- Research article
- Open Access
Objective quantification of ligament balancing using VERASENSE in measured resection and modified gap balance total knee arthroplasty
BMC Musculoskeletal Disorders volume 19, Article number: 266 (2018)
Soft tissue balancing which is above all most important factor of total knee arthroplasty, has been performed by subjective methods. Recently objective orthosensor has been developed for compartment pressure measurement. The purpose of this study was: (1) to quantify the compartment pressure of the joint throughout the range of motion during TKA using orthosensor, (2) to determine the usefulness of orthosensor by analyzing correlation between the pressure in both compartment with initial trial and after final implantation, and (3) to evaluate the types and effectiveness of additional ligament balancing procedures to compartment pressure.
Eighty-four patients underwent total knee arthroplasty (TKA) using VERASENSE Knee System. TKA was performed by measured resection and modified gap balance technique. Compartment pressure was recorded on full extension, 30°, 60°, 90° and full flexion at initial (INI), after each additional procedure, and after final (FIN) implantation. “Balanced” knees were defined as when the compartment pressure difference was less than 15 pounds.
Thirty patients (35.7%) showed balanced knee initially and 79 patients (94.0%) showed balance after final implantation. The proportion of balanced knee after initial bony resection, modified gap balancing TKAs showed significantly higher proportion than measured resection TKAs (P = 0.004) On both compartment, the pressure was generally decreased throughout the range of motion. Linear correlation on both compartment showed statistically significant throughout the range on motion, with higher correlation value on the lateral compartment. Total 66 additional ligament balancing procedures were performed.
Using orthosensor, we could obtain 94% quantified balance knee, consequently. And between the techniques, measured resection TKA showed less balanced knee and also required more additional procedures compared to modified gap balancing TKA. Furthermore, with the acquired quantified data during appropriate ligament balancing, the surgeon could eventually reduce the complications associated with soft tissue imbalance in the future.
A successful outcome in total knee arthroplasty depends on various surgical factors including precise bone resection, correct alignment of the components, femoral and tibial rotation, and above all adequate soft tissue balancing [1,2,3]. Technological advances over the past few years have facilitated accurate alignment and rotation in the surgical field via three-dimensionally guided bone resection such as navigation or ROBODOC systems [4, 5]. However, adequate soft tissue balancing still remains a challenge for many surgeons, especially younger surgeons who lack surgical experience. Further, experienced surgeons traditionally obtain soft tissue balance using their own subjective “feeling” rather than a scientific perspective [6,7,8,9]. The soft tissue balancing “feeling” is affected by factors such as surgical experience, patient’s BMI, gender, generalized laxity, degree of joint contracture and even the surgeon’s daily condition [10, 11].
Despite the excellent long-term clinical results of TKA, with reported survival greater than 95% at 15 years [12, 13], a few patients experience failure. Many factors contributing to dysfunction and pain have been reported . Especially, revision arthroplasty due to instability has been estimated in more than 20% each year, which may be due to inappropriate ligament balancing . Soft tissue balancing of the knee during total knee arthroplasty is a significant factor for postoperative patient satisfaction and implant longevity . Unfortunately, until recently, soft tissue balancing, which can lead to catastrophic failure of the implant, has been determined only by surgeon’s judgement with subjective feeling.
Combining the need for objective soft tissue balancing and advanced technology, VERASENSE Knee System (OrthoSensor Inc., Dania Beach, FL) has been introduced, recently. VERASENSE is an orthosensor that enables surgeons to quantify ligament balance based on real-time, evidence-based data during primary and revision TKA (Fig. 1). This disposable device delivers wireless data to an intra-operative monitor to facilitate informed decision-making regarding implant position and soft-tissue releases to improve balance and stability through a full range of motion.
The purpose of this study was: (1) to quantify the compartment pressure of the joint throughout the range of motion during TKA using orthosensor, (2) to determine the usefulness of orthosensor by analyzing correlation between the pressure in both compartment with initial trial and after final implantation, and (3) to evaluate the types and effectiveness of additional ligament balancing procedures to compartment pressure.
