- Research article
- Open Access
Comparison of monoplanar and polyaxial screw fixation systems in percutaneous intermediate fixation for thoracolumbar fractures
BMC Musculoskeletal Disorders volume 23, Article number: 172 (2022)
The newly developed monoplanar pedicle screws (MPPSs) can mobile in axial plane but fixed in the sagittal plane, which holds potential to combine ease of rod placement with sagittal plane strength theoretically. So far, few clinical studies focused on the outcomes of MPPSs for treatment of thoracolumbar fractures (TLFs). The aim of this study was to compare the efficacy of MPPSs to polyaxial pedicle screws (PAPSs) in percutaneous intermediate fixation of TLFs.
Seventy-eight patients who sustained TLFs without neurological deficits and underwent percutaneous intermediate fixation using MPPSs (40 patients) or PAPSs (38 patients) with a minimum 1-year follow-up were included in this study. The operation time, blood loss, local Cobb angle (LCA), vertebral wedge angle (VWA), anterior body height ratio (ABHR), visual analogue scale (VAS) and Oswestry Disability Index (ODI) were collected.
No significant differences were observed in baseline demographics, clinical characteristics, operation time or blood loss between the two groups (P > 0.05). The postoperative LCA, VWA and ABHR were significantly corrected compared to these parameters preoperatively in both groups (#P < 0.05). The postoperative LCA, VWA and ABHR in the MPPS group were significantly better corrected than those in the PAPS group (*P < 0.05). Furthermore, the correction loss of LCA, VWA and ABHR in the MPPS group was significantly lower than that in the PAPS group (*P < 0.05). However, no significant difference in VAS and ODI scores was observed between the two groups.
MPPSs showed similar efficiency as PAPSs in percutaneous intermediate fixation surgical procedures. More importantly, MPPSs achieved better radiological performance than PAPSs in the correction of TLFs and the prevention of correction loss.
Spine fractures are commonly observed in traffic accidents, height crashes and other high-energy injuries, and they account for 5% of all trauma patients . Approximately 60–70% of all traumatic spinal fractures are thoracolumbar fractures (TLFs, from T11 to L2), which is ascribed to the special biomechanical characteristics of the thoracolumbar spine that translate from the rigid, kyphotic thoracic spine to the mobile, lordotic lumbar spine . Surgical or conservative management of TLFs remains controversial, especially for patients without neurological deficits [3, 4].
In recent years, minimally invasive percutaneous pedicle screw fixation (PPSF) for TLFs has been growing in popularity [5,6,7]. The typical PPSF for TLFs is 4-screw fixation with pedicle screws introduced to one level above and below the injured vertebra. Recently, a series of studies have documented that percutaneous intermediate fixation (PIF), which adds two screws in the fractured vertebra, provides stronger fixation than traditional 4-screw fixation [2, 8, 9]. In addition, PIF is more effective in restoring and maintaining fractured vertebral height [2, 9,10,11]. The thoracolumbar spine is the junctional area from a narrow thoracic spine to a wider lumbar spine; it requires inserting screws with high alignment, and slight deviation of the ipsilateral three screws might make it difficult to insert the longitudinal rod, especially using fixed-axis pedicle screws in PIF (6-screw fixation) [12, 13]. The polyaxial pedicle screw (PAPS) increases the degrees of freedom at the screw-rod interface and provides greater ease for rod insertion, making PAPSs more favourable than traditional fixed-axis pedicle screws in minimally invasive surgeries [12, 13]. PAPSs show less stiffness in the sagittal plane and inferior fracture reduction compared to fixed-axis pedicle screws in TLFs [12, 14].
Recently, a novel type of pedicle screw, monoplanar pedicle screw (MPPS), has been introduced. MPPS is designed to be mobile in the axial plane but fixed in the sagittal plane. Thus, MPPS behaves as a fixed-axis pedicle screw in the sagittal plane and a PAPS in the axial plane. A series of recent biomechanical studies have demonstrated that MPPSs significantly increase stiffness in the sagittal plane compared to PAPSs [15,16,17]. Theoretically, MPPSs may combine the ease of rod insertion with sagittal plane strength.
To our knowledge, few clinical studies have focused on the outcomes of MPPSs for the treatment of TLFs. The objectives of this study were to compare the efficacy of MPPSs to PAPSs in PIF for TLFs without neurological deficits.
