- Research article
- Open Access
- Open Peer Review
Biomechanical comparison of menisci from different species and artificial constructs
- Gunther H Sandmann2,
- Christopher Adamczyk3,
- Eduardo Grande Garcia4,
- Stefan Doebele2,
- Andreas Buettner5,
- Stefan Milz6,
- Andreas B Imhoff1,
- Stefan Vogt1,
- Rainer Burgkart†4 and
- Thomas Tischer†1, 7Email author
© Sandmann et al.; licensee BioMed Central Ltd. 2013
- Received: 16 May 2013
- Accepted: 12 November 2013
- Published: 17 November 2013
Loss of meniscal tissue is correlated with early osteoarthritis but few data exist regarding detailed biomechanical properties (e.g. viscoelastic behavior) of menisci in different species commonly used as animal models. The purpose of the current study was to biomechanically characterize bovine, ovine, and porcine menisci (each n = 6, midpart of the medial meniscus) and compare their properties to that of normal and degenerated human menisci (n = 6) and two commercially available artificial scaffolds (each n = 3).
Samples were tested in a cyclic, minimally constraint compression–relaxation test with a universal testing machine allowing the characterization of the viscoelastic properties including stiffness, residual force and relative sample compression. T-tests were used to compare the biomechanical parameters of all samples. Significance level was set at p < 0.05.
Throughout cyclic testing stiffness, residual force and relative sample compression increased significantly (p < 0.05) in all tested meniscus samples. From the tested animal meniscus samples the ovine menisci showed the highest biomechanical similarity to human menisci in terms of stiffness (human: 8.54 N/mm ± 1.87, cycle 1; ovine: 11.24 N/mm ± 2.36, cycle 1, p = 0.0528), residual force (human: 2.99 N ± 0.63, cycle 1 vs. ovine 3.24 N ± 0.13, cycle 1, p = 0.364) and relative sample compression (human 19.92% ± 0.63, cycle 1 vs. 18.72% ± 1.84 in ovine samples at cycle 1, p = 0.162). The artificial constructs -as hypothesized- revealed statistically significant inferior biomechanical properties.
For future research the use of ovine meniscus would be desirable showing the highest biomechanical similarities to human meniscus tissue. The significantly different biomechanical properties of the artificial scaffolds highlight the necessity of cellular ingrowth and formation of extracellular matrix to gain viscoelastic properties. As a consequence, a period of unloading (at least partial weight bearing) is necessary, until the remodeling process in the scaffold is sufficient to withstand forces during weight bearing.
- Animal model
- Meniscus scaffolds
The important role of the meniscus in load transmission and shock absorption is well known [1, 2]. It is obvious that loss of meniscus tissue leads to focal overload of articular cartilage and finally to a higher incidence of osteoarthritis [1, 3]. Unfortunately, meniscus tears are very common and (partial) resection of the damaged tissue is necessary in many cases [3, 4]. To avoid the negative side effects of (total or partial) meniscectomy [5, 6], alternative treatment options, especially for young individuals, are required. Meniscus allograft transplantation has been reported to reduce pain and improve knee joint function . However, there are still problems like the sizing of implants, risk of disease transmission, and limited availability [8, 9]. In addition, degeneration, tears and structural alterations were noticed throughout the remodeling process in several studies dealing with early meniscus allograft transplantation [10, 11].
Alternatively, artificial scaffolds such as Menaflex (formerly CMI) or Actifit can be implanted. Menaflex is derived from devitalized bovine collagen (Achilles tendon) and this scaffold allows cellular ingrowth and improves tissue regeneration [10, 12]. The use of Bone marrow derived stem cells under perfusion and cyclic compression has been shown to have a positive effect on the cellular ingrowth in vitro . So far, several studies have proven the positive effect of Menaflex implantation on the clinical outcome measured by IKDC, Tegner Score and VAS [14–16]. Actifit is a polyurethane polymer scaffold and good biocompatibility and cellular ingrowth have been reported leading to an improvement of the frictional properties as reported by Galley et al. . Nevertheless, the long-term benefits of these scaffolds in terms of chondroprotection are still a matter of debate , although mid-term results are promising [19, 20]. Especially, studies about the biomechanical properties of theses polyurethane scaffolds at time of implantation are missing.
