- Research article
- Open Access
- Open Peer Review
Comparison of computer-navigated and conventional total knee arthroplasty in patients with Ranawat type-II valgus deformity: medium-term clinical and radiological results
© Huang et al.; licensee BioMed Central Ltd. 2014
- Received: 24 April 2014
- Accepted: 11 November 2014
- Published: 22 November 2014
Arthritic knees with Ranawat type-II valgus deformity present with soft tissue contracture and osseous anomalies that make total knee arthroplasty (TKA) difficult. We hypothesized that computer-navigated-TKA (CN-TKA) may be superior to conventional techniques and provide better mid-term radiographic and clinical outcomes in such cases.
Between January 2002 and January 2009, patients with Ranawat type-II valgus deformity who underwent primary TKA were entered into this retrospective study. Conventional TKA and CN-TKA were compared for the accuracy of component placement, joint line level, and postoperative limb alignment. International Knee Society scores and patellar scores were used for clinical assessment.
A total of 62 patients (70 knees) with a minimum of 5 years of follow-up were studied. Conventional TKA was performed in 36 knees and CN-TKA in 34 knees. A significantly higher rate of lateral retinaculum release was recorded in the conventional TKA group compared to the CN-TKA group. Proper restoration of joint line was achieved using CN-TKA. The range of motion of the knees was similar in both groups preoperative and postoperatively. There were no significant differences in reconstructed mechanical axes, accuracy of component positioning, and difference in perioperative hemoglobin level between the two groups. At a mean follow-up of 6.2 years, both groups had significant postoperative improvements in clinical performance, however the difference did not reach statistical significance between both techniques.
Our findings demonstrate that CN-TKA can properly restored the joint line level for arthritic knees with Ranawat type II valgus deformity. However, no differences in clinical function, limb and component alignment, or survival of the prostheses were noted between the CN-TKA and conventional TKA groups at a mean follow-up of 6.2 years.
- Genu valgus deformity
- Computer-navigated surgery
- Total knee arthroplasty
- Cruciate retaining
Total knee arthroplasty (TKA) is a successful procedure for end-stage arthritis of the knee joint . Studies on the biomechanical and long-term survival have reported that the reconstructed mechanical axis and component positions should be within 3 degrees of neutral to prevent abnormal wear, premature mechanical loosening of the components, and patellofemoral problems [2–5]. Following the development of computer-navigated TKA (CN-TKA), more accurate bone cuts, more precise component placement in coronal, sagittal and axial planes, better restoration of coronal limb alignment and joint line, and less gap asymmetry have been reported [6–13].
Approximately 10% of patients requiring TKA have a valgus deformity, and this deformity was classified into three types by Ranawat et al. [14, 15]. The type-I deformity accounts for 80% of all valgus knees, and is defined as a preoperative valgus angulation of anatomical axis <10° based on hip-to-ankle standing radiography. The medial collateral ligament (MCL) is functional and intact in this type of deformity, and performing TKA is not more challenging than usual. The type-III deformity (5% of valgus knees) includes an anatomical axis >20° and an impaired medial soft-tissue sleeve. A varus-valgus constrained type of prosthesis is often required to compensate for the lax MCL. The type-II valgus deformity is seen in 15% of valgus knees, and has a more substantial deformity (anatomical axis ≥10°) with medial soft tissue being stretched and elongated, but functional. Furthermore, the osseous deformity in Ranawat type-II valgus deformity (including hypoplasia of the distal femur, rotational deformity of the tibia and femur, unusual proximal femoral neck-shaft angles, and maltracking of the patella) may influence the accuracy of conventional instrumentation and contribute to improper bone cuts. In this circumstance, conventional TKA is technically challenging and may have greater risks of malposition of the components, improper joint line and unplanned conversion to a constrained prosthesis intraoperatively [14–16]. Some surgeons find it difficult to correct an advanced valgus deformity using conventional instrumentation without a constrained prosthesis [17–19].
CN-TKA has the theoretical advantages in dealing with Ranawat type-II valgus deformity of precise bone cuts in conjunction with meticulous soft tissue release with real-time and quantitative feedback . However, only a few cohort studies have been published regarding whether or not radiographic improvements achieved with CN-TKA in valgus deformity lead to an improvement in functional scores or implant survival after mid- or long-term follow-up [21, 22]. We hypothesized that CN-TKA would be superior to conventional TKA, and thus the aim of this study was to retrospectively investigate the effect of CN-TKA compared with conventional TKA on mid-term clinical function, alignment, and implant survival.