A retrospective study was adopted to review patients who underwent PIF surgery using MPPSs or PAPSs from January 2017 to May 2020. A total of 78 patients were enrolled in this study. The patients were divided into two groups: those treated with the MPPS system were included in the MPPS group; those treated with the PAPS system were included in the PAPS group. All protocols were approved by the Ethics Committee of General Hospital of Central Theater Command (approval number: 040) and were performed in compliance with the Helsinki Declaration.
Inclusion and exclusion criteria
The inclusion criteria for patients were as follows: traumatic fracture of T11-L2; type A3 and A4 with or without type B2 in AO classification of spinal fracture ; without neurologic deficit; age 18 to 60 years; less than 2 weeks between trauma and surgery; and followed up for more than 12 months. Patients with the following criteria were excluded: fractures of more than 2 vertebrae; presence of nerve injury symptoms; pregnancy; pathologic or osteoporotic fracture; bilateral pedicle fracture; and a history of previous spinal surgery. The general information of the patients is summarized in Table 1.
All surgeries were performed by fully qualified spine surgeons as reported previously (Fig. 1) [19,20,21]. Under general endotracheal anaesthesia, all patients were operated in the prone position on a Jackson operating table with chest and pelvis supported by a pad and abdomen suspended. The location of the pedicles of the fractured level, and one level above and below the injured vertebra were marked according to the posteroanterior fluoroscopy. One surgeon applied manual forces on the fractured vertebra by his hands under intermittent C-arm fluoroscopic guidance (Fig. 1b). Continuous neuromonitoring was utilized to monitor neural function. Skin incisions were made 1.5-cm lateral from the marks of those pedicles. Each pedicle was cannulated by a Jamshidi needle with proper direction and depth. A guidewire was placed into the vertebral body through the needle. The pedicle screw was inserted into the pedicle and vertebral body along the wire. There were six percutaneous pedicle screws (one level above and below the injured vertebra as well as the fractured level) implanted into each patient’s thoracolumbar region in both MPPS group and PAPS group . After implantation of all six screws, rods with appropriate length and bending were inserted (Fig. 1d). The TLF was further corrected by applying a hyperlordosing force through the posterior elements before tightening the screws (Fig. 1e). Posteroanterior fluoroscopy and lateral fluoroscopy were conducted to verify the reduction effect (Fig. 1f). The duration of operation time and the amount of blood loss were recorded. Postoperative management was performed as reported previously [23, 24]. All patients were encouraged to partake in ambulatory activities while wearing a brace 3 days after surgery.
Low back pain was evaluated by the VAS (0–10 scale). The functional outcomes were assessed by ODI as described previously . The VAS score and ODI were assessed preoperatively, 5 days postoperation, 1 month postoperation and at the last follow-up.
Thoracolumbar anterior-posterior and lateral X-rays in addition to thoracolumbar computed tomography (CT) and magnetic resonance imaging (MRI) were obtained before the operation to evaluate the fracture. The LCA, VWA and ABHR (Fig. 2) were evaluated by lateral X-rays. The LCA was measured between the superior endplate of the vertebra above the injured vertebra and inferior endplate of the vertebra below the injured vertebra. VWA was defined as the Cobb angle of the fractured vertebra, and ABHR was defined as the percentage of the anterior body high of fractured vertebra to the mean value of the adjacent vertebrae. All data were measured by two independent observers who were blinded to the group assignment.
Continuous variables are presented as the mean ± SD and were evaluated by Student’s t test. Categorical variables are presented as numbers and were compared utilizing the chi-square test or Fisher’s exact test. A statistical significance level of P < 0.05 was applied. Statistical analyses were conducted by SPSS 23.0 (IBM, New York, USA).
In total, 78 patients with traumatic TLFs were included as follows: 40 patients received MPPSs, and 38 patients received PAPSs. The demographic data and clinical characteristics of all patients are shown in Table 1. There was no significant difference in age, sex, fracture mechanism, fracture level or AO classification between the two groups (P > 0.05). The average TLICS was 4.0 ± 0.7 (3–5) in both groups (P > 0.05). The average time of follow-up was 15.9 ± 3.2 (12.2–24.8) months for the MPPS group and 16.6 ± 3.9 (12.5–26.3) months for the PAPS group (P > 0.05). The mean operation time was 80.5 ± 15.4 (64.3–119.5) min in the MPPS group and 78.3 ± 16.2 (63.8–125.5) min in the PAPS group (P > 0.05). The blood loss was 87.5 ± 37.1 (45–155) ml in the MPPS group and 80.3 ± 34.0 (40–165) ml in the PAPS group (P > 0.05).