Whereas human meniscus has been characterized relatively well in several studies [21–23], biomechanical data of meniscus tissue of suitable animal models serving as a base for comparative studies are therefore of particular interest [24–26]. Sweigart et al. examined the biomechanical properties of baboon, bovine, canine, lapine, porcine and human meniscus samples . They found significant variations in the material properties, which underline the problems in finding the perfect animal model and transferring the results to the human knee joint. As Chevrier et al. found the highest structural similarities in terms of vascularity and collagen structure between human and ovine menisci, our study aimed to close this gap .
Therefore, we characterized the biomechanical properties of different human (normal and degenerative) and animal (bovine, ovine, porcine) menisci and compared the results to two commercially available artificial scaffolds (Menaflex, Actifit). The midpart of the meniscus was chosen as it represents a well-defined region, which always can be identified and examined in all species irrespective of size differences.
Animal meniscus samples
Meniscus samples were harvested from fresh bovine (n = 6), ovine (n = 6) and porcine (n = 6) knee joints obtained from a local slaughterhouse. All animals were young at slaughter but skeletally mature with closed epiphyseal plates (bovine: average weight 135 kg, average age 7.3 months, ovine: average weight 85.2 kg, average age 12.1 months, porcine: average weight 92.5 kg, average age 11.3 months). Knees with macroscopic signs of degenerative or traumatic changes to the cartilage or obvious meniscal tears were excluded. Using a cylindrical device for “osteochondral autologous transplantation” (OATS donor device, Arthrex, Naples, Fl) three samples from the midpart (pars intermedia) of each medial meniscus, 8 mm in diameter and 4 mm in height were harvested. The meniscus specimens were stored in physiologic saline solution at 8°C until biomechanical testing (performed at room temperature). Testing was performed within 6 hours to avoid alterations due to prolonged storage.
Human meniscus samples
Normal human meniscus samples were obtained from the Department of Forensic medicine and excluded if there were any signs of macroscopic degenerative knee disease or traumatic knee injury. Medial meniscus samples of 6 Caucasian male individuals with a mean age of 30.7 years (range 25-34 years, SD 3.27) were used. All had died due to diseases or accidents not affecting the knee. In contrast, the degenerative human meniscus samples were collected during operations with partial (n = 4) or total knee replacements (n = 2) with a mean patient age of 55.3 years (range 50-60 years, SD 3.67). In 2 patients the unicondylar prostheses became necessary due to partial meniscectomies in their twenties. The remaining four patients had posttraumatic arthritis after tibial head fractures explaining the relatively young age of our patients at time of prosthesis implantation. All human samples were treated as described above and three cylinders were harvested from each meniscus.
The study was approved by the Institutional Ethical Review Board of the Institute of Forensic Medicine. All patients approved the use of their tissues for this research project.
The artificial scaffolds Menaflex (Regen Biologics, Hackensack, NJ, USA), formerly known as collagen meniscus implant (CMI), and Actifit (Orteq, London, UK) were used. From both scaffolds three samples (n = 3), were used to test their biomechanical properties. The samples were harvested with the OATS donor device (Arthrex, Naples, Fl), stored in physiologic saline solution for 10 minutes and kept moist throughout testing.
For an overview of the principal tissue or scaffold organization, histologic sections were obtained from selected remaining tissue samples or scaffold material. After fixation of the human and animal meniscus samples in 90% methanol in 4°C for 48 h, specimens were infiltrated overnight in PBS with 5% sucrose at pH 7.4 and afterwards mounted on chucks in Jung tissue embedding medium (Leica, Germany), frozen in a HM 500 OMV cryostat (Mikrom, Germany) and cryosectioned at 12 μm. Sections were stained with hematoxylin and eosin and Safranin O staining. Artificial specimens were embedded in paraffin, sectioned and stained with Safranin O and light green. Safranin O staining highlights the accumulation of acidophil molecular complexes (proteoglycans and glycosaminoglycans) in the tissue.
