- Research article
- Open Access
- Open Peer Review
Initial experience with 3D isotropic high-resolution 3 T MR arthrography of the wrist
BMC Musculoskeletal Disorders volume 17, Article number: 30 (2016)
Our study was performed to evaluate the image quality of 3 T MR wrist arthrograms with attention to ulnar wrist structures, comparing image quality of isotropic 3D proton density fat suppressed turbo spin echo (PDFS TSE) sequence versus standard 2D 3 T sequences as well as comparison with 1.5 T MR arthrograms.
Eleven consecutive 3 T MR wrist arthrograms were performed and the following sequences evaluated: 3D isotropic PDFS, repetition time/echo time (TR/TE) 1400/28.3 ms, voxel size 0.35x0.35x0.35 mm, acquisition time 5 min; 2D coronal sequences with slice thickness 2 mm: T1 fat suppressed turbo spin echo (T1FS TSE) (TR/TE 600/20 ms); proton density (PD) TSE (TR/TE 3499/27 ms). A 1.5 T group of 18 studies with standard sequences were evaluated for comparison. All MR imaging followed fluoroscopically guided intra-articular injection of dilute gadolinium contrast. Qualitative assessment related to delineation of anatomic structures between 1.5 T and 3 T MR arthrograms was carried out using Mann–Whitney test and the differences in delineation of anatomic structures among each sequence in 3 T group were analyzed with Wilcoxon signed-rank test. Quantitative assessment of mean relative signal intensity (SI) and relative contrast measurements was performed using Wilcoxon signed-rank test.
Mean qualitative scores for 3 T sequences were significantly higher than 1.5 T (p < 0.01), with isotropic 3D PDFS sequence having highest mean qualitative scores (p < 0.05). Quantitative analysis demonstrated no significant difference in relative signal intensity among the 3 T sequences. Significant differences were found in relative contrast between fluid-bone and fluid-fat comparing 3D and 2D PDFS (p < 0.01).
3D isotropic PDFS sequence showed promise in both qualitative and quantitative assessment, suggesting this may be useful for MR wrist arthrograms at 3 T. Primary reasons for diagnostic potential include the ability to make reformations in any obliquity to follow the components of ulnar side wrist structures including triangular fibrocartilage complex. Additionally, isotropic imaging provides thinner slice thickness with less partial volume averaging allowing for identification of subtle injuries.
The wrist, given its relative small size compared to other joints and complex arrangement of ligaments, requires high-resolution, high signal-to-noise, and small field of view MR imaging for accurate assessment of anatomy and pathology. When attention is required at the ulnar side of the wrist for evaluation of triangular fibrocartilage complex (TFCC) or capsular injury, direct MR arthrography has become an increasingly utilized modality, allowing for improved visualization of the closely grouped ligamentous structures following distention of the joint with injected contrast [1–3]. In addition to joint distension, intra-articular contrast allows for visualization of contrast leakage/extravasation and has been shown to be particularly useful in evaluating partial ligamentous tears including of the TFCC . With increasingly widespread use of 3 T MR in musculoskeletal imaging, the wrist included, 3 T MR has been shown to be sensitive and specific for wrist ligament tears in addition to improved visualization of normal anatomy [5, 6]. The addition of 3D isotropic imaging has also shown utility in musculoskeletal imaging, including of the knee [7, 8]. More recent literature involving 3 T 3D isotropic imaging of the wrist in healthy volunteers has been carried out, highlighting the advantages of multi-planar reconstructions and additional post-processing capabilities afforded by isotropic imaging which prove useful in evaluating the intricate anatomy of the wrist .
Our study was performed to evaluate image quality of 3 T MR arthrograms which include 3D isotropic proton density weighted fat suppressed sequence (PDFS). Comparison was carried out with standard 1.5 T MR arthrogram studies and additional quantitative analysis of relative signal intensity and relative contrast of the individual 3 T sequences was also performed.
Retrospective review of MR images was carried out following University of California, Irvine institutional review board approval. Given our study was a retrospective review, informed consent was waived by the IRB. For the 3 T group, there were 11 total patients (ages 17 to 58 years, average age 33 years; four females and seven males; nine right and two left wrist) collected from studies performed from 11/2012 to 2/2015. Ten of the 11 included 3D isotropic PDFS sequence. 3 T sequence imaging parameters can be found in Table 1. All 3 T images were acquired on a Phillips 3 T system using 8 channel wrist coil (Achieva, Philips HealthcareVR, Best, The Netherlands). A parallel imaging technique called sensitivity encoding (SENSE) was used both in the 2D and 3D sequences at 3 T. All 3D images were obtained with the driven equilibrium (DRIVE) technique. The 3D isotropic PD FSE sequence uses flip angle sweep, short and nonvolume selective refocusing pulse, and SENSE, making it possible to get shorter echo spacing and a better signal-to-noise ratio .