As shown in Table 2, there was no significant difference in the preoperative LCA, VWA or ABHR between the MPPS and PAPS groups (P > 0.05). As expected, the postoperative LCA, VWA and ABHR were significantly corrected compared to these parameters preoperatively in both groups (Table 2, Fig. 3 and Fig. 4, #P < 0.05). The postoperative LCA in the MPPS group (6.0° ± 5.1°) was significantly lower than that in the PAPS group (9.2° ± 4.7°, *P < 0.05). Similarly, the postoperative VWA and ABHR in the MPPS group were also significantly better corrected than those in the PAPS group (*P < 0.05). Both the MPPS and PAPS groups showed correction loss with time. Importantly, the LCA, VWA and ABHR in the MPPS group at the last follow-up remained significantly better than those in the PAPS group (*P < 0.05). In addition, significant differences were also found in the correction loss of LCA, VWA and ABHR between the two groups (Table 2, *P < 0.05).
The VAS for back pain and ODI scores were significantly improved following surgery in both groups (Table 3, #P < 0.05) and gradually improved with time. However, no significant differences were found in VAS and ODI between the two groups at any time point (Table 3, P > 0.05). There were no major vascular injuries or neurologic complications. No patient required revision for correction loss or instrumentation failure at the last follow-up.
Traumatic TLFs are commonly observed in traffic accidents, height crashes and other high-energy injury situations . Open surgery has been recommended for patients with neurological deficits and unstable TLFs to decompress the nerve and stabilize the spine. For patients without neurological deficits, decompression is not required, and the treatment focuses on recovering the height of the fractured vertebra, restoring the stability of the spine and avoiding complications due to posttraumatic kyphosis and prolonged bed rest . Minimally invasive PPSF exhibits better fracture reduction and long-term clinical outcomes than nonoperative treatment, and it provides three-column fixation similar to open surgery and shows less soft tissue injury, lower infection risk, less postoperative pain and shorter rehabilitation time than traditional open surgery [5,6,7]. Therefore, PPSF has been growing in popularity for the treatment of TLFs . With the development of PPSF, more recent studies have demonstrated that PIF, which adds two screws to the fractured vertebra, is more effective in restoring and maintaining fractured vertebral height than the classic 4-screw PPSF . In the present study, all the patients received PIF using six pedicle screws. Our results showed that the LCA, VWA and ABHR were significantly corrected by surgery in both groups. Our results provided further evidence that PIF effectively recovers the height of fractured vertebrae and corrects kyphosis.
In PIF surgery, high alignment of the ipsilateral three screws is required for successful installation of the longitudinal connecting rod, especially using fixed-axis pedicle screws. Deviation in the position of screw placement may increase the difficulty of surgery and prolong the operation time [12, 13]. PAPSs increase the degrees of freedom at the screw-rod interface and provide greater ease for rod insertion. However, it has been reported that fixed-axis pedicle screws provide better correction of deformities than PAPSs . In an effort to combine the relative advantages of fixed-axis pedicle screws and PAPSs, MPPSs were developed. MPPSs are mobile in the axial plane (see Additional file 1), which may facilitate rod insertion and improve surgical efficiency. In the present study, the mean operation time and blood loss showed no significant difference between the MPPS group and the PAPS group (P > 0.05). These results indicated that the operating efficiencies of MPPSs and PAPSs were comparable.
MPPSs are rigid in the sagittal plane and behave as fixed-axis screws in the sagittal plane (see Additional file 1) , potentially achieving better fracture reduction and kyphotic angle correction similar to fixed-axis screws. In line with this hypothesis, our results demonstrated that the LCA, VWA, and ABHR were better corrected by MPPSs than PAPSs. In addition, MPPSs showed less correction loss than PAPSs with prolonged follow-up.
In the present study, statistically significant differences were not observed in the VAS and ODI scores between the two groups. The following reasons might account for this lack of statistical difference: 1) both MPPS and PAPS fixation procedures are minimally invasive procedures that cause little damage to paraspinal soft tissues [6, 26]; and 2) one of the most important reasons for functional defects after TLF is pain caused by the jiggle of the fractured vertebra under loading-bearing stress, buckling stress and rotation stress, and both MPPS and PAPS fixation provide sufficient strength to stabilize the fractured vertebra and minimize the pain caused by the jiggle of the fractured vertebra. Further studies with long-term follow-up after the removal of instrumentations might clarify whether the promising radiological results in the MPPS group translate to superior functional outcomes.