Statistical analysis of the biomechanical data was performed using the software package SPSS (Version 11.0, SPSS Inc., Chicago, Illinois). After testing for normal distribution (Kolmogorov-Smirnov test) we used t-tests to compare the biomechanical parameters of animal, human and artificial meniscus samples. Significance level was defined at p < 0.05.
The mean sample height was 3.9 mm (±0.2 mm) and all samples could be loaded up to 7 N without signs of plastic deformity. Slopes of the load curves were linear between 2 N and 5 N in all tests.
Summary of the biomechanical results showing the viscoelastic properties including stiffness, residual force and compression
Human (n = 6)
Human deg. (n = 6)
Bovine (n = 6)
Ovine (n = 6)
Porcine (n = 6)
Actifit (n = 3)
Menaflex (n = 3)
Stiffness [N/mm], cycle1
8.54 ± 1.87
7.94 ± 2.25
14.72 ± 2.07
11.24 ± 2.36
15.24 ± 0.95
2.83 ± 0.13
4.66 ± 0.35
Stiffness [N/mm], cycle 5
18.29 ± 2.88
16.59 ± 2.66
22.89 ± 2.01
19.84 ± 3.2
24.63 ± 1.28
3.88 ± 0.17
5,50 ± 0.33
Resiudal Force [N], cycle 1
2.99 ± 0.63
2.31 ± 0.69
3.13 ± 0.2
3.24 ± 0.13
2.05 ± 0.21
5.93 ± 0.017
5.15 ± 0.31
Residual Force [N], cycle 5
4.26 ± 0.54
3.74 ± 0.35
4.45 ± 0.12
4.49 ± 0.19
3.73 ± 0.14
8.18 ± 0.032
5.58 ± 0.32
Compression [%], cycle 1
19.92 ± 1.36
18.43 ± 055
14.05 ± 0.92
18.72 ± 1.84
15.21 ± 1.25
74.7 ± 4.76
72.8 ± 3.86
Compresson [%], cycle 5
13.58 ± 1.33
14.38 ± 1.14
11.10 ± 0.95
13.43 ± 1.61
10.31 ± 0.53
65.1 ± 3.38
71.89 ± 7.23
The objective of our study was to assess the biomechanical behavior of meniscal tissue from different large animals in comparison to normal and degenerated human meniscus tissue as well as two different artificial scaffolds. We found that the viscoelatisc properties of ovine meniscus samples show the highest congruence to human meniscus samples in our biomechanical setup and that the tested artificial scaffolds lack the typical meniscal properties at time zero.
Several different approaches have been described to evaluate the biomechanical properties of meniscus tissue: The creep indentation test characterizes the material properties with the aggregate modulus, Poisson’s ratio, permeability, and shear modulus . Uniaxial tension tests are used for evaluation of meniscal attachment function after transplantation , tensile forces are used to evaluate suture function in meniscus repair techniques [28, 29]. The ball indention test characterizes the viscoelastic properties of meniscal tissue including stiffness, residual force and compression behavior [9, 30] and we chose this test also in our biomechanical setup as the cyclic testing mimics physiological loading conditions.
Joshi et al.  found that among monkey, canine, bovine and porcine samples the ovine menisci were most similar to human menisci. These data can only partially be compared to our work because a different biomechanical setup with uniaxial confined compression testing was used. Nevertheless, we could also confirm the biomechanical similarities between human and ovine samples in our own setup. Based on the work of Sweigart et al. who report biomechanical differences in the different regions of the porcine medial meniscus, we assured to harvest the meniscus samples from the same topographic region to avoid any bias .
Normal human and degenerative meniscus samples from the same region revealed significantly different stiffness values with lower values for the degenerative samples. In addition, we could show that human meniscus samples exhibited statistically significant differences in their viscoelastic behavior compared to the large animal menisci tested, except the ovine samples. As a consequence meniscus research focusing on the biomechanical properties should be performed using an ovine animal model.
Data about the biomechanical properties of Menaflex and Actifit before cellular ingrowth are sparse. The Actifit polyurethane scaffold shows a classic stress-strain response  and has slightly more viscoelastic behavior (higher compression and residual force), whereas its stiffness is less (not significant) than in the Menaflex scaffold. Both artificial constructs had significantly different viscoelastic properties in comparison to all tested human and animal samples.