For the 1.5 T group there were 18 total patients (ages 15 to 58 years, average age 31 years, six females and 12 males, eight right and eight left wrists) from review of MR arthrograms from 2006 through 2013. 1.5 T imaging parameters can be found in Table 2. All images in this group were acquired on Siemens 1.5 T system (Avanto, Siemens AG, Berlin, Gemany).
All studies of both 1.5 T and 3 T groups were obtained following injection of 3–4 ml of dilute gadolinium contrast mixture (15 mL sterile saline, 5 mL iodinated contrast, and 0.1 mL of gadolinium contrast) into the radiocarpal joint under fluoroscopy using standard department protocol. The patients were imaged within 30 min of injection.
A total of 11 wrists from the 3 T group and 18 from the 1.5 T group were independently evaluated on AGFA Impax PACS workstation (AGFA Morsel, Belgium) by two experienced musculoskeletal radiologists (HY (R1), 26 years of experience and TN (R2), 13 years of experience) without knowing clinical data. For evaluation of the 3 T group, each of the 3 imaging sequences (T1FS, PDFS, 3D isotropic PDFS) were graded separately on the delineation of anatomic structures and quality for detection of pathology including presence of contrast leak, presence/absence of abnormality in the prestyloid recess, ability to visualize triangular ligament styloid attachment and presence/absence of any associated injury, status of remaining TFCC, and ability to visualize ulnar collateral ligament attachment and presence/absence of any related injury. For the above pathology except contrast leak, each 3 T sequence (T1FS, PDFS, 3D isotropic PDFS) image quality was graded from 0 to 4; 0 for no visualization, 1 for poor visualization, 2 satisfactory, 3 good, and 4 excellent. Contrast leak was classified as major or minor, with major leak being a contrast leakage outside the joint capsule with signal intensity approximately equal to signal intensity of the injected contrast within the radiocarpal joint and minor leak being less signal intensity than injected contrast. Confidence level of leak detection was graded from 0 to 3 with 0 for no leak, 1 possible, 2 probable, and 3 definite leak. For UCL attachment and injury, a fourth sequence was also evaluated (PD without fat suppression) because ligamentous attachment is generally better identified without fat suppression. One study was a post operative study from prior repair of TFCC injury and given post surgical changes, the anatomic evaluation was unable to be accurately performed, and only overall image quality of the sequences was evaluated. For the 18 wrist studies in the 1.5 T group, the same anatomic and pathologic structures were evaluated in a similar manner but instead of assigning a score from 0 to 4 for each individual sequence, an overall score for the study was assigned using the best image quality from the 3 sequences evaluated (T1, T1FS, and Multi-Echo Data Image Combination (MEDIC)).
Quantitative analysis was performed for the 11 3 T studies. For each of the 3 T wrist sequences (2D T1FS, 2D PDFS, isotropic 3D PDFS), ROIs for each patient were drawn and relative signal intensity (SI) and relative contrast were measured using. ROI’s were identical in size and placed in identical positions on matching sections. Fluid was measured from the space adjacent to TFCC, prestyloid recess, or intercarpal space (Fig. 1). The mean and standard deviation (SD) of SI were determined for the disc of the TFCC, fluid, cartilage, and bone marrow. Then, relative SI and relative contrast were used for direct comparison of image quality between the 2D FSE and 3D isotropic FSE MR images both with and without fat suppression, because all sequences in this study were obtained with a parallel imaging technique. Relative SI of each structure was calculated as SI/SD, and relative contrast of structure A (a) to structure B (b) was calculated as (SIa-SIb) / (SDa 2 + SDb 2)1/2 [9, 11].
