This article has Open Peer Review reports available.
Novel MR imaging method – MAVRIC – for metal artifact suppression after joint replacement in musculoskeletal tumor patients
© Susa et al. 2015
Received: 25 September 2015
Accepted: 1 December 2015
Published: 4 December 2015
Standard imaging modality for the follow-up after prosthetic replacements for musculoskeletal tumor patients has been conventional radiography. This technique is effective in detecting subtle changes in bone adjacent to metal implants, but in many cases, radiographs do not lead to definitive diagnosis of postoperative adverse events such as acute infection, local recurrence of soft tissue tumor or soft tissue local recurrence of osseous sarcoma. Conventional MRI sequences have not been effective due to metal artifacts. In this study, we tried to elucidate the effectiveness of metal artifact suppression using novel sequence, multiacquisition variable-resonance image combination (MAVRIC), after musculoskeletal tumor surgeries.
We retrospectively analyzed 5 cases of malignant bone and soft tissue sarcoma patients who were reconstructed with metal prosthesis after wide resection of tumors. Images obtained using MAVRIC and short tau inversion recovery (STIR) were compared side by side. The paired MAVRIC and STIR images were qualitatively compared independently by two specialists for 4 parameters: visualization of bone - implant interface, visualization of surrounding soft tissues, image blurring, and overall image quality. Quantitatively, paired images were reviewed to identify the slice where the metal artifact was maximal, and a region of interest encompassing the implant and surrounding artifact was drawn using Advantage Workstation (GE Healthcare, Japan).
There were no local recurrences that were detected. By utilizing MAVRIC, visualization of the bone - implant interface and visualization of the surrounding soft tissue were significantly improved in MAVRIC compared to STIR. Although blurring was worse on the MAVRIC acquisitions, the overall image quality was still better on MAVRIC. Quantitatively, the area of metal artifact measured using MAVRIC was markedly less compared to STIR (61.4 cm2 vs 135.9 cm2).
Despite the relatively small number of cases in the present study, our observation strongly suggests that MAVRIC is able to improve the quality of images by decreasing the artifact caused by endoprosthesis, frequently utilized in reconstruction of musculoskeletal tumor patients. Further installments of conventional imaging sequences with the addition of gadolinium - enhancement will enable increased accuracy in diagnosing local recurrences of sarcoma patients.
KeywordsMRI MAVRIC Metal artifact
Multimodal therapy for sarcoma patients has improved the prognosis in the past decades, but surgery is still the mainstay of treatment for malignant tumors. Surgery for sarcoma of both bone and soft tissue usually consists of wide resection including the cuff of normal tissue, which leads to large defects. When skeletal structure is compromised, especially near the joint, endoprosthesis is frequently utilized for the reconstruction. Traditionally, radiograph was utilized to detect the change in bone, but it has not enabled the detection of devastating postoperative events such as soft tissue recurrences. The usage of massive metal implants have prevented the usage of magnetic resonance imaging (MRI) which has high spatial resolution and tissue contrasts compared to plain radiographs due to metal artifacts.
Recently, improvements in metal composition of prosthesis [1, 2] and artifact-reducing MRI sequences have been reported [3–5]. Multiacquisition variable-resonance image combination (MAVRIC) was first reported in 2009, where it uses multiple excitations to excite the overall volume being imaged . MAVRIC excites a series of limited spectral distortion and uses the three-dimensional (3D) spin-echo acquisition to resolve the profile of each excited slice in the region of interest. MAVRIC has been reported to reduce the metal artifacts around smaller metal implants usually reserved for conventional orthopaedic procedure [4, 6, 7]. In this study, we analyzed the effectiveness of MAVRIC for suppression of metal artifacts around larger tumor endoprosthesis for early possible detection of tumor recurrences.