Patient demographic data
Between July 2017 and March 2018, 84 patients (84 knees) who showed varus deformed osteoarthritic knees underwent unilateral posterior cruciate ligament-retaining (CR) or -sacrificing (PS) primary total knee arthroplasty using VERASENSE Knee System (OrthoSensor Inc., Dania Beach, FL) in our hospital. The surgical procedure was approved by our institutional review board (IRB) of Chonnam National University Hwasun Hospital, and written informed consent was obtained from all patients. All patients were implanted with the same total knee system (NexGen™, Zimmer, Inc., Warsaw, IN, USA). Bone resection was performed by either measured resection technique using ROBODOC® system or modified gap balancing technique. Among 84 knees, measured resection TKA was performed in 34 patients (17 CR implants and 17 PS implants) and modified gap balancing TKA was performed in 50 patients (all PS implants). Patients were consisted of 11 men and 73 women ranging in age from 51 to 83 years (mean age: 69.6 years old) at the time of surgery. There were 46 left knees and 38 right knees. The mean mechanical axis was 10.5° varus (range 0.5 to 26.7°). The CR implant was used in 17 patients with mean 9.2° varus deformity (range 0.7 to 19.5°) and the PS implant was used in 67 patients with mean 10.8° varus deformity (range 0.5 to 26.7°). Patients younger than 50 years, those who underwent revision TKA, or previous ligamentous reconstruction, osteotomy, or any other procedures, were excluded from the study.
Preoperatively, all patients underwent standard anteroposterior, lateral radiographic examination, Merchant & Lauren’s view, Rosenberg view and standing extremity teleoroentgenography for evaluation of mechanical alignment. Further, we obtained the preoperative valgus and varus stress views to evaluate the ligament laxity of the knee preoperatively, and helical computed tomography (CT) to prepare ROBODOC® planning for measured resection TKA. All the surgical procedures were carried out by a senior author (EKS) using either ROBODOC® (Curexo Technology Corp, Fremont, CA, USA)-assisted TKA or conventional TKA using modified gap balancing technique.
Surgical procedure and VERASENSE application
Bone resection (measure resection and modified gap balance technique)
All knees were exposed with a standard midline incision with medial parapatellar arthrotomy, and the patella was everted laterally. In ROBODOC®-assisted TKA, the patient’s leg was rigidly connected to the robot and bony landmarks of both femur and tibia were registered using a probe. After the registration was successfully accomplished, bone resection was automatically conducted by ROBODOC® according to preoperative planning using ORTHODOC system with measured resection technique. In the coronal plane, distal femur and proximal tibia cutting were used to cut perpendicular to the mechanical axis, with 7° of posterior slope to the mechanical axis of the tibia in the sagittal plane. Femoral rotational alignment was planned perpendicular to the trans-epicondylar axis and the tibial rotational axis was planned parallel to that of the femur . In conventional TKA, after exposing the knee joint, bone resection was performed by tibia first modified gap balance technique. Tibial preparation was carried out using extramedullary cutting guide with proximal tibial resection perpendicular to the mechanical axis with 7° posterior slope. Tibial component rotation was aligned to the line connecting the posterior cruciate ligament insertion site and the medial edge to medial 1/3 of the tibial tuberosity. Distal femoral preparation was carried out by intramedullary rod guide with 5° of valgus, and femoral rotation was determined according to the balanced flexion gap .
Intraoperative compartment pressure evaluation using VERASENSE
The VERASENSE Knee System (OrthoSensor Inc., Dania Beach, FL) is a wireless and disposable articular loading quantification device, which is inserted in the tibial component tray during the surgery. The sensor allows quantification of the intraoperative real-time loading data in both medial and lateral compartments of the knee via the full range of movement (Fig. 2).
Once the bony resection was carried out either with the ROBODOC®-assisted measured resection or modified gap balance technique, initially femoral and tibial trial components were implanted and the VERASENSE sensor was inserted into the tibial component tray for initial (INI) quantitative assessment of compartment pressure. After reducing the patella and closing the capsule with one or two towel clips, a gentle range of motion was performed by the surgeon. The surgeon held the leg in a neutral position and monitored the medial and lateral loading forces from full extension to full flexion. In order to observe additional details, we divided the range of motion into 5 categories and recorded compartment pressure in: full extension, 30°, 60°, 90° and full flexion. Less than differential loading of 15 pounds between the medial and lateral compartments, was considered as adequately “balanced” according to previous studies . After initial ligament balance assessment, if the joint showed imbalance (loading difference > 15 pounds), additional soft tissue releases (s) or bony resection were performed and recorded. After each additional ligament balancing procedure, the VERASENSE was inserted in the tibial tray and compartment pressure was re-measured. The orthosensor was routinely re-zeroed before the insertion to minimize the error due to plastic deformation of the sensor. Resultingly, the compartment pressures were recorded initially right after the bone resection (INI), after each additional procedure, and after the final implantation of the real femoral and tibial components with the bone cement (FIN).