In the qualitative assessments, differences in the delineation of anatomic structures between 1.5 T and 3 T MR arthrography were statistically analyzed using the Mann–Whitney test, and differences in the delineation of anatomic structures among each sequence (2D T1FS, 2D PDFS, isotropic 3D PDFS and 2D PD) analyzed using the Wilcoxon signed-rank test. Inter-rater reliability was measured with percentage of exact agreement and agreement within a range of +/−1 score point . Cohen’s kappa was also calculated to assess inter-rater reliability for the presence or absence of each anatomical structure’s injury. The following ratings for the interpretation of kappa were used; poor (<0.40), moderate (0.40–0.59), good (0.60–0.80), and excellent (>0.80). In the quantitative assessments, comparison of mean relative SI and relative contrast between 3 T 2D FSE and 3 T 3D isotropic FSE MR images were performed using the Wilcoxon signed-rank test. P values of less than 0.05 were considered to indicate a significant difference. All statistical analyses were performed using R version 3.0.2 for windows software (R Development Core Team, Vienna, Austria).
Mean values of qualitative analysis calculated from individual scores are presented in Table 3. Each 3 T sequence mean value was statistically significantly higher than that for overall image quality of 1.5 T studies (see Table 3). Within the 3 T group, the 3D sequence had the highest average scores for the five separate categories (contrast leak, prestyloid recess abnormality, styloid attachment of TFC, remaining TFC, and UCL), and the 3D isotropic sequence was statistically significantly superior in the image quality of the triangular ligament compared with the T1FS (p = 0.047) and PDFS (p = 0.026). Figure 2 shows examples 3 T coronal MR arthrogram images as well as oblique coronal MPR image created from 3D isotropic PDFS sequence. Oblique coronal MPR clearly demonstrates ulnar attachment of the triangular ligament of the TFCC. In Fig. 3, only oblique coronal MPR is able to demonstrate a focal partial tear of the distal lamina of the triangular ligament. At the level of the disc of the TFCC, the ulnar styloid is not included in the coronal plane (see the top row in Fig. 3) and at the level of ulnar styloid, the TFCC is not shown in the coronal plane (see the middle row in Fig. 3). This happened very frequently in a true coronal plane because the transverse axis of the radius, which is parallel to the coronal plane, is not the identical to the transverse axis of the ulnar head (for example oblique axial image in Figs. 2 and 3).
Contrast leak was identified in 10 of 11 3 T studies; 4 classified as major and 6 as minor leak with higher confidence than 1.5 T, (Table 4). For 1.5 T studies, 12 of 18 showed contrast leak with 4 major and 8 minor.
The results of inter-rater (R1-R2) agreement for each evaluation are shown in Table 5. All sequences at 3 T and 1.5 T demonstrated high inter-rater agreement within one point (90.9–100 at 3 T and 94.4–100 % at 1.5 T). Inter-rater agreement of the presence or absence of each anatomical structure’s injury was excellent (k = 0.83–1.00) on 3.0 T-MR arthrography, and good or excellent (k = 0.71–0.89) on 1.5 T-MR arthrography. 3 T-MR arthrography was superior compared with 1.5 T-MR arthrography on the evaluation of each anatomical structure’s injury.
Mean values for quantitative measurements of relative signal intensity of fluid, bone marrow, TFCC, and fat as well as relative contrast of fluid to bone, fluid to TFCC, and fluid to fat are shown in Table 6. There were no significant differences in relative SI between the three evaluated 3 T sequences, however the 3D isotropic PDFS sequence showed significantly higher relative contrast of fluid to bone and fluid to fat compared to the 2D PDFS sequence (p < 0.01). The 2D T1FS sequence also showed significantly higher relative contrast of fluid to fat compared to 2D PDFS (p < 0.05).
Evaluation of the ulnar side of the wrist remains a challenge in everyday medical practice, particularly due to the complex nature of triangular fibrocartilage complex. This is complicated by discussion involving anatomic nature of ulnar wrist structures including ulnomeniscal homologue [13–17] and ulnar collateral ligament complex [18–21], both of which have been debated in the literature. Despite the nuances in the anatomy, direct MR arthrography with injected intra-articular gadolinium contrast has become the chosen modality for evaluation of the ulnar side of the wrist, with several prior studies demonstrating improved diagnostic ability compared routine 2D MRI [22–27] and previously supplanting conventional arthrography due to conventional arthrography’s low specificity and accuracy .