Since the Menaflex samples are made from bovine collagen (mostly type I), its structure is well adapted to withstand tensile forces. During processing, water binding molecules like proteoglycans and glycosaminoglycans are almost entirely removed (Figure 5a), which explains the absence of viscoelastic behavior. In such scaffolds in-growing cells need to produce the specific extracellular matrix of meniscus, in order to gain “meniscal biomechanical properties” . The Actifit as a fully artificial construct (Figure 5b) also lacks the viscoelastic properties of normal meniscus tissue. The polymer implants induced fibrous ingrowth with cartilaginous areas, which resembled neo-meniscal tissue. Though there have been little data in large weight-bearing models such as the ovine, it might be that the chondroprotective effects are more due to an improvement of the contact area than biologic effects . A study by Welsing et al.  found that the porous polymer implants were fully integrated after 2 years in a dog model. Still there were no viable cells in all parts of the meniscus and their biomechanical properties were intermediate to normal meniscus and the scaffold before implantation. In contrast, Maher et al  could prove that the implantation of a polyurethane scaffold in a partial meniscectomy ovine model promotes tissue ingrowth without affecting the articulating cartilage. Transferring these results to daily clinical life, the artificial scaffolds must be protected from weight bearing until cellular in-growth and remodeling processes can occur and allow for maturation of the constructs and induce a gain of biomechanical strength. Considering their minor biomechanical properties at time zero partial weight bearing or even load removal of the operated leg for 6 weeks would be recommendable. Possibly, the cellularization of scaffolds before implantation might be beneficial in terms of biomechanical behavior and biocompatibility [13, 30].
We believe that a large-animal model is necessary in meniscus experiments: First there are anatomical similarities between the human knee and the knee of a quadruped, like the sheep [35, 36]. This includes the existence of cruciate ligaments, the menisci and the asymmetrical collateral ligaments with a bicondylar distal femur . There is also a visible similarity when comparing cross-sectional shape of the midpart region of sheep and healthy human menisci (Figure 5). Second, viscoelastic meniscus behavior in a large animal model is more comparable to human menisci than it is in small animals due to the limited size of rabbit or rat meniscus. Third, small animals like rabbit or rat walk in a much more flexed knee position compared to humans or larger animals. Gait analysis in goats showed that these larger quadrupeds show similar knee flexion and full extension during walking or trotting and have comparable values to human patients . Finally, Chevrier et al.  found that in the important histological terms of vascularization and cell density the ovine menisci were much more similar to human meniscus samples than rabbit menisci underlining our results.
Nevertheless, this study has several limitations. The tested meniscus samples were all taken from the intermediate part, so that no information about the anterior or posterior meniscal part can be made. In addition, the histological results are only descriptive- a systematic analysis is missing.
Due to the small size with difficult meniscus surgery in small animals we performed no biomechanical testing of small animals. However, there are some studies using the rabbit model for basic science research [39, 40].
In summary, we could show that from the large animals (bovine, ovine and porcine) the biomechanical and histological properties of the ovine meniscus samples revealed the highest similarity to human samples, in terms of stiffness, compression and residual force and that therefore the desirable animal model for meniscus research would be ovine meniscus tissue. Artificial scaffolds have inferior biomechanical properties and need to be protected from load transfer until remodeling processes might improve the viscoelastic properties.
This project was partially supported by the BMBF project QuReGe. There are no conflicts of interest.