There were several limitations to this study. First, this was a retrospective study without randomization, potentially resulting in selection bias. A randomized, prospective study is warranted to further confirm these findings. Second, this was a one-center study, and the sample size remained small. Third, the follow-up time was relatively short without removal of instrumentations. It would be interesting to investigate the long-term functional outcomes, correction loss and adjacent segment degeneration after the removal of instrumentations in future studies.
In summary, MPPSs and PAPSs showed similar operation times, blood loss and clinical outcomes in PIF of thoracolumbar fractures. The LCA, VWA, and ABHR were better corrected by MPPSs than PAPSs. In addition, MPPSs showed less correction loss than PAPSs with prolonged follow-up. These results highlighted that MPPS fixation system was a highly efficient fixation system as PAPS in PIF surgical procedures and MPPSs achieved better radiological performance than PAPSs in the correction of TLFs and the prevention of correction loss.
Availability of data and materials
The datasets analysed during the current study are available from the corresponding author on reasonable request.
Anterior body height ratio
Local Cobb angle
Magnetic resonance imaging
Monoplanar pedicle screws
Oswestry Disability Index
Polyaxial pedicle screws
Percutaneous intermediate fixation
Visual analogue scale
Vertebral wedge angle
Liu P, Yao Y, Liu M, Fan W, Chao R, Wang Z, et al. Spinal trauma in mainland China from 2001 to 2007: an epidemiological study based on a nationwide database. Spine (Phila Pa 1976). 2012;37(15):1310–5.
Li K, Li Z, Ren X, Xu H, Zhang W, Luo D, et al. Effect of the percutaneous pedicle screw fixation at the fractured vertebra on the treatment of thoracolumbar fractures. Int Orthop. 2016;40(6):1103–10.
Caruso G, Lombardi E, Andreotti M, Lorusso V, Gildone A, Padovani S, et al. Minimally invasive fixation techniques for thoracolumbar fractures: comparison between percutaneous pedicle screw with intermediate screw (PPSIS) and percutaneous pedicle screw with kyphoplasty (PPSK). Eur J Orthop Surg Traumatol. 2018;28(5):849–58.
Aleem IS, Nassr A. Cochrane in CORR(®): surgical versus non-surgical treatment for thoracolumbar burst fractures without neurological deficit. Clin Orthop Relat R. 2016;474(3):619–24.
Siebenga J, Leferink VJM, Segers MJM, Elzinga MJ, Bakker FC, Haarman HJTM, et al. Treatment of traumatic thoracolumbar spine fractures: a multicenter prospective randomized study of operative versus nonsurgical treatment. Spine (Phila Pa 1976). 2016;31(25):2881–90.
Kim D, Lee S, Chung SK, Lee H. Comparison of multifidus muscle atrophy and trunk extension muscle strength: percutaneous versus open pedicle screw fixation. Spine (Phila Pa 1976). 2005;30(1):123–9.
Vanek P, Bradac O, Konopkova R, de Lacy P, Lacman J, Benes V. Treatment of thoracolumbar trauma by short-segment percutaneous transpedicular screw instrumentation: prospective comparative study with a minimum 2-year follow-up. J Neurosurg Spine. 2014;20(2):150.
Dick JC, Jones MP, Zdeblick TA, Kunz DN, Horton WC. A biomechanical comparison evaluating the use of intermediate screws and cross-linkage in lumbar pedicle fixation. J Spinal Disord. 1994;7(5):402–7.
Norton RP, Milne EL, Kaimrajh DN, Eismont FJ, Latta LL, Williams SK. Biomechanical analysis of four- versus six-screw constructs for short-segment pedicle screw and rod instrumentation of unstable thoracolumbar fractures. Spine J. 2014;14(8):1734–9.
Ye C, Luo Z, Yu X, Liu H, Zhang B, Dai M. Comparing the efficacy of short-segment pedicle screw instrumentation with and without intermediate screws for treating unstable thoracolumbar fractures. Medicine. 2017;96(34):e7893.
Liu H, Wang H, Liu J, Li C, Zhou Y, Xiang L. Biomechanical comparison of posterior intermediate screw fixation techniques with hybrid monoaxial and polyaxial pedicle screws in the treatment of thoracolumbar burst fracture: a finite element study. J Orthop Surg Res. 2019;14(1):122.
Wang H, Zhao Y, Mo Z, Han J, Chen Y, Yu H, et al. Comparison of short-segment monoaxial and polyaxial pedicle screw fixation combined with intermediate screws in traumatic thoracolumbar fractures: a finite element study and clinical radiographic review. Clinics. 2017;72(10):609–17.
