This article has Open Peer Review reports available.
Revision TKA with a condylar constrained prosthesis using metaphyseal and surface cementation: a minimum 6-year follow-up analysis
© Sanz-Ruiz et al.; licensee BioMed Central. 2015
Received: 12 October 2014
Accepted: 29 January 2015
Published: 25 February 2015
The increasing number of revision knee arthroplasty result in the more frequently use of a constraint implant but results from previous reports are difficult to interpret. The purpose of this study was to compare the long-term outcomes of superficial cemented versus metaphyseal cemented in revision total knee arthroplasty with a condylar constrained arthroplasty.
The study was a retrospective analysis of clinical and radiographic outcomes in a series of revision total knee arthroplasties performed with a constrained condylar knee prosthesis and press-fit modular stems. We hypothesized that the clinical and radiographic outcome of surface cementation would be inferior to that of metaphyseal cementation. Fifty-two consecutive revision cases were followed for a median of 8.2 years (range, 6 to 10 years).
Substantial improvements in range of motion and Knee Society score were achieved in all patients, although these were not significant between groups. Significantly more radiolucent lines were visible on the tibial component with surface cementation than with metaphyseal cementation, although the clinical differences were not relevant.
Radiologic outcome was better in revision total knee arthroplasty using metaphyseal cemented revision and components with press-fit cementless stems than in the surface cementation–based approach; however, the difference was not clinically relevant.
KeywordRevision TKA Constrained condylar Fixation Press-fit stem
The increasing number of patients undergoing primary total knee arthroplasty (TKA) has been accompanied by an increase in the number of revision knee arthroplasty procedures, with an estimated cost of €50,000-70,000 per procedure [1,2]. The goal of revision knee arthroplasty is to restore the anatomy and function of the joint. However, bone loss and soft tissue instability are substantial obstacles to obtaining adequate joint line restoration  and ligament balancing, which are crucial when deciding on a suitable choice of implant. The constrained condylar knee (CCK) prosthesis was developed to resist coronal plane stresses arising from soft-tissue instability.
Results from previous reports of patients treated with modern CCK prostheses are difficult to compare, because different types of prostheses were used and follow-up periods were fairly short. In this report, we aim to provide a more robust clinical and radiographic evaluation of the CCK arthroplasty procedures performed in our institution between 2004 and 2008 using the same type of prosthesis for patients with instability or severe bone loss. We evaluated the functional outcome of the knees, radiographic results, and potential complications.
All patients gave their informed consent before inclusion in the study. Ours was a retrospective study performed in accordance with the principles of the 1964 Declaration of Helsinki as revised in 2013 and was approved by the research ethics committee of the Gregorio Marañon General Hospital.
Data from the 53 procedures were entered into a computerized registry. Data on all the knees were reviewed by an orthopedic surgeon who was not involved in the original procedure. Clinical and radiologic follow-up evaluations were planned at postoperative intervals of 1 month, 3 months, 6 months, and 1 year, and yearly thereafter. Preoperative and postoperative review data were collected according to the scoring systems of the Knee Society .
Normotensive epidural anesthesia was used in all the procedures. All the knees were approached through a previous anterior midline incision scar. A quadriceps snip  was required in 13 knees, and a tibial tubercle osteotomy in 3. All the osteotomies and 5 snips were performed in septic cases during the second stage.
n = 10
n = 37
The functional and clinical results were evaluated according to the Knee Society Score and compared with the radiographic and demographic variables mentioned above.
Statistical analysis was performed using SPSS 20.0 for Windows (IBM Corp., Armonk, New York, USA). All data were checked for normality using the Shapiro-Wilk W test. The t test was used to compare scores for the normal distribution between the preoperative and postoperative data. The degree of satisfaction between the 2 groups was compared using the chi-square test. Statistical significance was set at p <0.05. Kaplan-Meier curves were used to analyze the survival of the prostheses.
