- Open Access
A consensus approach toward the standardization of spinal stiffness measurement using a loaded rolling wheel device: results of a Delphi study
BMC Musculoskeletal Disorders volume 22, Article number: 436 (2021)
Spinal stiffness assessment has the potential to become an important clinical measure. Various spinal stiffness-testing devices are available to help researchers objectively evaluate the spine and patient complaints. One of these is VerteTrack, a device capable of measuring posteroanterior displacement values over an entire spinal region. This study aimed to develop a best-practice protocol for evaluating spinal stiffness in human participants using VerteTrack.
Twenty-five individuals with research experience in measuring spinal stiffness, or who were trained in spinal stiffness measurement using the VerteTrack device, were invited to participate in this 3-Round Delphi study. Answers to open-ended questions in Round 1 were thematically analyzed and translated into statements about VerteTrack operation for spinal stiffness measurements. Participants then rated their level of agreement with these statements using a 5-point Likert scale in Rounds 2 and 3. A descriptive statistical analysis was performed. Consensus was achieved when at least 70% of the participants either strongly agreed, agreed, (or strongly disagreed, disagreed) to include a statement in the final protocol.
Twenty participants completed Round 1 (80%). All these participants completed Rounds 2 and 3. In total, the pre-defined consensus threshold was reached for 67.2% (123/183) of statements after three rounds of surveys. From this, a best-practice protocol was created.
Using a Delphi approach, a consensus-based protocol for measuring spinal stiffness using the VerteTrack was developed. This standard protocol will help to improve the accuracy, efficiency, and safety of spinal stiffness measurements, facilitate the training of new operators, increase consistency of these measurements in multicenter studies, and provide the synergy and potential for data comparison between spine studies internationally. Although specific to VerteTrack, the resulting standard protocol could be modified for use with other devices designed to collect spinal stiffness measures.
Low back pain (LBP) is the most burdensome of musculoskeletal conditions globally affecting ~ 7.5% of the world’s population (~ 577 million people) . For up to 90% of people presenting with LBP, the specific cause of their pain cannot be clearly identified resulting in a label of non-specific LBP . The current treatment of LBP mainly focuses on pain management while the causes of pain are rarely addressed. Quantitative assessments of the spine and patient complaints related to LBP may help with the identification of causes, improve the management of this condition, and reduce health care system costs.
Advances in science and technology over the past few decades have made several devices available to objectively assess clinical characteristics of patients including spinal stiffness. Stiffness is considered an important spinal biomechanical measure and has long been recognized by both patients and clinicians as one of the characteristic features of the back . Therefore, stiffness has been widely used in the management of patients with back pain for diagnosis, prognosis, clinical decision-making, and the evaluation of manipulative techniques .
An increase or decrease in spinal stiffness has been found to be related to LBP. Specifically, previous studies have demonstrated that some patients with LBP have abnormal levels of spinal stiffness  and that these patients experience an immediate and sustained decrease in spinal stiffness for 1 week following spinal manipulative therapy [6, 7]. Moreover, researchers reported an increase in posteroanterior (PA) stiffness in participants with LBP compared to when participants had little or no pain, while asymptomatic controls showed insignificant changes in PA stiffness over time . A reduction in stiffness has also been shown to be associated with self-reported measures of disability [6, 9]. These findings suggest that restoration of normal spinal stiffness and mobility plays an important role in some patients with LBP by improving spinal function and reducing pain although a casual relation between stiffness and these outcomes has not been confirmed. Therefore, further exploration of spinal stiffness assessment is warranted. While there are various spinal stiffness-testing devices available to objectively evaluate the spinal complaints [4, 5], there is no standard operating protocol for spinal stiffness measurement.
Having a standard data collection protocol for spinal stiffness assessment would facilitate comparison of devices and data between studies. Our research team developed a novel device, the VerteTrack, to improve on single-site spinal indentation by employing a loaded rolling wheel system. Several identical devices have been manufactured and are in use in multiple research centers over the past 6 years. In this Delphi study, our goal was to develop a best-practice protocol for evaluating spinal stiffness in human participants using VerteTrack, a spinal stiffness measurement device shown to be safe , reliable , and accurate .