- Aagaard H, Verdonk R: Function of the normal meniscus and consequences of meniscal resection. Scand J Med Sci Sports. 1999, 9 (3): 134-140.View ArticlePubMedGoogle Scholar
- Jones JC, Burks R, Owens BD, Sturdivant RX, Svoboda SJ, Cameron KL: Incidence and risk factors associated with meniscal injuries among active-duty US military service members. J Athl Train. 2012, 47 (1): 67-73.PubMedPubMed CentralGoogle Scholar
- Petty CA, Lubowitz JH: Does arthroscopic partial meniscectomy result in knee osteoarthritis? A systematic review with a minimum of 8 years´follow-up. Arthroscopy. 2011, 27 (3): 419-423. 10.1016/j.arthro.2010.08.016.View ArticlePubMedGoogle Scholar
- Noble J, Hamblen DL: The pathology of the degenerate meniscus lesion. J Bone Joint Surg Br. 1975, 57 (2): 180-186.PubMedGoogle Scholar
- Roos H, Laurén M, Adalberth T, Roos EM, Jonsson K, Lohmander LS: Knee osteoarthtritis after meniscectomy: prevalence of radiographic changes after twenty-one years, compared with matched controls. Arthritis Rheum. 1998, 41 (4): 687-693. 10.1002/1529-0131(199804)41:4<687::AID-ART16>3.0.CO;2-2.View ArticlePubMedGoogle Scholar
- Hede A, Larsen E, Sandberg H: The long term outcome of open total and partial meniscectomy related to the quantity and site of the meniscus removed. Int Orthop. 1992, 16 (2): 122-125.View ArticlePubMedGoogle Scholar
- Saltzman BM, Bajaj S, Salata M, Daley EL, Strauss E, Verma N, Cole BJ: Prospective long-term evaluation of meniscal allograft transplantation procedure: a minimum of 7-year follow-up. J Knee Surg. 2012, 25 (2): 165-175.View ArticlePubMedGoogle Scholar
- Yoon JR, Kim TS, Wang JH, Yun HH, Lim H, Yang JH: Importance of independent measurement of width and length of lateral meniscus during preoperative sizing for meniscal allograft transplantation. Am J Sports Med. 2011, 39 (7): 1541-1547. 10.1177/0363546511400712.View ArticlePubMedGoogle Scholar
- Sandmann GH, Eichhorn S, Vogt S, Adamczyk C, Aryee S, Hoberg M, Milz S, Imhoff AB, Tischer T: Generation and characterization of a human acellular meniscus scaffold for tissue engineering. J Biomed Mater Res A. 2009, 91 (2): 567-574.View ArticlePubMedGoogle Scholar
- Amin S, Niu J, Guermazi A, Grigoryan M, Hunter DJ, Clancy M, LaValley MP, Genant HK, Felson DT: Cigarette smoking and the risk for cartilage loss and knee pain in men with knee osteoarthritis. Ann Rheum Dis. 2007, 66 (1): 18-22.View ArticlePubMedPubMed CentralGoogle Scholar
- Binnet MS, Akan B, Kaya A: Lyophilised medial meniscus transplantation in ACL-deficient knees: a 19-year follow-up. Knee Surg Sports Traumatol Arthrosc. 2012, 20: 109-113. 10.1007/s00167-011-1556-3.View ArticlePubMedGoogle Scholar
- Martinek V, Ueblacker P, Braeun K, Nitschke S, Mannhardt R, Specht K, Gansbacher B, Imhoff AB: Second generation of meniscus transplantation: in-vivo study with tissue engineered meniscus replacement. Arch Orthop Trauma Surg. 2006, 126 (4): 228-234. 10.1007/s00402-005-0025-1.View ArticlePubMedGoogle Scholar
- Petri M, Ufer K, Toma I, Becher C, Liodakis E, Brand S, Haas P, Liu C, Richter B, Haasper C: Effects of perfusion and cyclic compression on in vitro tissue engineered meniscus implants. Knee Surg Sports Traumatol Arthrosc. 2012, 20 (2): 223-231. 10.1007/s00167-011-1600-3.View ArticlePubMedGoogle Scholar
- Zaffagnini S, Marcheggiani Mucciolli GM, Lopomo N, Bruni D, Giordano G, Ravazzolo G, Molinari M, Marcacci M: Prospective long-term outcomes of the medial collagen meniscus implant versus partial medial meniscectomy: a minimum 10-year follow-up study. Am J Sports Med. 2011, 39 (5): 977-985. 10.1177/0363546510391179.View ArticlePubMedGoogle Scholar