Additional ligament balancing procedures
In “unbalanced” knees, various ligament balancing procedures were performed. For soft tissue balancing procedures, we generally used “pie-crusting” (PC) technique with an 18-guage needle while releasing the ligaments . Medial collateral ligament (MCL) release using the PC technique was most frequently used in soft tissue releases, followed by sub-periosteal superficial MCL release using a narrow curved chisel, lateral collateral ligament (LCL) release using PC technique, and iliotibial band (ITB) release. When the joint was too tight to balance by soft-tissue release alone or in valgus-aligned knee after the initial procedure, additional bony resection was performed. Varus or additional proximal tibial recut was mostly performed followed by varus or additional distal femur recut, additional osteophyte removal on medial compartment and posterior tibial slope resection.
Postoperative radiologic evaluation
For radiologic evaluation, standard anteroposterior, lateral and standing extremity teleoreontgenography were used to evaluate the postoperative alignment. Mechanical axis, femorotibial angle (FTA), and posterior slope of the tibia (PTS) were measured from the postoperative radiograph. Also to evaluate the adequate positioning of the implants, coronal and sagittal inclination of femoral and tibial components (α, β, γ, δ) were measured using anteroposterior and lateral radiographs. To determine the intra-observer variation, the radiographic measurement was repeated by the author (observer A) after 1 week. To determine the inter-observer variation, the measurements were performed by 3 experts: the author (observer A), another knee orthopedic surgeon (observer B), and a radiologist (observer C).
Means, standard deviations and frequencies were calculated, and paired t-tests were used to evaluate the continuous variables and chi-square test was used for categorical variables. Multiple regression analyses were made to evaluate the intraoperative effectiveness of additional procedures on compartment pressure. Linear correlation analysis was performed to evaluate the relationship between the initial and final pressure differences in both compartments. For radiographic measurement, intra-observer consistency between the two sets of measurements obtained by observer A and inter-observer consistency between the three sets of measurements obtained by observers A, B, and C, were analyzed using Pearson’s correlation coefficient and the intra-class correlation coefficient (ICC). ICC > 0.75 was regarded as excellent, ICC 0.40–0.75 was fair to good, and ICC < 0.40 was poor (20). A P value less than 0.05 was considered statistically significant and all statistical analyses were performed using SPSS 24.0 (SPSS, Chicago, IL). Finally, a power analysis was conducted to estimate the required number of the study. The power was set at 80% or higher with p < 0.05, and at least 64 knees were required for the study.
Absolute mediolateral compartment pressure difference determined initially (INI) and finally (FIN) were recorded and are shown in Fig. 3. Throughout the range of motion, the measured resection group showed higher initial pressure difference than the modified gap balancing group. Upon initial trial and final measurement after implantation, the proportion of the knees according to the compartment pressure (sum of medial and lateral compartment pressure) is shown in Fig. 4. After final implantation, 95% of the patients showed total compartment pressure less than 200 lbf.
Overall (T), measured resection TKAs (M) and modified gap balancing TKAs (G) initial (INI) and final (FIN) compartment pressure on medial and lateral side are shown in Fig. 5. In both medial lateral compartments, the pressure was generally decreased throughout the ROM in both TKA methods. In the medial compartment, the overall final compartment pressure (FIN) was significantly decreased compared with initial pressure (INI) throughout all range of motion (p < 0.05), and in the lateral compartment, the overall final compartment pressure was also decreased throughout the ROM especially during 30° flexion and full flexion (p < 0.05) (Table 1). According to TKA methods, both TKA techniques showed significant decrease in the medial compartment at the final measurement, although the lateral compartment did not show significant decrease in both technique (Table 2).