This study used a standard Delphi methodology to achieve consensus. The Delphi method is a reliable and structured method of obtaining a consensus of opinion from a group of experts or knowledgeable participants  in areas where existing research is limited. The Delphi method is particularly recommended for areas where controversy, debate, or a lack of clarity exist .
Selection of participants
As our lab manufactured the device in question, we know of all the research centers that possess the device and all the staff who were trained on the device. We contacted these centers and asked them to provide us with an updated contact list of those who were trained/used the device since their initial training session. Thus, all individuals trained in VerteTrack methods and/or having previous experience using the VerteTrack device were invited to participate in the Delphi process (n = 25 individuals from 9 different institutions in 7 different countries). Potential participants were asked to participate in the study if they were willing to participate, have access to the internet over the course of the study, and were able to commit time to complete the surveys. Written consent was obtained from all participants after being informed about the project by adding a consent question to the start of Round 1.
The Delphi survey involved three sequential rounds of deidentified online questionnaires provided over 4 months (Sep-Dec 2020). Study data were collected and managed using REDCap  electronic data capture tools provided by the Women & Children’s Health Research Institute at the University of Alberta. We contacted the research centers that are equipped with the device and asked them to send us the email addresses of those who were trained or collected data using the device. E-mail addresses were then entered into the REDcap website. All potential participants were sent an invitation email to participate in the Delphi process containing a link to the online survey. Participants were requested to complete each questionnaire within 2 weeks. Two automated e-mail reminders per round were sent out to non-responders at 1 week and the day before the due date. If participants were not able to complete the questionnaires within the 2 weeks, they were provided with additional reminders and extra time to respond. Each survey took 20–30 min to complete. Participants were allowed to save their answers and return to complete the questionnaire over several sessions.
Prior to the commencement of this study, consensus was defined when at least 70% of the participants in Rounds 2 and 3 either strongly agreed, agreed, (or strongly disagreed, disagreed) to include a statement in the final protocol. These levels of agreement have been considered appropriate in previous Delphi studies [13, 16,17,18,19]. Figure 1 summarizes the stages of the Delphi method in this study.
In order to improve the structure and readability of questions, the Round 1 questionnaire was first piloted with three colleagues. Based on their feedback, Round 1 questions were revised and finalized. MH and GNK designed the Round 1 of the survey. This round included questions regarding basic demographic information and 21 open-ended questions inquiring about participant recruitment for VerteTrack testing, device safety, instructions given to research participants, and technical issues. This round aimed to review the comprehensiveness and relevance of the items and provide suggestions for the eventual protocol. Items for Round 2 of the survey were generated by comments from the first round that suggested removing, aggregating, or retaining items from the first round.
Only those who completed round 1 were invited to participate in Round 2. In this round, each participant received a survey comprising 171 statements. The goal of this round was to reach consensus on a standard protocol. In Round 2, participants were asked to indicate their anonymous opinion by ranking statements along a five-point Likert scale for agreement (“strongly agree”, “agree”, “neither agree nor disagree”, “disagree”, “strongly disagree”). Additionally, a free-text comment section for each question was available for participants to express any further thoughts or opinions. Round 2 also included four new open-ended questions derived from Round 1. Participants were required to rate every single item to be able to move on with the questionnaire.
Round 3 of the study comprised the same list and grading scale as Round 2 with an additional graphical description of findings from the previous round. The graphic information identified the percentage of total respondents that selected each possible score for the given item in Round 2. The respondents, therefore, were given an opportunity to modify or confirm their answers after viewing the scoring results using the same Likert scale from the previous round. The revised and new statements proposed by participants were added in Round 3 yielding a total of 183 statements. Using the consensus results obtained from Round 3, the authors created a written protocol for use of the VerteTrack device in collecting spinal stiffness measures.
Deidentified data were analyzed by encoding participants with their survey ID numbers. Data from the REDCap tool was downloaded into a Microsoft Excel version 16.45 after each round. Descriptive statistics were used to describe the participants’ demographic characteristics. Responses to open-ended questions in the Round 1 and participants’ comments in Round 2 were thematically analyzed with MH and GNK discussing the qualitative responses. MH, GNK and SF met to discuss the items for the consensus statements in Rounds 2 and 3. The quantitative responses from the participants’ ratings in Rounds 2 and 3 were analyzed descriptively using medians, ranges, and percentages.