Demographic and clinical data
Primary diagnosis, No. of knees (%)
Reason for revision, number of knees (%)
Prosthesis before revision (no [%] of knees)
Mean duration between primary and revision TKAs (months) (range)
Duration of follow-up (months) (range)
Clinical results at the end of follow-up
4.4 (−15 to 30)
2.44 (0 to 15)
90.4 (60 to 120)
105.5 (70 to 130)
Range of movement (range)
86 (30 to 120)
108 (70 to 130)
Knee Society knee score (points) (range)
39.4 (0 to 67)
78.7 (45 to 100
Knee Society function score (points) (range)
32.1 (0 to 73)
56 (5 to 100)
The Knee Society score increased in all patients after revision surgery. At 1 year, the clinical score had increased by a mean of 42.9 (14 to 74); however, at the end of follow-up this value had decreased by a mean of 3.6 points. No statistical differences were found between the time points. Clinical outcome was rated as excellent in 21 revisions (44.7%), good or moderate in 18 (38.3%), and poor in 8 cases (17%). The functional score increased by a mean of 29.3 (5 to 85) at 1 year after surgery; however, at the end of follow-up it had decreased by a mean of 5.4 (−30 to +40) (Table 3). No statistical differences were detected between the scores at 1 year or at the end of follow-up, when the score for function was excellent in 18 revisions (38.3%), good or moderate in 19 (40.4%), and poor in 10 cases (21.3%). No differences were observed in the Knee Society scores between aseptic and septic cases (p = 0.4) or between the different cementing techniques used.
Radiographic results at the end of follow-up
6 (0 to 20)
2.97 (0 to 6)
94.3 (86 to 104)
97.6 (87 to 100)
5.2 (−3 to 12)
4.17 (0 to 10)
85.4 (80 to 91)
87.4 (85 to 93)
88.2 (81 to 94)
84.2 (81 to 87)
Joint line (cm)
1.4 (0.9 to 2.1)
1.6 (0.8 to 2.3)
Radiolucency around the implants
Rupture of extensor apparatus (intraoperative)
Condylar fracture (intraoperative)
Favorable clinical outcomes with revision condylar constrained TKA have been reported by several authors and include improved quality of life, pain relief, increased walking distance, reduced deformity, and enhanced function [11-16]. However, comparison of clinical outcome and complications is difficult, as is comparison of condylar-constrained TKA studies in the literature because of the different types of prostheses and the relatively short follow-up periods.
Hossain et al.  reported the clinical results of 349 revision TKAs after a mean follow up of 57.7 months. The authors compared 3 different implant designs and found that functional outcome and range of motion improved significantly, irrespective of the implant used. Kim et al.  reported the clinical results of 114 LCCK-based revision TKAs after a follow-up of 7.2 years. Of the 114 knees studied, clinical improvement was achieved in 91%, and radiographically stable implant fixation was observed in 96%. The mean range of motion was 95° before surgery and 106° at the end of follow-up; however, both posterior stabilized and CCK liners were used. Harwin  reported excellent results with the 18 Kinemax Plus Superstabilizer prosthesis after a follow-up of 11.3 years. Septic revision, supracondylar fracture, and neuropathic arthropathy were excluded in this series.
In our study, the mean Knee Society clinical score of the 47 patients at the end of follow-up (mean, 6.5 years) was 78.7 (45 to 100); 39 patients (83%) had an excellent or good result. Our findings are comparable to those reported by Lee et al.  in the longest follow-up of a single prosthesis used for revision TKA with only a CCK liner. The series contained roughly similar numbers of septic and aseptic revisions.
Several clinical studies have suggested that radiolucent lines are more frequent after revision TKA than after primary TKA [21,22]. One early clinical study of revision TKA using the Total Condylar-III prosthesis found radiolucent lines in 22 of 36 patients (61%) . In another series of 14 TKAs, radiolucent lines were identified around the tibial component in 10 knees and around the femoral component in 4 knees . In 2 recent studies, radiolucent lines were observed in 36% (13 out of 36 knees) to 72% (28 out of 39 knees) after revision TKA using different system designs [23,24]. Kim et al.  identified radiolucent lines around the tibial component in 31% (35 out of 114 knees) and in 18% (21 knees) on the femoral side, although most implants were posterior stabilized. In the present study, radiolucent lines were identified in 49% (23), 34% (16) around the tibial component, especially in zone 3 in the lateral tibial view and zone 4 in the anteroposterior tibial view, and in 17% (8) around the femoral component, especially in zone 5 in the lateral femoral view.