There are many histological subtypes of sarcoma of both bone and soft tissues, and treatment is increasingly becoming case-specific with the advent of genetic testing. Several molecular targeting agents have been reported to improve the prognosis and many more are set to be reported. Although, these medical therapies are important, almost all sarcomas need some kind of surgical intervention to eradicate the tumor. It is especially important to perform a wide resection of the lesion to prevent local recurrence from the residual sarcoma cells. To reconstruct large defects, several methods have been proposed including endoprosthesis, allografts, autografts, and recycled bone by pasteurization or liquid nitrogen. When the defect is situated around the joint, endoprosthesis is often utilized due to the poor outcome using other reconstruction methods.
In order to improve the outcome of sarcoma patients, it is important to control the various postoperative adverse events such as acute infection and local recurrence as well as the distant metastasis which is a devastating event both physically and psychologically. Infection leads to multiple salvage operations and delay of chemotherapy which is a poor prognosticator of survival. Local recurrences have been reported to cause higher morbidity due to consequent distant metastasis . One of the reasons for poor outcome after local recurrence is the difficulty of additional surgery. It is difficult to ascertain the tumor infiltration after multiple operations and it is even more difficult after endoprosthesis placement. One of the method recently reported utilize positron emission tomography (PET) - CT to localize the recurrence in the vicinity of metal implants . We have also reported on the possibility of local recurrence detection using PET-CT, where SUVmax of 5.0 or greater might differentiate between tumor and fibrosis (data not shown). But, utilization of CT still causes artifacts due to beam hardening which pose significant challenge in delineating the tumor for successful surgery. Novel method for early diagnosis of postoperative adverse event is imperative to improve the outcome of sarcoma patients.
MRI has improved the prognosis of sarcoma patients by accurately detecting the localization and spread of the tumor prior to operation. It has the highest spatial resolution and tissue contrasts compared to other imaging modalities, but metal implants after surgery have impeded its accuracy by causing artifacts that results from metal disturbing the main magnetic field and inducing strong and spatially - varying local gradients . MRI implementation on tumor endoprosthesis including expandable prosthesis has been reported to be safe without local hotness or unintentional lengthening . MAVRIC and the slice encoding for metal artifact attenuation technique has shown promising results by reducing artifacts. MAVRIC has been reported in several reports after orthopaedic implant surgery as a possible solution for suppression of artifacts [4, 6, 7, 13]. MAVRIC has been able to detect joint abnormalities such as joint effusion and bursitis in painful hip  and shoulder , and showed clinical relevance after total knee replacements . Results from this study also demonstrate that MAVRIC correction for metal induced artifacts improved postoperative visualization around the endoprosthesis. Fortunately, no local recurrence was detected in this case series, but imaging finding of local fluid collection was confirmed by subsequent invasive treatment in one case.
A limitation of this study is its relatively small sample size, which is consistent with other past small case series. Although the preparation of images is still time consuming, it is well suited for dedicated application in areas under suspicion. Further improvement, such as the introduction of MAVRIC - T1WI will enable the use of gadolinium - enhancement and increase the sensitivity for small lesions near the metal implants.