Linear correlation in the medial and lateral compartments was both statistically significant throughout the range of motion, with higher correlation in the lateral compartment. Medial compartment coefficients of determination were as follows: full extension (R2 = 0.12; P = 0.01), 30° flexion (R2 = 0.084; P = 0.007), 60° flexion (R2 = 0.078; P = 0.010), 90° flexion (R2 = 0.071; P = 0.014), and full flexion (R2 = 0.107; P = 0.002). Lateral compartment coefficients of determination were as follows: full extension (R2 = 0.306; P = 0.000), 30° flexion (R2 = 0.366; P = 0.000), 60° flexion (R2 = 0.358; P = 0.000), 90° flexion (R2 = 0.376; P = 0.000), full flexion (R2 = 0.422; P = 0.000) (Fig. 6).
Thirty patients (35.7%) showed “balanced” knee without a need for additional ligament balancing upon initial measurement and 54 patients showed “imbalance” knee who underwent additional procedures. Among 30 balanced knees, 6 were measured resection TKAs and 24 were modified gap balancing TKAs. After initial bone resection, modified gap balancing TKAs showed significantly higher proportion of “balanced” knees than measured resection TKAs (24 of 50 TKAs vs. 6 of 34 TKAs) (P = 0.004). Upon final pressure measurement after ligament balancing and component implantation, 79 patients (94.0%) showed a “balanced” joint with medial and lateral pressure difference less than 15 pounds throughout the full range of motion. Compared with the initial “balanced” knee (35.7%), there was significant improvement following ligament balancing at the final implantation (94.0%) (P < 0.05).
For 54 imbalanced knee, total 66 additional ligament balancing procedures (including soft tissue release (s) and bony resection) were carried out: 38 procedures for 28 measured resection TKAs and 28 procedures for 26 modified gap-balancing TKAs. Nine distal femur recuts (varus and neutral resection), 14 proximal tibia recuts (varus and neutral resection), 4 osteophyte removals, and 1 tibia posterior slope recut constituted additional bony procedures; and 26 medical collateral ligament (MCL) releases using pie-crusting (PC) method, 5 sub-periosteal superficial MCL releases, 5 iliotibial band (ITB) releases using PC method, 2 lateral collateral ligament (LCL) releases using PC method were performed for additional soft tissue releases.
The change of compartment pressure due to each additional procedure has been evaluated individually throughout the range of motion, and statistically significant procedures analyzed by regression analysis at each motion, are shown in Table 3. Results of regression analysis of the additional procedures were calculated based on the contributing beta values for each procedure by subtracting the post-procedure compartment pressure from pre-procedure compartment pressure. A positive value indicates improvement in joint pressure, and negative value indicates increased joint pressure while the absolute value indicates effectiveness of the joint pressure improvement. The effect of LCL release using PC method and tibia posterior slope resection procedures could not be analyzed due to lack of cases. However, the loading has been calculated to identify the effect of additional procedures on the compartment pressure throughout the range of motion. On average, bony resection showed relatively higher beta values compared with soft tissue releases. Distal femur recut showed a statistically significant effect on full extension and 90° flexion; proximal tibia recut showed significant effect on full extension; MCL release with PC technique showed effect throughout the range of motion but only significant in full extension; and release of suferficial MCL showed small but significant effect throughout the full range of motion.
On postoperative radiograph, average mechanical axis (MA) was 1.8° varus with 4.7° valgus femorotibial angle (FTA) and 6.2° posterior tibial slope (PTS). According to implant radiologic measurements, femoral and tibial components were also adequate positioned overall (Table 4). And the radiologic measurements showed excellent intra-observer consistency and inter-observer consistency across the 3 observers (Table 5).
The most important implication of the study is that, not only measured resection TKA but also modified gap balancing TKA using subjective “feeling” of ligament balancing, can be inaccurate or can show variable results despite abundant experience. Objective quantification using real-time orthosensor improved the soft tissue balance in both TKA techniques.
Until now, most of the studies about kinematics of knee joint have been based on biomechanical models or cadaveric studies [21,22,23]. Unfortunately, these studies have limitations in understanding the dynamics of the real knee joint. It is difficult to extrapolate the findings from cadaveric studies to living human body due to bias associated with factors such as postmortem contracture and tissue atrophy. Furthermore, in the biomechanical model studies, data application was a demanding procedure, because data are generally acquired from limited specimens.
Further, according to previous studies, although TKAs were carried out in conventional methods with soft tissue balancing intraoperatively, the potential imbalance remains due to subjective ligament balancing. It resulted in a variety of postoperative complications such as instability, stiffness, loosening, etc., which resulted in a significant proportion of TKA failures and revision surgeries [24, 25].