The ideal method of fixation in revision TKR remains controversial. The use of cement for at least a portion of the femoral and tibial components is well accepted in most procedures; however, the issue of whether to cement the stemmed portion of the implant is not clear . Cemented fixation has the advantage that antibiotics can be added to the cement , although its disadvantages include difficulty in removing cement at re-revision (if necessary) and the potential for stress shielding . Murray et al.  showed sclerotic lines around long cemented stems and similar trabecular patterns around the stem in cemented and non-cemented stems at 5 years of follow-up and concluded that there was no stress shielding. Completo et al. , on the other hand, recently demonstrated stress shielding in cemented stems in vitro. Marx et al.  recommended surface cementation to avoid stress shielding. To our knowledge, there are no reports directly comparing the cementation technique with hybrid fixation in revision TKA.
In our series, we observed a higher proportion of radiolucent lines at the bone-cement interface with surface cementation than with metaphyseal cementation. These results are consistent with those published by Conlisk et al.  in an in vitro study that showed better initial stability when more cement was used. Wood et al.  reported 87% survival with metaphyseal cementation and press-fit stem at 12 years of follow-up in 127 patients. Our findings (80%) are consistent with those of other series. Lee et al.  reported a survival rate of 83% at 8 years in 79 revisions, and Peters et al.  reported a rate of 80% at 80 months in 43 revisions. This study showed a survival rate of 80%, which was lower than in other studies [33,34] but similar to that reported by Conlisk et al. .
All implants in our series were CCK and not posterior stabilized. To our knowledge, there are no reports directly comparing stem type and fixation method in revision TKA.
Our study is subject to a series of limitations. First, clinical and radiological data were analyzed retrospectively. Second, the cohort was too small to determine the implant survival rate. Third, no control group was included. Finally, as ours is a major referral center, referral bias is possible. Although these factors might account for some of the differences in implant survival, our results were comparable to those published for knee revision using similar constrained systems.
The strengths of our study include the homogeneity of the approach used, namely, a single knee revision system, with a CCK liner in all cases. Furthermore, all the procedures were performed at a single institution by 2 senior surgeons, and postoperative data were collected using a uniform, controlled protocol through the joint registry; consequently, the likelihood of differential measurement bias is reduced. Nevertheless, a multicenter trial is necessary to ensure more robust results. Finally, our analysis combined both implant survival with long-term clinical outcome measures and patient-reported subjective measures to evaluate the performance of these implants.
To our knowledge, ours is the first study to provide encouraging results for range of movement, clinical scores, implant fixation stability, and freedom from complications with a follow-up of more than 5 years based on the same CCK arthroplasty system. Although no significant functional differences were recorded between the groups, further re-evaluations of these series could elucidate the potential long-term significance of the radiolucent lines, which are more frequent in the group with surface cementation.
In conclusion, the modular, fixed-bearing LCCK TKA (Zimmer™) system has an acceptable long-term survival rate in both septic and aseptic revisions, although the frequency of complications is high. Metaphyseal cementation provides better radiologic outcome than surface cementation; however, the difference was not clinically significant. For patients with long-term follow-up and implant survival, values for objective clinical outcome measures and subjective patient-reported outcome measures were significantly better than the preoperative values.
The authors wish to thank the Department of Statistics of the Gregorio Marañón General Hospital and Mr Jose Maria Bellon for his assistance in the statistical analysis. There was no external funding.