Given the shape and complex geometry of the TFCC it had been proposed as a candidate for 3D imaging . Additional studies using 3D images of the wrist have been done including a study in healthy volunteers , and a study using 3D T1 sequence examining scapholunate, lunotriquietral, and TFCC tears . These studies also showed improved diagnostic potential with 3D imaging. A consensus among these studies as a factor in improved diagnostic potential using 3D isotropic sequence include ability to construct multi-planar reformatted images along an arbitrary obliquity to follow the TFCC and additional ulnar sided structures (or any ligament). Another reason for improved diagnostic potential is the smaller slice thickness of 3D isotropic sequence (0.35 mm versus 2–3 mm) with subsequently decreased partial volume averaging. With smaller slices there is potential for improved identification of subtle abnormalities of the TFCC and capsular complex, which may not bee seen on 2D sequences. Image blur was not a major factor in image quality of the 3 T images, although it has been implicated in the past as a disadvantage to 3 T imaging using long echo train length FSE sequences . Improvement in the imaging of the TFCC is important for many reasons including defining MR anatomy, patient selection for surgical versus non surgical treatment, pre-operative planning, and ultimately leading to improved patient outcomes.
In our study we evaluated 3 T MR wrist arthrogram studies with attention to ulnar wrist structures including TFCC and ulnar capsular complex and compared the image quality of the 3 T sequences to one another and to that of 1.5 T MR wrist arthrogram, along with quantitative analysis amongst the 3 T images using relative SI and relative contrast. The results showed 3 T image quality to be significantly superior to that of 1.5 T and that the 3D isotropic PDFS sequence had the highest scores for image quality among the 3 T sequences. Inter-rater reliability was high for 3 T 3D sequence. In addition, as previous studies, the capability of multi-planar reformations in any obliquity planes proved usefulness for 3D isotropic MR arthrograms at 3 T for improved visualization of TFCC and capsular injuries, especially at the ulnar styloid attachment.
As part of the qualitative grading we looked at the presence of contrast leakage outside the joint capsule at the ulnar wrist, and graded it on confidence of leak and whether it was major or minor, because contrast leakage is frequently seen and is a specific finding with MR arthrogram. The clinical significance of leak at the ulnar side of the wrist, and furthermore whether major or minor plays a role, remains to be seen and is a source of further investigation. Prior studies characterizing leakage on conventional arthrography identified leakage at the ulnar capsular complex as pathologic, supported by improvement in symptoms following repair . However, it has also been found that TFC “communicating defects” on conventional arthrography happen frequently in asymptomatic patients and have also been shown in a significant percentage of cadaveric studies and may in fact represent changes related to degeneration [33–36]. This discussion of contrast extravasation at the ulnar side of the wrist is similar to that found with MR arthrogram of the shoulder, where capsular leakage is also not uncommon, having both pathologic causes including a number of labral tears and conditions such as adhesive capsulitis, but also identified through the subscapular recess along subscapularis and adjacent to axillary recess, both unrelated to needle path and without adjacent related pathologic findings . Regardless, future studies with focus on clinical outcomes will be needed to clarify the significance of contrast material outside the capsule at the ulnar side of the wrist on MR arthrography.
Limitations of our study include the relatively small size, with 11 3 T studies and 10 of which included 3D isotropic sequence. Additionally, the 1.5 T studies were collected over a number of years and the imaging protocols had changed, leaving comparison with similar but not identical imaging parameters for both 3 T and 1.5 T studies. Another limitation is related to the relative longer acquisition time of 3D sequence, which adds approximately 5–6 min to total scan time. The ability to quickly and seamlessly create MPR images is also a limitation, usually requiring additional functional PACS capabilities or additional software, which may not be widely available. The fact that this 3D sequence is PD weighted rather than T1, differentiation between subtle edema and minor contrast leakage occasionally can be difficult. This may suggest 3D isotropic PDFS could potentially replace 2D PDFS, but not 2D T1FS. Finally, selection bias also plays a role given that the majority of patient’s had chronic ulnar wrist pain and had been evaluated by a hand surgeon prior to imaging and initially referred for MR arthrography.
The results of our initial study investigating the image quality of 3 T MR arthrogram with 3D isotropic PDFS sequence adds to some existing evidence that 3D isotropic PDFS sequences may be useful when imaging the wrist at 3 T. Given the inherent advantages of direct MR arthrography combined with isotropic high-resolution images, this technique shows promise for the evaluation of ulnar wrist pathology including TFCC and capsular injuries.
Maizlin ZV, Brown JA, Clement JJ, Grebenyuk J, Fenton DM, Smith DE, et al. MR arthrography of the wrist: controversy and concepts. Hand. 2009;4:66–73.