Of the 25 individuals invited to participate in this Delphi study, 20 participants completed Round 1 (80% response rate), 20/20 completed Round 2 (100.0% response rate), and 20/20 completed Round 3 (100.0% response rate). The reasons for 5/25 participants not responding to the initial invitation email were not identified. Table 1 presents the demographic characteristics of participants at baseline. Participants had different experiences working with the device that ranged from receiving training to performing measurements of spinal stiffness in a population of 180 patients with back pain.
In total, the pre-defined consensus threshold was reached for 67.2% (123/183) of statements after three rounds of surveys. Results from Round 3 were presented in Table 2. The number of consensus statements under each category was listed in Table 3. Items with 70% or more consensus from Round 3 were used to create the best practice protocol for the VerteTrack device (Additional file 1).
In this Delphi study, 20 panelists reached consensus on the majority of items relating to VerteTrack spinal stiffness measurements covering a wide range of domains including recruitment criteria, familiarization procedure, instructions for participants/ operators, technical issues, and safety. This is the first time, to our knowledge, that consensus has been used to obtain a common protocol on instrumented spinal stiffness measurements.
It is important to stress that the key feature of the approach used in this study is the consensus of individuals in the field of spinal manipulative therapy and low back pain research who had experienced working with VerteTrack. Therefore, the intent was not to find “the best” protocol for measuring spinal stiffness or to present an instrument as “the only” mechanical method for measuring spinal stiffness. Our goal was to develop a standard protocol for measuring spinal stiffness using a loaded rolling wheel device that could be used as a common resource in future studies.
The surveys identified some previously known considerations when measuring stiffness including the participant’s testing position, trunk muscles contraction, intra-abdominal pressure, respiratory cycle, and relocation of target spinal landmarks [4, 5]. This supports the quality and validity of our participants’ answers as these items have been developed over years in this field and the literature. For instance, one of our participant’s recommendations was to ask the patient to relax their back muscles during the assessment which is in line with an early study that showed spinal extensor muscle activities could induce changes in the mechanical responses to posteroanterior stiffness testing . Furthermore, the surveys identified other factors not described previously in the literature including optimizing participant’s safety, a definition for a good/ bad trial, procedures to ensure a good trial, placing the device over the test area, instructions for reaching the same position in case of multiple assessments, and fixing software program crashes. This emphasizes the importance of group opinion over that of individuals for bringing new topics into focus that can be validated and studied in future works.
Interestingly, there was one specific area where no agreement was reached: the exclusion of pregnant participants from spinal stiffness measurements. One explanation for this lack of agreement is that different respondents may have different experiences in this area through diverse research designs that would, or would not, allow participants to be enrolled at different stages of pregnancy. This speculation is supported by studies to date that have employed VerteTrack. Of six studies using VerteTrack in human participants to date, three excluded pregnant participants [11, 21, 22], one excluded pregnant participants in the second or third trimester of pregnancy  and the remaining studies did not mention pregnancy at all [23, 24].
All items for which consensus was reached were consolidated into a final best practice protocol (Additional file 1) for using the VerteTrack. The resulting standard protocol is expected to improve the accuracy and efficiency of spinal stiffness measurements using the VerteTrack, facilitate the training of new operators, increase consistency of these measurements in multicenter studies, and finally provide the synergy and potential for data comparison between spine studies internationally. Our final protocol provides directions for researchers and clinicians who use the VerteTrack to measure spinal stiffness. However, caution should be used if between-patient comparisons are made (for many reasons including differences in plinth rigidity as well as between-person variations). The final protocol could be useful for other technologies that assess stiffness and even manual assessment of spinal stiffness. We encourage researchers in this area to review this protocol and consider adopting it for their own purpose. While the technical part of the protocol explaining how to operate the device may not be useful for manual assessments or devices that test participants in sitting position, however, some general information for spinal stiffness measurements has been provided and may be of benefit.