Between January 2002 and January 2009, patients who had arthritic knees with Ranawat type-II valgus deformity (a preoperative valgus angulation of anatomical axis ≥10° based on hip-to-ankle standing radiography) and who underwent primary TKA were enrolled in this retrospective study. The cost of CN was not reimbursed in our national health insurance, the type of surgery was chosen by the patients themselves after well explaining the merits of both techniques to patients.
The clinical and radiographic data and functional outcomes were reviewed. The clinical data included age, gender, diagnosis, intraoperative procedures, tourniquet time (the tourniquet was inflated before skin incision and was deflated after the setting of cement), difference in perioperative hemoglobin level (which was checked before the operation and in the morning of the second postoperative day), and antibiotic prophylaxis. The radiographic and functional assessments before surgery and at the time of each follow-up were analyzed and recorded by a research associate (Yu-Shuan, Lin) who was not part of the operative team and was blinded to technique allocation. This study was approved by the Ethics Committee and Institutional Review Board of Chang Gung Memorial Hospital (99-2386B). Informed consent for participation in the study was obtained from all participants.
A tourniquet was used for all patients. The procedure was carried out through a midline skin incision with the use of a medial parapatellar arthrotomy. All cases received the same cruciate-retaining type of cemented total knee prosthesis (DePuy PFC Knee Systems, DePuy Orthopaedics, Warsaw, Indiana).
The CN-TKA was implanted with the use of a computed tomography (CT)-free navigation system (BrainLAB, Munich, Germany). After exposing the knee joint by using an anterior midline longitudinal skin incision and a medial parapatellar arthrotomy, the synovium was adequately removed to make registration precise. The femoral reference arrays were fixed to the distal femur using two bi-cortical half-pins, and the tibial reference arrays were fixed to the proximal tibia. An infrared camera was used to track the femoral and tibial reference arrays. The implant size and orientation were identified by dragging the pointer along the bone surface to reconstruct the three-dimensional bone model. The rotational alignment of the femur was checked by the transepicondylar axis under navigation. The contracted soft tissue was sequentially released using Whiteside’s method , and the tension of the PCL was assessed using the method of Scott and Chmell  with real-time and quantitative feedback of the navigation. The assessment and management of bony defect were performed. The femoral and tibial reference arrays were retained until the cement had fully set, and were removed after the alignment was verified under navigation. The tourniquet was then deflated and hemostasis. Assessment of patellar tracking were performed by using “The no thumb test”. All procedures were performed by a senior surgeon (Hsu RWW) who has extensive experience in the use of both conventional mechanical guides and computer-assisted navigation.
A continuous passive motion machine was used for passive range of motion exercises four times daily (30 minutes per exercise period) from the day of surgery until the day of discharge from the hospital. Under the supervision of a physical therapist, the patients started active knee-motion exercises and began standing at the bedside or walking with crutches or a walker twice daily for 30-minute periods. All patients used crutches or a walker with full weight-bearing for six weeks, and a cane when needed thereafter.
All patients received hip-to-ankle standing radiography along with anteroposterior and lateral radiography of the knees and the skyline view of the patellofemoral joint before and after surgery, following a protocol described in detail in a previous publication by the senior author . The lateral patellar tilt and displacement were measured according to the method of Laurin et al. [27, 28]. The postoperative patellar tilting angle was measured by the method described by Kawahara S et al. . Radiographic parameters including the mechanical axis and four component alignment angles (the femoral valgus angle, tibial valgus angle, femoral flexion angle, and tibial flexion angle ) were assessed. The position of the prosthetic joint line was measured on radiographs taken before surgery and at the last follow-up using the method proposed by Figgie et al. . In brief, the length of the joint line measured in the lateral view was the perpendicular distance from a weight bearing surface of the tibial plateau to the tibial tubercle of the natural tibia. The relative change in joint line distance was recorded. All measurements were done by a blinded observer using digital radiographs.
We considered a tibial component positioned at a valgus angle of 90° in the coronal plane and at a flexion angle of 87° in the sagittal plane (3° of posterior slope) to be ideal. A femoral component positioned at a flexion angle of 0° in the sagittal plane was considered to be ideal, and the desired valgus angle in the coronal plane was determined according to the valgus correction angle of the distal femur which was measured by hip-to-ankle standing radiography. The goal of alignment was to correct the postoperative mechanical axis to within 3° of neutral, and all component angles to within 3 degrees of the ideal.