- Rodkey WG, DeHaven KE, Montgomery WH, Baker CL, Beck CL, Hormel SE, Steadman JR, Cole BJ, Briggs KK: Comparison of the collagen meniscus implant with partial meniscectomy. A prospective randomized trial. J Bone Joint Surg Am. 2008, 90 (7): 1413-1426. 10.2106/JBJS.G.00656.View ArticlePubMedGoogle Scholar
- Hirschmann MT, Keller L, Hirschmann A, Schenk L, Berbig R, Luthi U, Amsler F, Friederich NF, Arnold MP: One-year clinical and MR imaging outcome after partial meniscal replacement in stabilized knees using a collagen meniscus implant. Knee Surg Sports Traumatol Arthrosc. 2013, 21 (3): 740-747. 10.1007/s00167-012-2259-0.View ArticlePubMedGoogle Scholar
- Galley NK, Gleghorn JP, Rodeo S, Warren RF, Maher SA, Bonassar LJ: Frictional properties of the meniscus improve after scaffold-augmented repair of partial meniscectomy: a pilot study. Clin Orthop Relat Res. 2011, 469 (10): 2817-2823. 10.1007/s11999-011-1854-6.View ArticlePubMedPubMed CentralGoogle Scholar
- Hannink G, van Tienen TG, Schouten AJ, Buma P: Changes in articular cartilage after meniscectomy and meniscus replacement using a biodegradable porous polymer implant. Knee Surg Sports Traumatol Arthrosc. 2011, 19 (3): 441-451. 10.1007/s00167-010-1244-8.View ArticlePubMedGoogle Scholar
- Verdonk P, Beaufils P, Bellemans J, Dijan P, Heinrichs EL, Huysse W, Laprell H, Siebold R, Verdonk R: Succesful treatment of painful irreparable partial meniscal deficits with a polyurethane scaffold: two- year safety and clinical outcomes. Am J Sports Med. 2012, 40 (4): 844-853. 10.1177/0363546511433032.View ArticlePubMedGoogle Scholar
- De Connick T, Huysse W, Willemot L, Verdonk R, Verstraete K, Verdonk P: Two-year follow-up study on clinical and radiological outcomes of polyurethane meniscal scaffolds. Am J Sports Med. 2013, 41 (1): 64-72. 10.1177/0363546512463344.View ArticleGoogle Scholar
- Messner K, Gao J: The menisci of the knee joint. Anatomical and functional characteristics, and a rationale for clinical treatment. J Anat. 1998, 193 (Pt 2): 161-178.View ArticlePubMedPubMed CentralGoogle Scholar
- Makris EA, Hadidi P, Athanasiou KA: The knee meniscus: structure-function, pathophysiology, current repair techniques, and prospects for regeneration. Biomaterials. 2011, 32 (30): 7411-7431. 10.1016/j.biomaterials.2011.06.037.View ArticlePubMedPubMed CentralGoogle Scholar
- Fithian DC, Kelly MA, Mow VC: Material properties and structure-function relationships in the menisci. Clin Orthop Relat Res. 1990, 252: 19-31.PubMedGoogle Scholar
- Chevrier A, Nelea M, Hurtig MB, Hoemann CD, Buschmann MD: Meniscus structure in human, sheep, and rabbit for animal models of meniscus repair. J Orthop Res. 2009, 27 (9): 1197-1203. 10.1002/jor.20869.View ArticlePubMedGoogle Scholar
- Sweigart MA, Zhu CF, Burt DM, DeHoll PD, Agrawal CM, Clanton TO, Athanasiou KA: Intraspecies and interspecies comparison of the compressive properties of the medial meniscus. Ann Biomed Eng. 2004, 32 (11): 1569-1579.View ArticlePubMedGoogle Scholar
- Joshi MD, Suh JK, Marui T, Woo SL: Interspecies variation of compressive biomechanical properties of the meniscus. J Biomed Mater Res. 1995, 29 (7): 823-828. 10.1002/jbm.820290706.View ArticlePubMedGoogle Scholar
- Maes JA, Haut Donahue TL: Time dependent properties of bovine meniscal attachments: stress relaxation and creep. J Biomech. 2006, 39 (16): 3055-3061. 10.1016/j.jbiomech.2005.09.025.View ArticlePubMedGoogle Scholar