- Garrido-Gomez J, Arrabal-Polo MA, Giron-Prieto MS, Cabello-Salas J, Torres-Barroso J, Parra-Ruiz J. Descriptive analysis of the economic costs of periprosthetic joint infection of the knee for the public health system of Andalusia. J Arthroplasty. 2013;28:1057–60.View ArticlePubMedGoogle Scholar
- Kurtz SM, Ong KL, Schmier J, Mowat F, Saleh K, Dybvik E, et al. Future clinical and economic impact of revision total hip and knee arthroplasty. J Bone Joint Surg Am. 2007;89 Suppl 3:144–51.View ArticlePubMedGoogle Scholar
- Partington PF, Sawhney J, Rorabeck CH, Barrack RL, Moore J. Joint line restoration after revision total knee arthroplasty. Clin Orthop Relat Res. 1999;367:165–71.View ArticlePubMedGoogle Scholar
- Engh GA, Ammeen DJ. Bone loss with revision total knee arthroplasty: defect classification and alternatives for reconstruction. Instr Course Lect. 1999;48:167–75.PubMedGoogle Scholar
- Insall JN, Dorr LD, Scott RD, Scott WN. Rationale of the Knee Society clinical rating system. Clin Orthop Relat Res. 1989;248:13–4.PubMedGoogle Scholar
- Garvin KL, Scuderi G, Insall JN. Evolution of the quadriceps snip. Clin Orthop Relat Res. 1995;321:131–7.PubMedGoogle Scholar
- Servien E, Viskontas D, Giuffre BM, Coolican MR, Parker DA. Reliability of bony landmarks for restoration of the joint line in revision knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2008;16:263–9.View ArticlePubMedGoogle Scholar
- Villanueva M, Rios A, Pereiro J, Chana F, Fahandez-Saddi H. Hand-made articulating spacers for infected total knee arthroplasty: a technical note. Acta Orthop. 2006;77:329–32.View ArticlePubMedGoogle Scholar
- Villanueva Martínez M, Rios A, Chana F, Pereiro J, Fahandez-Saddi H, García-Lechuz J, et al. Two step surgical treatment of infected total knee replacement: comparative study of two different types of spacers. Rev Ortop Traumatol. 2005;49:346–57.View ArticleGoogle Scholar
- Ewald FC. The Knee Society total knee arthroplasty roentgenographic evaluation and scoring system. Clin Orthop Relat Res. 1989;248:9–12.PubMedGoogle Scholar
- Kim YH. Salvage of failed hinge knee arthroplasty with a Total Condylar III type prosthesis. Clin Orthop Relat Res. 1987;221:272–7.PubMedGoogle Scholar
- Haas SB, Insall JN, Montgomery 3rd W, Windsor RE. Revision total knee arthroplasty with use of modular components with stems inserted without cement. J Bone Joint Surg Am. 1995;77:1700–7.PubMedGoogle Scholar
- Rosenberg AG, Verner JJ, Galante JO. Clinical results of total knee revision using the Total Condylar III prosthesis. Clin Orthop Relat Res. 1991;273:83–90.PubMedGoogle Scholar
- Rand JA. Revision total knee arthroplasty using the total condylar III prosthesis. J Arthroplasty. 1991;6:279–84.View ArticlePubMedGoogle Scholar
- Peters CL, Hennessey R, Barden RM, Galante JO, Rosenberg AG. Revision total knee arthroplasty with a cemented posterior-stabilized or constrained condylar prosthesis: a minimum 3-year and average 5-year follow-up study. J Arthroplasty. 1997;12:896–903.View ArticlePubMedGoogle Scholar
- Sheng PY, Jamsen E, Lehto MU, Konttinen YT, Pajamaki J, Halonen P. Revision total knee arthroplasty with the Total Condylar III system in inflammatory arthritis. J Bone Joint Surg Br. 2005;87:1222–4.View ArticlePubMedGoogle Scholar
- Hossain F, Patel S, Haddad FS. Midterm assessment of causes and results of revision total knee arthroplasty. Clin Orthop Relat Res. 2010;468:1221–8.View ArticlePubMedPubMed CentralGoogle Scholar