Several variations of metal artifact suppression technique such as FSE, patient positioning, and use of high bandwidth have been reported [5, 15–17], but there is still significant challenge in detecting small lesions near metal implants. MAVRIC significantly improved the image quality and has the potential to improve patient managements. However, further prospective randomized studies are needed to establish the optimum use of MAVRIC for early diagnosis of local recurrences after sarcoma surgeries.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Wang JC, Yu WD, Sandhu HS, Tam V, Delamarter RB. A comparison of magnetic resonance and computed tomographic image quality after the implantation of tantalum and titanium spinal instrumentation. Spine (Phila Pa 1976). 1998;23(15):1684–8.View ArticleGoogle Scholar
- Lee MJ, Kim S, Lee SA, Song HT, Huh YM, Kim DH, et al. Overcoming artifacts from metallic orthopedic implants at high-field-strength MR imaging and multi-detector CT. Radiographics. 2007;27(3):791–803.View ArticlePubMedGoogle Scholar
- Koch KM, Lorbiecki JE, Hinks RS, King KF. A multispectral three-dimensional acquisition technique for imaging near metal implants. Magn Reson Med. 2009;61(2):381–90.View ArticlePubMedGoogle Scholar
- Kretzschmar M, Nardo L, Han MM, Heilmeier U, Sam C, Joseph GB, et al. Metal artefact suppression at 3 T MRI: comparison of MAVRIC-SL with conventional fast spin echo sequences in patients with Hip joint arthroplasty. Eur Radiol. 2015;25(8):2403–11.View ArticlePubMedGoogle Scholar
- Vandevenne JE, Vanhoenacker FM, Parizel PM, Butts Pauly K, Lang RK. Reduction of metal artefacts in musculoskeletal MR imaging. JBR-BTR. 2007;90(5):345–9.PubMedGoogle Scholar
- Choi SJ, Koch KM, Hargreaves BA, Stevens KJ, Gold GE. Metal artifact reduction with MAVRIC SL at 3-T MRI in patients with hip arthroplasty. AJR Am J Roentgenol. 2015;204(1):140–7.PubMed CentralView ArticlePubMedGoogle Scholar
- Hayter CL, Koff MF, Shah P, Koch KM, Miller TT, Potter HG. MRI after arthroplasty: comparison of MAVRIC and conventional fast spin-echo techniques. AJR Am J Roentgenol. 2011;197(3):W405–11.View ArticlePubMedGoogle Scholar
- Gutierrez LB, Do BH, Gold GE, Hargreaves BA, Koch KM, Worters PW, et al. MR imaging near metallic implants using MAVRIC SL: initial clinical experience at 3T. Acad Radiol. 2015;22(3):370–9.View ArticlePubMedGoogle Scholar
- Trovik CS, Gustafson P, Bauer HC, Saeter G, Klepp R, Berlin O, et al. Consequences of local recurrence of soft tissue sarcoma: 205 patients from the Scandinavian Sarcoma Group Register. Acta Orthop Scand. 2000;71(5):488–95.View ArticlePubMedGoogle Scholar
- Gelfand MJ, Sharp SE. [F-18]2-fluoro-2-deoxyglucose (FDG) positron emission tomography after limb salvage surgery: post-surgical appearance, attenuation correction and local complications. Pediatr Radiol. 2015;45(8):1182–8.View ArticlePubMedGoogle Scholar
- Wendt 3rd RE, Wilcott 3rd MR, Nitz W, Murphy PH, Bryan RN. MR imaging of susceptibility-induced magnetic field inhomogeneities. Radiology. 1988;168(3):837–41.View ArticlePubMedGoogle Scholar
- Ogg RJ, McDaniel CB, Wallace D, Pitot P, Neel MD, Kaste SC. MR safety and compatibility of a noninvasively expandable total-joint endoprosthesis. Magn Reson Imaging. 2005;23(7):789–94.View ArticlePubMedGoogle Scholar
- Sutter R, Ulbrich EJ, Jellus V, Nittka M, Pfirrmann CW. Reduction of metal artifacts in patients with total hip arthroplasty with slice-encoding metal artifact correction and view-angle tilting MR imaging. Radiology. 2012;265(1):204–14.View ArticlePubMedGoogle Scholar
- Chen CA, Chen W, Goodman SB, Hargreaves BA, Koch KM, Lu W, et al. New MR imaging methods for metallic implants in the knee: artifact correction and clinical impact. J Magn Reson Imaging. 2011;33(5):1121–7.PubMed CentralView ArticlePubMedGoogle Scholar
- Butts K, Pauly JM, Gold GE. Reduction of blurring in view angle tilting MRI. Magn Reson Med. 2005;53(2):418–24.View ArticlePubMedGoogle Scholar
- Cho ZH, Kim DJ, Kim YK. Total inhomogeneity correction including chemical shifts and susceptibility by view angle tilting. Med Phys. 1988;15(1):7–11.View ArticlePubMedGoogle Scholar
- Walde TA, Weiland DE, Leung SB, Kitamura N, Sychterz CJ, Engh Jr CA, et al. Comparison of CT, MRI, and radiographs in assessing pelvic osteolysis: a cadaveric study. Clin Orthop Relat Res. 2005;437:138–44.View ArticlePubMedGoogle Scholar