Therefore, we used the VERASENSE system to ensure adequate ligament balancing objectively via quantification of the joint intraoperatively, which would reduce the postoperative complications in long-term follow-up.
The initial (INI) and final (FIN) absolute pressure difference between medial and lateral compartments was higher in measured resection TKAs compared with gap balancing technique TKAs. As the femur resection was conducted according to the soft tissue balance in the modified gap balance technique, the quantification results were obtained as expected. But the difference was relatively higher than expected in spite of modified gap technique bone resection. Theoretically, modified gap balance bone resection should not show pressure difference between medial and lateral compartments because the bone resection was already based on patient’s ligament balance. In previous studies, the superiority of measured resection and gap balancing technique was still disputed. A few studies reported better outcomes with the gap balancing technique TKAs, whereas other reports showed no significant difference between the techniques in long-term clinical outcomes [3, 26, 27].
The overall initial (INI) and final (FIN) average compartment pressure showed a significant decrease throughout the range of motion in the medial compartment (p < 0.05). In the lateral compartment, the overall final loading values were significantly decreased under 30° flexion and full flexion (p < 0.05). Correlation analysis suggested that both medial and lateral compartments showed a significant relationship between initial (INI) and final (FIN) pressure throughout the range of motion. Our study indicated that variability of pressure between the initial (INI) and final (FIN) measurements was equal in both compartments, which facilitated surgical prediction of similar loading measurements with both VERASENSE system and final implanted components. Consequently, the surgeon could expect the same measured compartment pressure with the final implanted components.
Another important implication of the study was that only 30 patients (35.7%), constituting only 1/3 of the patients showed initially “balanced” knee after measured resection or gap balancing TKA. In spite of the abundant experience of the surgeon (EKS) in total knee arthroplasty for more than 30 years, subjective human “feeling” is variable and inaccurate. Fortunately, after additional ligament balancing procedures, a total of 79 patients (94.0%) completed the “balanced” TKA procedure, with load difference lower than 15 lbs. In a previous report of TKAs with a load difference lower than 15 lbs., better shorter-term clinical outcomes were observed compared with “unbalanced” knees with a difference greater than 15 lbs. , which underscored the significance of the objective quantification orthosensor, for adequate and satisfactory results. As mentioned before, the subjective and inaccurate procedures associated with traditional ligament balancing  have been elucidated through the study.
Excluding the 30 patients who were initially “balanced”, 54 patients underwent additional ligament balancing procedure. A total of 66 additional procedures were performed, which accounted for an average of 1.2 procedures per patient. According to our study, generally measured resection TKA patients underwent more additional ligament balancing procedures than modified gap balance TKA patients, 1.36 per person and 1.1 per person, respectively. Bony resection showed higher changes in joint loading, especially distal femur recut was effective in full extension to mid-flexion, and proximal tibia recut was effective during the mid-flexion. These results were similar to previous qualitative studies. For example, Mihalko et al.  demonstrated that distal femur resection was effective for mid-flexion contractures. Ahn et al.  demonstrated the effectiveness of proximal tibia varus resection on severe varus deformities. Among soft tissue release (s), MCL release with PC technique was most frequently used and was also most effective through flexion, which also was similar to many previous studies using MCL release as an important procedure during total knee arthroplasty in varus deformity [31,32,33]. Although superficial MCL has been known to have a lesser effect in flexion stability , our study showed that sub-periosteal superficial MCL release showed small but significant effect on joint loading throughout the range of motion. Postoperative radiographic alignment of patients including mechanical axis, femorotibial angle, posterior slope, and implant positions were within satisfactory range.
However, this study also has a few limitations. First, the group only consisted of 84 patients, and a larger group may have yielded more generalized analyses of the knee joint. Fortunately, our study findings were similar to those reported previously in published studies. Second, we used only a single implant system in all patients. Other implant systems showed different loading values due to differences in implant design. Also, the PCL release being a strong influence on flexion ligament balancing, using two types of implant (CR and PS type) may have produced some bias during interpretation of the data. Third, although the pressure was quantified and presented as digited results on the screen, the measurement was done by a single surgeon, suggesting possible bias. Fourth, we did not compare the long-term clinical and radiological outcomes between objectively balanced TKAs and traditionally subjectively balanced TKAs.