Liu PLai P, Lin C. A biomechanical investigation of different screw head designs for vertebral derotation in scoliosis surgery. Spine J. 2017;17(8):1171–9.
Chung WH, Eu WC, Chiu CK, Chan CYW, Kwan MK. Minimally invasive reduction of thoracolumbar burst fracture using monoaxial percutaneous pedicle screws: surgical technique and report of radiological outcome. J Orthop Surg-Hong K. 2020;28(1):920548385.
Ye B, Yan M, Zhu H, Duan W, Hu X, Ye Z, et al. Novel screw head design of pedicle screw for reducing the correction loss in the patients with thoracolumbar vertebral fractures: a biomechanical study. Spine (Phila Pa 1976). 2017;42(7):E379–84.
Schroerlucke SR, Steklov N, Mundis GM, Marino JF, Akbarnia BA, Eastlack RK. How does a novel monoplanar pedicle screw perform biomechanically relative to monoaxial and polyaxial designs? Clin Orthop Relat Res. 2014;472(9):2826–32.
Lam FCGroff MW, Alkalay RN. The effect of screw head design on rod derotation in the correction of thoracolumbar spinal deformity: laboratory investigation. J Neurosurg Spine. 2013;19(3):351.
Reinhold M, Audigé L, Schnake KJ, Bellabarba C, Dai L, Oner FC. AO spine injury classification system: a revision proposal for the thoracic and lumbar spine. Eur Spine J. 2013;22(10):2184–201.
Xu Y, Zhou X, Yu C, Cheng M, Dong Q, Qian Z. Effectiveness of postural and instrumental reduction in the treatment of thoracolumbar vertebra fracture. Int Orthop. 2008;32(3):361–5.
Jeon C, Lee Y, Youn S, Lee H, Chung N. Factors affecting postural reduction in posterior surgery for thoracolumbar burst fracture. J Spinal Disord Tech. 2015;28(4):E225–30.
Li L, Dong Y, He Y, Liu X, Tong W. Treatment of thoracolumbar fractures by percutaneous pedicle screw fixation technique combined with three-step reduction. J Neurol Surg A Cent Eur Neurosurg. 2017;78(03):231–7.
Yang P, Chen K, Zhang K, Sun J, Yang H, Mao H. Percutaneous short-segment pedicle instrumentation assisted with O-arm navigation in the treatment of thoracolumbar burst fractures. J Orthop Transl. 2020;21:1–7.
Lyu J, Chen K, Tang Z, Chen Y, Li M, Zhang Q. A comparison of three different surgical procedures in the treatment of type a thoracolumbar fractures: a randomized controlled trial. Int Orthop. 2016;40(6):1233–8.
Wang X, Dai L, Xu H, Chi Y. Kyphosis recurrence after posterior short-segment fixation in thoracolumbar burst fractures. J Neurosurg Spine. 2008;8(3):246.
Lue YJ, Hsieh CL, Huang MH, Lin GT, Lu YM. Development of a Chinese version of the Oswestry Disability Index version 2.1. Spine (Phila Pa 1976). 2008;33(21):2354–60.
Yang M, Zhao Q, Hao D, Chang Z, Liu S, Yin X. Comparison of clinical results between novel percutaneous pedicle screw and traditional open pedicle screw fixation for thoracolumbar fractures without neurological deficit. Int Orthop. 2019;43(7):1749–54.
This work was supported by the National Natural Science Foundation of China (No. 81902213), the Medical Youth Top Talent Project of Hubei Province (No. 48#), the Training Plan for Young and Middle-Aged Medical Backbone Talents in Wuhan (No. 51#) and the Health Commission of Hubei Province Scientific Research Project (No. WJ2021Q054).
Ethics approval and consent to participate
Approval was obtained from Clinical Research Ethics Committee of the General Hospital of Central Theater Command (approval number: 040). Written informed consent was obtained from all patients.
Consent for publication
The authors declare that they have no conflicts of interest.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Huang, L., Xiong, C., Guo, Z. et al. Comparison of monoplanar and polyaxial screw fixation systems in percutaneous intermediate fixation for thoracolumbar fractures. BMC Musculoskelet Disord 23, 172 (2022). https://doi.org/10.1186/s12891-022-05129-8
- Thoracolumbar fracture
- Monoplanar pedicle screw
- Polyaxial pedicle screw
- Percutaneous intermediate fixation
- Percutaneous pedicle screw fixation