Elentuck D, Palmer WE. Direct magnetic resonance arthrography. Eur Radiol. 2004;14:1956–67.
Steinbach LS, Palmer WWE, Schweitzer ME. Special focus session. MR arthrography. Radiographics. 2002;22:1223–46.
Meier R, Schmitt R, Krimmer H. Wrist lesions in MRI arthrography compared with wrist arthroscopy. Handchir Mikrochir Plast Chir. 2005;37:85–9.
Magee T. Comparison of 3-T MRI and arthroscopy of intrinsic wrist ligament and TFCC tears. AJR Am J Roentgenol. 2009;192:80–5.
Saupe N, Prussmann KP, Luechinger R, Bosiger P, Marincek BWeishaupt D. MR imaging of the wrist: comparison between 1.5- and 3-T MR imaging–preliminary experience. Radiology. 2005;234:256–64.
Gold GE, Busse RF, Beehler C, Han E, Brau ACS, Beatty PJ, et al. Isotropic MRI of the knee with 3D fast spin-echo extended echo-train acquisition (XETA): initial experience. AJR Am J Roentgenol. 2007;188:1287–93.
Kijowski R, Davis KW, Woods MA, Lindstrom MJ, DeSmet AA, Gold GE, et al. Knee joint: comprehensive assessment with 3D isotropic resolution fast spin-echo imaging–diagnostic performance compared with that of conventional MR imaging at 3.0 T. Radiology. 2009;252:486–95.
Yamabe E, Anavim A, Sakai T, Miyagi R, Nakamura T, Hitt D, et al. Comparison between high-resolution isotropic three-dimensional and high-resolution conventional two-dimensional FSE MR images of the wrist at 3 tesla: a pilot study. J Magn Reson Imaging. 2013;40:603–8.
Choo HJ, Lee SJ, Kim OH, Seo SS, Kim JH. Comparison of three- dimensional isotropic T1-weighted fast spin-echo MR arthrography with two-dimensional MR arthrography of the shoulder. Radiology. 2012;262:921–31.
Zhang J, Israel GM, Hecht EM, Krinsky GA, Babb JS, Lee VS. Isotropic 3D T2-weighted MR cholangiopancreatography with parallel imaging: feasibility study. AJR Am J Roentgenol. 2006;187:1564–70.
Fischer M, Ruegg S, Czaplinski A, Strohmeier M, Lehmann A, Tschan F, et al. Inter-rater reliability of the full outline of unresponsiveness score and the Glasgow Coma Scale in critically ill patients: a prospective observational study. Crit Care. 2010;14:R64.
Buck FM, Gheno R, Nico MA, Haghighi P, Trudell DJ, Resnick D. Ulnomeniscal homologue of the wrist: correlation of anatomic and MR imaging findings. Radiology. 2009;253:771–9.
Taleisnik J. The ligaments of the wrist. J Hand Surg [Am]. 1976;1:110–8.
Bowers WH. The distal radioulnar joint. In: Green DP, editor. Operative hand surgery, vol. 1. 3rd ed. New York: Churchill Living- stone; 1993. p. 973–1019.
Ishii S, Palmer AK, Werner FW, Short WH, Fortino MD. An anatomic study of the ligamentous structure of the triangular fibrocartilage complex. J Hand Surg [Am]. 1998;23:977–85.
Nishikawa S, Toh S. Anatomical study of the carpal attachment of the triangular fibrocartilage complex. J Bone Joint Surg (Br). 2002;84:1062–5.
Spinner M, Kaplan EB. Extensor carpi ulnaris: its relationship to the stability of the distal radio-ulnar joint. Clin Orthop. 1970;68:124–9.
Kleinman WB. Scapho-trapezio-trapezoid joint arthrodesis. In: Blair WF, editor. Techniques in hand surgery. Baltimore: Williams & Wilkins; 1996. p. 835–48.
Buterbaugh GA. Radiocarpal arthroscopy portals and normal anatomy. Hand Clin. 1994;10:567–76.
Osterman AL. Atlas of right wrist arthroscopy. Hand Clin. 1995;11:103–13.
Smith DK. MR imaging of normal and injured wrist ligaments. Magn Reson Imaging Clin N Am. 1995;3(2):229–48.