- Kim YH, Kim JS. Revision total knee arthroplasty with use of a constrained condylar knee prosthesis. J Bone Joint Surg Am. 2009;91:1440–7.View ArticlePubMedGoogle Scholar
- Harwin SF. Revision total knee arthroplasty using the Kinemax Plus Superstabilizer prosthesis: minimum 10-year follow-up. Knee. 2006;13:93–7.View ArticlePubMedGoogle Scholar
- Lee KJ, Moon JY, Song EK, Lim HA, Seon JK. Minimum two-year results of revision total knee arthroplasty following infectious or non-infectious causes. Knee Surg Relat Res. 2012;24:227–34.View ArticlePubMedPubMed CentralGoogle Scholar
- Jacobs MA, Hungerford DS, Krackow KA, Lennox DW. Revision total knee arthroplasty for aseptic failure. Clin Orthop Relat Res. 1988;226:78–85.PubMedGoogle Scholar
- Insall JN, Dethmers DA. Revision of total knee arthroplasty. Clin Orthop Relat Res. 1982;170:123–30.PubMedGoogle Scholar
- Mow CS, Wiedel JD. Revision total knee arthroplasty using the porous-coated anatomic revision prosthesis: six- to twelve-year results. J Arthroplasty. 1998;13:681–6.View ArticlePubMedGoogle Scholar
- Takahashi Y, Gustilo RB. Nonconstrained implants in revision total knee arthroplasty. Clin Orthop Relat Res. 1994;309:156–62.PubMedGoogle Scholar
- Peters CL, Erickson J, Kloepper RG, Mohr RA. Revision total knee arthroplasty with modular components inserted with metaphyseal cement and stems without cement. J Arthroplasty. 2005;20:302–8.View ArticlePubMedGoogle Scholar
- Elia EA, Lotke PA. Results of revision total knee arthroplasty associated with significant bone loss. Clin Orthop Relat Res. 1991;309;271:114–21.Google Scholar
- Completo A, Fonseca F, Simoes JA. Strain shielding in proximal tibia of stemmed knee prosthesis: experimental study. J Biomechanics. 2008;41:560–6.View ArticleGoogle Scholar
- Murray PB, Rand JA, Hanssen AD. Cemented long-stem revision total knee arthroplasty. Clin Orthop Relat Res. 1994:116–23.Google Scholar
- Marx R, Qunaibi M, Wirtz DC, Niethard FU, Mumme T. Surface pretreatment for prolonged survival of cemented tibial prosthesis components: full- vs. surface-cementation technique. Biomed Eng Online. 2005;4:61.View ArticlePubMedPubMed CentralGoogle Scholar
- Conlisk N, Gray H, Pankaj P, Howie CR. The influence of stem length and fixation on initial femoral component stability in revision total knee replacement. Bone Joint Res. 2012;1:281–8.View ArticlePubMedPubMed CentralGoogle Scholar
- Wood GC, Naudie DD, MacDonald SJ, McCalden RW, Bourne RB. Results of press-fit stems in revision knee arthroplasties. Clin Orthop Relat Res. 2009;467:810–7.View ArticlePubMedGoogle Scholar
- Lee JK, Lee S, Kim D, Lee SM, Jang J, Seong SC, et al. Revision total knee arthroplasty with varus-valgus constrained prosthesis versus posterior stabilized prosthesis. Knee Surg Sports Traumatol Arthrosc. 2013;21:620–8.View ArticlePubMedGoogle Scholar
- Lee JK, Kim SJ, Choi CH, Chung HK. Revision total knee arthroplasty using a constrained condylar knee prosthesis in conjunction with a posterior stabilized articular polyethylene. J Arthroplasty. 2013;28:566–9.View ArticlePubMedGoogle Scholar
- Manopoulos P, Havet E, Pearce O, Lardanchet JF, Mertl P. Mid- to long-term results of revision total knee replacement using press-fit intramedullary stems with cemented femoral and tibial components. J Bone Joint Surg Br Vol. 2012;94:937–40.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.