By objective quantification using orthosensor, we observed significant decrease in both medial and lateral compartments pressure after TKA, and could obtain 94% balanced knee, consequently. And between the techniques, measured resection TKA showed less balanced knee in the initial pressure measurement and also required more additional procedures compared to modified gap balancing TKA. But also, we suggest that regardless of TKA surgical methods, additional procedures could be needed for adequate “patient-specific” ligament balancing. Furthermore, with the consistent data of the orthosensor acquired during appropriate ligament balancing, the surgeon could eventually reduce the complications associated with soft tissue imbalance in the future.
Intra-class correlation coefficient
Lateral collateral ligament
Medical collateral ligament
Posterior tibial slope
Total knee arthroplasty
Lustig S, Bruderer J, Servien E, Neyret P. The bone cuts and ligament balance in total knee arthroplasty: the third way using computer assisted surgery. Knee. 2009;16(2):91.
Hananouchi T, Yamamoto K, AndoW FK, Ohzono K. The intraoperative gap difference (flexion gap minus extension gap) is altered by insertion of the trial femoral component. Knee. 2012;19:601–5.
Matsumoto T, Muratsu H, Kawakami Y, Takayama K, Ishida K, Matsushita T, et al. Soft-tissue balancing in total knee arthroplasty: cruciate-retaining versus posterior-stabilised, and measured-resection versus gap technique. Int Orthop. 2014;38(3):531–7.
Siddiqi A, Hardaker WM, Eachempati KK, Sheth NP. Advances in computer-aided technology for total knee arthroplasty. Orthopedics. 2017;40(6):338–52.
Liow MH, Xia Z, Wong MK, Tay KJ, Yeo SJ, Chin PL. Robot-assisted total knee arthroplasty accurately restores the joint line and mechanical axis. A prospective randomised study. J Arthroplast. 2014;29(12):2373–7.
Nagai K, Muratsu H, Takeoka Y, Tsubosaka M, Kuroda R, Matsumoto T. The influence of joint distraction force on the soft-tissue balance using modified gap-balancing technique in posterior-stabilized total knee arthroplasty. J Arthroplast. 2017;32(10):2995–9.
Lee SY, Lim HC, Jang KM, Bae JH. What factors are associated with femoral component internal rotation in TKA using the gap balancing technique? Clin Orthop Relat Res. 2017;475(8):1999–2010.
Ferreira MC, Franciozi CES, Kubota MS, Priore RD, Ingham SJM, Abdalla RJ. Is the use of spreaders an accurate method for ligament balancing? J Arthroplast. 2017;32(7):2262–7.
Kim SH, Lim JW, Jung HJ, Lee HJ. Influence of soft tissue balancing and distal femoral resection on flexion contracture in navigated total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2017;25(11):3501–7.
Heesterbeek PJC, Haffner N, Wymenga AB, Stifter J, Ritschl P. Patient-related factors influence stiffness of the soft tissue complex during intraoperative gap balancing in cruciate-retaining total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2017;25(9):2760–8.
Wyss TF, Schuster AJ, Münger P, Pfluger D, Wehrli U. Does total knee joint replacement with the soft tissue balancing surgical technique maintain the natural joint line? Arch Orthop Trauma Surg. 2006;126(7):480–6.
Ranawat CS, Flynn WF Jr, Saddler S, Hansraj KK, Maynard MJ. Long-term results of the total condylar knee arthroplasty. A 15-year survivorship study. Clin Orthop Relat Res. 1993;286:94–102.
Ranawat CS, Flynn WF Jr, Saddler S, Hansraj KK, Maynard MJ. Long-term results followup of anatomic graduated components posterior cruciate-retaining total knee replacement. Clin Orthop Relat Res. 1993;388:51–7.
Fehring TK, Odum S, Griffin WL, Mason JB, Nadaud M. Early failures in total knee arthroplasty. Clin Orthop Relat Res. 2001;392:315–8.
Mulhall KJ, Ghomrawi HM, Scully S, Callaghan JJ, Saleh KJ. Current etiologies and modes of failure in total knee arthroplasty revision. Clin Orthop Relat Res. 2006;446:45–50.
Mihalko WM, Saleh KJ, Krackow KA, Whiteside LA. Soft-tissue balancing during total knee arthroplasty in the varus knee. J Am Acad Orthop Surg. 2009;17(12):766–74.