Schweitzer ME, Brahme SK, Hodler J, Hanker GJ, Lynch TP, Flannigan BD, et al. Chronic wrist pain: spin-echo and short tau inversion recovery MR imaging and conventional and MR arthrography. Radiology. 1992;182(1):205–11.
Zanetti M, Bram J, Hodler J. Triangular fibrocartilage and intercarpal ligaments of the wrist: does MR arthrography improve standard MRI? J Magn Reson Imaging. 1997;7(3):590–4.
Scheck RJ, Kubitzek C, Hierner R, Szeimies U, Pfluger T, Wilhelm K, et al. The scapholunate interosseous ligament in MR arthrography of the wrist: correlation with non-enhanced MRI and wrist arthroscopy. Skeletal Radiol. 1997;26(5):263–71.
Scheck RJ, Romagnolo A, Hierner R, Pfluger T, Wilhelm K, Hahn K. The carpal ligaments in MR arthrography of the wrist: correlation with standard MRI and wrist arthroscopy. J Magn Reson Imaging. 1999;9(3):468–74.
Moser T, Dosch JC, Moussaoui A, Dietemann JL. Wrist ligament tears: evaluation of MRI and combined MDCT and MR arthrography. AJR Am J Roentgenol. 2007;188(5):1278–86.
Weiss AP, Akelman E, Lambiase R. Comparison of the findings of triple-injection cinearthrography of the wrist with those of arthroscopy. J Bone Joint Surg Am. 1996;78:348–56.
Yoshioka H, Burns JE. Magnetic resonance imaging of triangular fibrocartilage. J Magn Reson Imaging. 2012;35:764–78.
Lee HY, Choi YR, Kim S, Song TH, Suh JS. Intrinsic ligament and triangular fibrocartilage complex (TFCC) tears of the wrist: comparison of isovolumetric 3D-THRIVE sequence MR arthrography and conventional MR images at 3 T. Magn Reson Imaging. 2013;31:221–6.
Constable RT, Gore JC. The loss of small objects in variable TE imaging: implications for FSE, RARE, and EPI. Magn Reson Med. 1992;28:9–24.
Levinsohn EM, Rosen ID, Palmer AK. Wrist arthrography: value of the three-compartment injection method. Radiology. 1991;179:231–9.
Manaster BJ, Mann RJ, Rubenstein S. Wrist pain: correlation of clinical and plain film findings with arthrographic results. J Hand Surg [Am]. 1989;14:466–73.
Kirschenbaum D, Sieler S, Solonick D, Loeb DM, Cody RP. Arthrography of the wrist: assessment of the integrity of the ligaments in young asymptomatic adults. J Bone Joint Surg Am. 1995;77:1207–9.
Yin YM, Evanoff B, Gilula LA, Pilgram TK. Evaluation of selective wrist arthrography of contralateral asymptomatic wrists for symmetric ligamentous defects. Am J Roentgenol. 1996;166:1067–73.
Metz VM, Mann FA, Gilula LA. Lack of correlation between site of wrist pain and location of noncommunicating defects shown by three-compartment wrist arthrography. Am J Roentgenol. 1993;160:1239–43.
Ogul H, Kantarci M, Topal M, Karaca L, Tuncer K, Pirimoglu B, et al. Extra-articular contrast material leaks into locations unrelated to injection path in shoulder MR arthrography. Eur Radiol. 2014;24:2006–613.
Our study was partially supported by JA Kyosai Research Institute Fund.
The authors declare they have no competing interests
JS contributed to data acquisition including image evaluation, compiled studies for evaluation, and drafted the manuscript. TN contributed to data acquisition including image evaluation and statistical analysis and drafted portions of the manuscript. YK contributed to data acquisition including statistical analysis and drafted portions of manuscript. HJY contributed to image acquisition and provided key elements for MRI sequence optimization. GR provided clinical support for patient selection and clinical insight for manuscript draft. DH provided technical guidance for MRI image optimization. HY provided project concept and oversight, contributed to image evaluation and data acquisition and contributed to manuscript drafting and revisions. All authors approved the final manuscript.
About this article
Cite this article
Sutherland, J.K., Nozaki, T., Kaneko, Y. et al. Initial experience with 3D isotropic high-resolution 3 T MR arthrography of the wrist. BMC Musculoskelet Disord 17, 30 (2016). https://doi.org/10.1186/s12891-016-0890-5
- Musculoskeletal MRI
- 3D MR imaging
- MR arthrography
- Wrist MRI