pedicle screw fixation of the thoracolumbar spine. Spine J 2017;17:837–844. https://doi.org/https://doi.org/10.1016/j.spinee.2017.01.009.
Hatzantonis C, Czyz M, Pyzik R, Boszczyk BM. Intracardiac bone cement embolism as a complication of vertebroplasty: management strategy. Eur Spine J 2017;26:3199–3205. https://doi.org/https://doi.org/10.1007/s00586-016-4695-x.
Guo HZ, Tang YC, Guo DQ, Zhang SC, Li YX, Mo GY, et al. The cement leakage in cement-augmented pedicle screw instrumentation in degenerative lumbosacral diseases: a retrospective analysis of 202 cases and 950 augmented pedicle screws. Eur Spine J. 2019;https://doi.org/https://doi.org/10.1007/s00586-019-05985-4.
Guo HZ, Tang YC, Li YX, Yuan K, Guo DQ, Mo GY, et al. The effect and safety of Polymethylmethacrylate-augmented sacral pedicle screws applied in osteoporotic spine with lumbosacral degenerative disease: a 2-year follow-up of 25 patients. World Neurosurg. 2019;121:e404-404e410. https://doi.org/https://doi.org/10.1016/j.wneu.2018.09.121.
Moon BJ, Cho BY, Choi EY, Zhang HY. Polymethylmethacrylate-augmented screw fixation for stabilization of the osteoporotic spine : a three-year follow-up of 37 patients. J Korean Neurosurg Soc. 2009;46:305–311. https://doi.org/https://doi.org/10.3340/jkns.2009.46.4.305.
Dakhil-Jerew F, Jadeja H, Cohen A, Shepperd JA. Inter-observer reliability of detecting Dynesys pedicle screw using plain X-rays: a study on 50 post-operative patients. Eur Spine J 2009;18:1486–1493. https://doi.org/https://doi.org/10.1007/s00586-009-1071-0.
Fan HT, Zhang RJ, Shen CL, Dong FL, Li Y, Song PW, et al. The biomechanical properties of pedicle screw fixation combined with trajectory bone cement augmentation in osteoporotic vertebrae. Clin Spine Surg 2016;29:78–85. https://doi.org/https://doi.org/10.1097/BSD.0b013e3182a14870.
Bullmann V, Liljenqvist UR, Rödl R, Schulte TL. [Pedicle screw augmentation from a biomechanical perspective]. Orthopade. 2010;39:673–678. https://doi.org/https://doi.org/10.1007/s00132-010-1602-8.
Fölsch C, Goost H, Figiel J, Paletta JR, Schultz W, Lakemeier S. Correlation of pull-out strength of cement-augmented pedicle screws with CT-volumetric measurement of cement. Biomed Tech (Berl). 2012;57:473–480. https://doi.org/https://doi.org/10.1515/bmt-2012-0012.
Elder BD, Lo SF, Holmes C, Goodwin CR, Kosztowski TA, Lina IA, et al. The biomechanics of pedicle screw augmentation with cement. Spine J 2015;15:1432–1445. https://doi.org/https://doi.org/10.1016/j.spinee.2015.03.016.
Li T, Shi L, Luo Y, Chen D, Chen Y. One-level or multilevel Interbody fusion for multilevel lumbar degenerative diseases: a prospective randomized control study with a 4-year follow-up. World Neurosurg 2018;110:e815-815e822. https://doi.org/https://doi.org/10.1016/j.wneu.2017.11.109.
Schmoelz W, Heinrichs CH, Schmidt S, Piñera AR, Tome-Bermejo F, Duart JM, et al. Timing of PMMA cement application for pedicle screw augmentation affects screw anchorage. Eur Spine J 2017;26:2883–2890. https://doi.org/https://doi.org/10.1007/s00586-017-5053-3.
Arabmotlagh M, Rickert M, Lukas A, Rauschmann M, Fleege C. Small cavity creation in the vertebral body reduces the rate of cement leakage during vertebroplasty. J Orthop Res 2017;35:154–159. https://doi.org/https://doi.org/10.1002/jor.23215.
Ulusoy OL, Kahraman S, Karalok I, Kaya E, Enercan M, Sever C, et al. Pulmonary cement embolism following cement-augmented fenestrated pedicle screw fixation in adult spinal deformity patients with severe osteoporosis (analysis of 2978 fenestrated screws). Eur Spine J 2018;27:2348–2356. https://doi.org/https://doi.org/10.1007/s00586-018-5593-1.
Krueger A, Bliemel C, Zettl R, Ruchholtz S. Management of pulmonary cement embolism after percutaneous vertebroplasty and kyphoplasty: a systematic review of the literature. Eur Spine J 2009;18:1257–1265. https://doi.org/https://doi.org/10.1007/s00586-009-1073-y.
Mueller JU, Baldauf J, Marx S, Kirsch M, Schroeder HW, Pillich DT. Cement leakage in pedicle screw augmentation: a prospective analysis of 98 patients and 474 augmented pedicle screws. J Neurosurg Spine 2016;25:103–109. https://doi.org/https://doi.org/10.3171/2015.10.SPINE15511.
Wong WJ, Lai DM, Wang SF, Wang JL, Hsu WL. Changes of balance control in individuals with lumbar degenerative spine disease after lumbar surgery: a longitudinal study. Spine J. 2019;https://doi.org/https://doi.org/10.1016/j.spinee.2019.02.015.
Erdem MN, Karaca S, Sarı S, Yumrukcal F, Tanli R, Aydogan M. Application of cement on strategic vertebrae in the treatment of the osteoporotic spine. Spine J 2017;17:328–337. https://doi.org/https://doi.org/10.1016/j.spinee.2016.10.001.
This study supported by the innovation and strength project of The First Affiliated Hospital of Guangzhou University of Chinese Medicine (2019IIT32). The funders had no role in study design, data collection or analysis, decision to publish, or preparation of the manuscript.
Ethics approval and consent to participate
This study was approved by the Ethics Committee of The First Affiliated Hospital of Guangzhou University of Chinese Medicine. All participation in the study was voluntary and written informed consent was obtained from the participants.
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Tang, Yc., Guo, Hz., Guo, Dq. et al. Effect and potential risks of using multilevel cement-augmented pedicle screw fixation in osteoporotic spine with lumbar degenerative disease. BMC Musculoskelet Disord 21, 274 (2020). https://doi.org/10.1186/s12891-020-03309-y
- Multilevel fixation
- Cement-augmented pedicle screw instrumentation
- Lumbar degenerative disease