Yim JH, Song EK, Khan MS, Sun ZH, Seon JK. A comparison of classical and anatomical total knee alignment methods in robotic total knee arthroplasty: classical and anatomical knee alignment methods in TKA. J Arthroplast. 2013;28(6):932–7.
Oh CS, Song EK, Seon JK, Ahn YS. The effect of flexion balance on functional outcomes in cruciate-retaining total knee arthroplasty. Arch Orthop Trauma Surg. 2015;135(3):401–6.
Gustake KA, Golladay GJ, Roche MW, et al. A new method for defining balance: promising short-term clinical outcomes of sensor-guided TKA. J Arthroplast. 2014;29:955–60.
Bellemans J, Vandenneucker H, Van Lauwe J, et al. New surgical technique for medial collateral ligament balancing. J Arthroplast. 2010;25(7):1151–6.
Onishi Y, Hino K, Watanabe S, Watamori K, Kutsuna T, Miura H. The influence of tibial resection on the PCL in PCL-retaining total knee arthroplasty: a clinical and cadaveric study. J Orthop Sci. 2016;21(6):798–803.
Wada K, Hamada D, Tamaki S, Higashino K, Fukui Y, Sairyo K. Influence of medial collateral ligament release for internal rotation of tibia in posterior-stabilized total knee arthroplasty: a cadaveric study. J Arthroplast. 2017;32(1):270–3.
Shoifi Abubakar M, Nakamura S, Kuriyama S, Ito H, Ishikawa M, Furu M, et al. Influence of posterior cruciate ligament tension on knee kinematics and kinetics. J Knee Surg. 2016;29(8):684–9.
Postler A, Lützner C, Beyer F, Tille E, Lützner J. Analysis of total knee arthroplasty revision causes. BMC Musculoskelet Disord. 2018;19(1):55–60.
Naudie DD, Ammeen DJ, Engh GA, Rorabeck CH. Wear and osteolysis around total knee arthroplasty. J Am Acad Orthop Surg. 2007;15(1):53–64.
Churchill JL, Khlopas A, Sultan AA, Harwin SF, Mont MA. Gap-balancing versus measured resection technique in total knee arthroplasty: a comparison study. J Knee Surg. 2018;31(1):13–6.
Teeter MG, Perry KI, Yuan X, Howard JL, Lanting BA. Contact kinematic differences between gap balanced vs measured resection techniques for single radius posterior-stabilized total knee arthroplasty. J Arthroplasty. 2017;32(6):1834–8.
Elmallah RK, Mistry JB, Cherian JJ, Chughtai M, Bhave A, Roche MW, et al. Can we really “feel” a balanced total knee arthroplasty? J Arthroplast. 2016;31(9 Suppl):102–5.
Mihalko WM, Whiteside LA. Bone resection and ligament treatment for flexion contracture in knee arthroplasty. Clin Orthop Relat Res. 2003;406(1):141–7.
Ahn JH, Back YW. Comparative study of two techniques for ligament balancing in total knee arthroplasty for severe varus knee: medial soft tissue release vs. bony resection of proximal medial tibia. Knee Surg Relat Res. 2013;25(1):13–8.
Ha CW, Park YB, Lee CH, Awe SI, Park YG. Selective medial release technique using the pie-crusting method for medial tightness during primary total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2015;23(6):1816–23.
Koh IJ, Kwak DS, Kim TK, Park IJ, In Y. How effective is multiple needle puncturing for medial soft tissue balancing during total knee arthroplasty? A cadaveric study. J Arthroplast. 2014;29(12):2478–83.
Bellemans J. Multiple needle puncturing: balancing the varus knee. Orthopedics. 2011;34(9):e510–2.
Whiteside LA. Soft tissue balancing: the knee. J Arthroplast. 2002;17(4 Suppl 1):23–7.
Availability of data and materials
The datasets used and/or analysed during the current study available from the corresponding author on reasonable request.
Consent to publish
Ethics approval and consent to participate
The surgical procedure was approved by our institutional review board of Chonnam National University Hwasun Hospital, and written informed consent was obtained from all patients.
On behalf of all authors, the corresponding author states that there is no competing of interest.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Cho, KJ., Seon, JK., Jang, WY. et al. Objective quantification of ligament balancing using VERASENSE in measured resection and modified gap balance total knee arthroplasty. BMC Musculoskelet Disord 19, 266 (2018). https://doi.org/10.1186/s12891-018-2190-8
- Total knee arthroplasty
- Ligament balancing