- Zantop T, Temmig K, Weimann A, Eggers AK, Raschke MJ, Petersen W: Elongation and structural properties of meniscal repair using suture techniques in distraction and shear force scenarios: biomechanical evaluation using a cyclic loading protocol. Am J Sports Med. 2006, 34 (5): 799-805.View ArticlePubMedGoogle Scholar
- Staerke C, Brettschneider O, Grobel KH, Becker R: Tensile forces on sutures in the human lateral knee meniscus. Knee Surg Sports Traumatol Arthrosc. 2009, 17 (11): 1354-1359. 10.1007/s00167-009-0875-0.View ArticlePubMedGoogle Scholar
- Maier D, Braeun K, Steinhauser E, Ueblacker P, Oberst M, Kreuz PC, Roos N, Martinek V, Imhoff AB: In vitro analysis of an allogenic scaffold for tissue-engineered meniscus replacement. J Orthop Res. 2007, 25 (12): 1598-1608. 10.1002/jor.20405.View ArticlePubMedGoogle Scholar
- Brophy RH, Cottrell J, Rodeo SA, Wright TM, Warren RF, Maher SA: Implantation of a synthetic meniscal scaffold improves joint contact mechanics in a partial meniscectomy cadaver model. J Biomed Mater Res A. 2010, 92 (3): 1154-1161.PubMedGoogle Scholar
- Reguzzoni M, Manelli A, Ronga M, Raspanti M, Grassi FA: Histology and ultrastructure of a tissue-engineered collagen meniscus before and after implantation. J Biomed Mater Res B Appl Biomater. 2005, 74 (2): 808-816.View ArticlePubMedGoogle Scholar
- Welsing RT, van Tienen TG, Ramrattan N, Heijkants R, Schouten AJ, Veth RP, Buma P: Effect on tissue differentiation and articular cartilage degradation of a polymer meniscus implant: a 2-year follow-up study in dogs. Am J Sports Med. 2008, 36 (10): 1978-1989. 10.1177/0363546508319900.View ArticlePubMedGoogle Scholar
- Maher SA, Rodeo SA, Doty SB, Brophy R, Potter H, Foo LF, Rosenblatt L, Deng XH, Turner AS, Wright TM: Evaluation of a porous polyurethane scaffold in a partial meniscal defect ovine model. Arthroscopy. 2010, 26 (11): 1510-1519. 10.1016/j.arthro.2010.02.033.View ArticlePubMedGoogle Scholar
- Osterhoff G, Loffler S, Steinke H, Feja C, Josten C, Hepp P: Comparative anatomical measurements of osseous structures in the ovine and human knee. Knee. 2011, 18 (2): 98-103. 10.1016/j.knee.2010.02.001.View ArticlePubMedGoogle Scholar
- Tischer T, Ronga M, Tsai A, Ingham SJ, Ekdahl M, Smolinski P, Fu FH: Biomechanics of the goat three bundle anterior cruciate ligament. Knee Surg Sports Traumatol Arthrosc. 2009, 17 (8): 935-940. 10.1007/s00167-009-0784-2.View ArticlePubMedGoogle Scholar
- Dye SF: An evolutionary perspective of the knee. J Bone Joint Surg Am. 1987, 69 (7): 976-983.PubMedGoogle Scholar
- Bogey RA, Barnes LA, Perry J: Computer algorithms to characterize individual subject EMG profiles during gait. Arch Phys Med Rehabil. 1992, 73 (9): 835-841.PubMedGoogle Scholar
- Zellner J, Hierl K, Mueller M, Pfeifer C, Berner A, Dienstknecht T, Krutsch W, Geis S, Gehmert S, Kujat R: Stem cell-based tissue-engineering for treatment of meniscal tears in the avascular zone. J Biomed Mater Res B Appl Biomater. 2013, 101 (7): 1133-1142. 10.1002/jbm.b.32922.View ArticlePubMedGoogle Scholar
- Naumann A, Dennis JE, Awadallah A, Carrino DA, Mansour JM, Kastenbauer E, Caplan AI: Immunochemical and mechanical characterization of cartilage subtypes in rabbit. J Histochem Cytochem. 2002, 50 (8): 1049-1058. 10.1177/002215540205000807.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/14/324/prepub
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.