Preoperative and postoperative functional scores were calculated using the International Knee Society (IKS) scoring system  and patellar score . The patellar score was designed by Feller et al. to assess the management of the patellofemoral joint. Active maximum range of motion of the knee was measured using a goniometer. The patients who had extra-articular deformity of the femur or tibia related to trauma or previous surgery and those with incomplete medical records, radiographic analysis and functional evaluation were excluded.
All data were entered into an Excel spreadsheet (Microsoft Corp, Redmond, WA), checked for missing and illogical entries, and subsequently copied into SPSS version 13.0 (SPSS Inc., Chicago, IL). Statistical analysis was performed by an independent statistician blinded to the surgical procedures. The independent t test and chi-square test were used for comparisons of preoperative and postoperative (the last follow-up visit) radiographic parameters and functional results. The level of statistical significance was set at p <0.05.
A total of 62 patients (70 knees) were enrolled in this study, with a mean age of 71.3 years (range, 50 to 86 years) at the time of surgery. The diagnosis was degenerative osteoarthritis in 68 knees and rheumatoid arthritis in the other two. The mean body height was 156.2 cm (range, 133.0 to 174.0 cm), mean body weight 64.8 kg (range, 35.0 to 95.0 kg), and mean body mass index 27.5 kg/m2 (range, 19.8 to 38.1 kg/m2). The mean follow-up time was 6.2 years (range, 5–11 years).
Demographic data of the patients in the conventional and CN-TKA groups
N = 36
N = 34
Body height (cm)
155.6 (147.0 – 171.5)
154.4 (133.0 – 174.0)
Body weight (kg)
65.7 (47.2 – 79.1)
66.5 (35.4 – 95.3)
Body mass index (kg/m 2 )
26.8 (21.8 – 31.1)
27.1 (19.8 – 38.1)
Reason for surgery
35 (97.2 %)
33 (97.1 %)
1 (2.8 %)
1 (2.9 %)
Hospital stay (days)
Follow-up time (years)
Perioperative and radiographic data of the conventional and CN-TKA groups
N = 36
N = 34
Difference of perioperative hemoglobin level (g/dL)
1.5 ( 0.6–2.7)
Tourniquet time (minutes)
Lateral retinaculum for patellar tracking
Preoperative AA (deg)
Postoperative MA (deg)
Preoperative congruent angle (deg)
Postoperative patellar tilting angle (deg)
2.4° (range, 1°−4°)
Joint line elevation (mm)
Femoral valgus angle (deg)
Femoral flexion angle (deg)
Tibial valgus angle (deg)
Tibial flexion angle (deg)
Comparison of percentage of postoperative lower limb axis (within 3° deviation) and component positioning between the conventional and CN-TKA groups
No. of postoperative component positioning within 3°deviation
N = 36
N = 34
Mechanical axis within 3° deviation
Femoral valgus angle
Femoral flexion angle
Tibial valgus angle
Tibial flexion angle
Comparison of preoperative and postoperative knee scores between the conventional and CN-TKA groups
N = 36
N = 34
Preoperative functional score
Patellar score (points)
IKS score for pain (points)
IKS score for clinical knee score (points)
IKS score for functional knee score (points)
Active range of motion in degrees (deg)
Postoperative functional score
Patellar score (points)
IKS score for pain (points)
IKS score for clinical knee score (points)
IKS score for functional knee score (points)
Active range of motion in degrees (deg)
No patient required advancement of MCL or was converted to a posterior-stabilized prosthesis or varus-valgus constrained components because of excessive bony resection or inadequate soft tissue release during the operation. There were no cases of wound infection, peroneal nerve neurapraxia, pulmonary emboli, or deep vein thrombosis. No periprosthetic fractures, joint instability, or patellar tracking problems were encountered. No patients showed loosening or osteolysis on radiography at the time of the last follow-up, and no patients received revision surgery for any reason.
Proper alignment of limb and component positions as well as soft tissue balance during TKA is critical in maximizing implant survival [2–5]. It has been reported that CN-TKA allows for a higher accuracy in radiographic parameters and soft tissue balance [6–13]. The osseous deformity in Ranawat type-II valgus deformity may influence the accuracy of conventional instrumentation and contribute to improper bone cuts [14–19]. However, compared to conventional instrumentation, CN-TKA only focuses on the center of the hip, knee, and ankle joints and overlooks extra-articular deformities of the femur and tibia. In addition, CN-TKA provides real-time and quantitative feedback to the surgeon . These characteristics are theoretically beneficial for Ranawat type-II valgus deformities.