Strengths and limitations
The strengths of this study include the development of a consensus-based protocol based on 80% of the global population of persons with VerteTrack training and experience for Round 1 and 100% follow-up responses for Rounds 2 and 3. The relative heterogeneity in our participants may enhance the generalizability of the protocol and may have ensured that a greater spectrum of opinions was considered. The initial pilot survey improved the structure and readability of the questions before executing the full-scale project. In addition, Round 1 of our Delphi study provided the possibility of open responses and gave the participants the freedom to elaborate on the research topic which may increase the richness of the data collected. Although author bias cannot be completely eliminated from this type of research, it was minimized through implementing a Delphi consensus process using anonymous participant ratings and comments. The deidentification anonymity of participants’ answers to the questions also provided more open and honest feedback and prevented response bias.
It is acknowledged that the Delphi method itself has inherent limitations including Level V in the hierarchy of evidence-based medicine and the small sample size required. Although the final protocol was developed based on Delphi participants’ responses to 3 rounds of questions, it was not distributed to them for approval at the end of the study. Further, lack of interaction between participants in the Delphi (e.g., face-to-face meetings) may deprive panelists of exchanging important information, such as clarification of reasons for disagreements.
Using a Delphi approach, a consensus-based protocol for measuring spinal stiffness using the VerteTrack was developed. This standard protocol was designed to i) improve the accuracy, efficiency, and safety of spinal stiffness measurements using the VerteTrack, ii) facilitate the training of new operators iii) increase consistency of these measurements in multicenter studies, and iv) provide the synergy and potential for data comparison between spine studies internationally.
Availability of data and materials
The datasets used and analyzed during the current study are available from the corresponding author on reasonable request.
Low Back Pain
Research Electronic Data Capture
Wu A, March L, Zheng X, Huang J, Wang X, Zhao J, et al. Global low back pain prevalence and years lived with disability from 1990 to 2017: estimates from the global burden of disease study 2017. Ann Transl Med. 2020;8(6):299. https://doi.org/10.21037/atm.2020.02.175.
Maher C, Underwood M, Buchbinder R. Non-specific low back pain. Lancet. 2017;389(10070):736–47. https://doi.org/10.1016/S0140-6736(16)30970-9.
Colloca CJ. The place of chiropractic care in the treatment of low back pain. In: Szpalski M, Gunzburg R, Rydevik B, Le Huec JC, Mayer H, editors. Surgery for Low Back Pain. Springer, Berlin, Heidelberg; 2010: 85–94. https://doi.org/10.1007/978-3-642-04547-9_12.
Snodgrass SJ, Haskins R, Rivett DA. A structured review of spinal stiffness as a kinesiological outcome of manipulation: its measurement and utility in diagnosis, prognosis and treatment decision-making. J Electromyogr Kinesiol. 2012;22(5):708–23. https://doi.org/10.1016/j.jelekin.2012.04.015.
Wong AYL, Kawchuk GN. The clinical value of assessing lumbar posteroanterior segmental stiffness: a narrative review of manual and instrumented methods. PM R. 2017;9(8):816–30. https://doi.org/10.1016/j.pmrj.2016.12.001.
Wong AYL, Parent EC, Dhillon SS, Prasad N, Kawchuk GN. Do participants with low back pain who respond to spinal manipulative therapy differ biomechanically from nonresponders, untreated controls or asymptomatic controls? Spine (Phila Pa 1976). 2015;40(17):1329–37. https://doi.org/10.1097/BRS.0000000000000981.
Jun P, Pagé I, Vette A, Kawchuk G. Potential mechanisms for lumbar spinal stiffness change following spinal manipulative therapy: a scoping review. Chiropr Man Therap. 2020;28(1):15. https://doi.org/10.1186/s12998-020-00304-x.
Latimer J, Lee M, Adams R, Moran CM. An investigation of the relationship between low back pain and lumbar posteroanterior stiffness. J Manip Physiol Ther. 1996;19(9):587–91.
Fritz JM, Koppenhaver SL, Kawchuk GN, Teyhen DS, Hebert JJ, Childs JD. Preliminary investigation of the mechanisms underlying the effects of manipulation: exploration of a multivariate model including spinal stiffness, multifidus recruitment, and clinical findings. Spine (Phila Pa 1976). 2011;36(21):1772–81. https://doi.org/10.1097/BRS.0b013e318216337d10.1097/BRS.0b013e318216337d
Brown BT, Blacke A, Carroll V, Graham PL, Kawchuk G, Downie A, et al. The comfort and safety of a novel rolling mechanical indentation device for the measurement of lumbar trunk stiffness in young adults. Chiropr Man Ther. 2017;25(1):1-10. https://doi.org/10.1186/s12998-017-0153-z.
Hadizadeh M, Kawchuk GN, Parent E. Reliability of a new loaded rolling wheel system for measuring spinal stiffness in asymptomatic participants. BMC Musculoskelet Disord. 2019;20(1):176. https://doi.org/10.1186/s12891-019-2543-y.
Young A, Swain MS, Kawchuk GN, Wong AYL, Downie AS. The bench-top accuracy of the VerteTrack spinal stiffness assessment device. Chiropr Man Therap. 2020;28(1):42. https://doi.org/10.1186/s12998-020-00331-8.
Hasson F, Keeney S, Mckenna H. Research guidelines for the Delphi survey technique. J Adv Nurs. 2000;32(4):1008–15. https://doi.org/10.1046/j.1365-2648.2000.t01-1-01567.x.
Iqbal S, Pipon-Young L. The Delphi method. The Psychologist. 2009;22(7):598–601.
Harris PA, Taylor R, Thielke R, Payne J, Gonzalez N, Conde JG. Research electronic data capture (REDCap)-a metadata-driven methodology and workflow process for providing translational research informatics support. J Biomed Inform. 2009;42(2):377–81. https://doi.org/10.1016/j.jbi.2008.08.010.
Keeney S, Hasson F, Mckenna H. Consulting the oracle: ten lessons from using the Delphi technique in nursing research. J Adv Nurs. 2006;53(2):205–12. https://doi.org/10.1111/j.1365-2648.2006.03716.x.
Keeney S, Hasson F, Mckenna H. The Delphi technique in nursing and health research. West Sussex: Wiley-Blackwell; 2011. https://doi.org/10.1002/9781444392029.
Wells C, Kolt GS, Marshall P, Bialocerkowski A. The definition and application of pilates exercise to treat people with chronic low back pain: a Delphi survey of Australian physical therapists. Phys Ther. 2014;94(6):792–805. https://doi.org/10.2522/ptj.20130030.
Akins RB, Tolson H, Cole BR. Stability of response characteristics of a Delphi panel: application of bootstrap data expansion. BMC Med Res Methodol. 2005;5:37. https://doi.org/10.1186/1471-2288-5-37.
Lee M, Svensson NL. Effect of loading frequency on response of the spine to lumbar posteroanterior forces. J Manip Physiol Ther. 1993;16(7):439–46.
Fritz JM, Sharpe J, Greene T, Lane E, Hadizadeh M, Mcfadden M, et al. Optimization of spinal manipulative therapy protocols: a factorial randomized trial within a multiphase optimization framework. J Pain. 2020. https://doi.org/10.1016/j.jpain.2020.11.008.
Nim CG, Kawchuk GN, Schiøttz-Christensen B, O’Neill S. The effect on clinical outcomes when targeting spinal manipulation at stiffness or pain sensitivity: a randomized trial. Sci Rep. 2020;10(1):14615. https://doi.org/10.1038/s41598-020-71557-y.
Pagé I, Kawchuk G. Effects of muscle activity on lumbar spinal stiffness in asymptomatic adults: an investigation using a novel rolling device. Musculoskelet Sci Pract. 2021;52:102301. https://doi.org/10.1016/j.msksp.2020.102301.
Nielsen J, Casper GN, O’Neill S, Boyle E, Hartvigsen J, Kawchuk GN. Self-reports vs. physical measures of spinal stiffness. Peer J. 2020;8:9598. https://doi.org/10.7717/peerj.9598.
The authors would like to thank all the respondents for their contribution to this consensus protocol development.
Ethics approval and consent to participate
This study was approved by the University of Alberta’s Humans Research Ethics Board (Pro00102734). All procedures performed in this study were in accordance with the ethical standards of the institutional research committee and with the declaration of Helsinki. Written informed consent was obtained from all participants in the study.
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Hadizadeh, M., Kawchuk, G. & French, S. A consensus approach toward the standardization of spinal stiffness measurement using a loaded rolling wheel device: results of a Delphi study. BMC Musculoskelet Disord 22, 436 (2021). https://doi.org/10.1186/s12891-021-04313-6
- Spinal stiffness
- Mechanical instruments
- Spinal pain