A few studies have shown that CN-TKA improved the overall accuracy of limb alignment and positioning of components in patients with valgus deformities [21, 22]. Hadjicostas et al.  studied 15 patients with valgus deformity of 17°-27°, and reported excellent mid-term results using CN-TKA combined with simultaneous osteotomy of the lateral femoral condyle. Chou et al.  reported a case-series study with a mean follow-up of 37.8 months, and concluded that CN-TKA in conjunction with preoperative radiographic templating is an alternative strategy to improve radiographic and clinical results in arthritic valgus knees. However, studies to confirm that the improved radiographic alignment achieved with CN-TKA translates into better mid- or long-term clinical outcomes or the durability of the TKA for Ranawat type-II valgus deformities are lacking. In addition, previous reports have been limited by small sample size, different severity of valgus deformity, short follow-up, and lack of a control group for comparisons to demonstrate the usefulness of CN-TKA. In the current study, CN-TKA provided proper radiographic and functional results without the need for additional osteotomy, and no significant differences with regards to limb alignment, component position and functional outcomes were noted between both techniques even at a mean follow-up of 6.2 years. The absence of significant differences in outliers in radiographic parameters may explain these results.
In TKA, restoration of the joint line is a crucial objective, as an incorrect joint line level may contribute to joint instability, anterior knee pain, limited range of motion, joint stiffness, patellofemoral contact forces, acceleration of polyethylene wear, and inferior clinical results [22, 31, 34, 35]. In order to obtain adequate stability of the knee joint, a thicker tibial polyethylene or unplanned conversion to a varus-valgus constrained type of prosthesis is necessary to compensate for improper bony resection and excessive soft tissue release. However, thicker tibial polyethylene would also result in elevation of the joint line level and increase the risk of peroneal nerve neurapraxia. Chou et al.  reported that the joint line was well restored with the use of navigation. Ensini et al.  concluded that a navigation system can support intraoperative joint line restoration by controlling the exact resection of the bone and the corresponding overall thickness of the components implanted. In the current study, we found a statistically significant difference in joint line elevation between CN-TKA and conventional TKA (p = 0.020) (Table 2).
In the current study, 13 knees (36.1%) in the conventional TKA group required release of the lateral retinaculum to obtain adequate patellar tracking, while only 2 knees (5.9%) required release in the CN-TKA group (p = 0.001). When using conventional femoral guidance jigs, deficiencies of the lateral femoral condyle often render the posterior condylar axis unreliable as a reference to determine femoral component rotation, resulting in an inadequately rotated femoral component. Malrotation of the femoral component affects patellofemoral tracking and may increase the contact pressure and lead to accelerated and excessive wear of the patellar button [36, 37].
Reliable evidence of the effect of CN-TKA on improvements in femoral rotational alignment and implant survival is limited because intraoperative and postoperative measurement techniques are often inaccurate, and because the optimal rotational alignment target has not been defined . Using postoperative CT scans, several authors have demonstrated significant improvements in the rotational alignment of the femoral component with CN-TKA [6, 38–40], however, other researchers have reported no differences between CN-TKA and conventional TKA [13, 41]. This discrepancy may reflect, at least in part, the different types of preoperative knee deformities (varus or valgus deformity) seen in their respective study populations. Further studies using CT to evaluate the rotational alignment are necessary to determine whether CN-TKA merits alternative use in conventional TKA in genu valgum or genu varum deformities.
There are several strengths to this study. First, despite the relative rarity of Ranawat type-II valgus deformity (approximately 10% of patients requiring TKA have a valgus deformity, and type-II deformity accounts for 15% of all valgus knees), we conducted a relatively large-scale cohort study (36 TKAs implanted using conventional instrumentation and 34 using CN-TKA) to compare the alignment and mid-term clinical outcomes. Second, all patients underwent surgery by a single surgeon, the single prosthesis used, the same surgical technique, and the same protocol allowed for decreases in confounding factors.
Several limitations to this study should be acknowledged. First, this was a retrospective study with all its inherent limitations and bias. Second, because the cost of CN was not reimbursed in our national health insurance, the type of surgery was chosen by the patients themselves based on their socioeconomic status. This potential bias may be related to outcome. Third, this study is limited by mid-term clinical follow-up, and we were unable to assess whether proper alignment would translate into long-term functional outcomes. Finally, a significantly higher rate of lateral retinaculum release was recorded in the conventional TKA group. However, both groups had similar patellar scores, and we are unable to state for certain whether CN-TKA would produce better benefits in this situation. We were also unable to provide data regarding rotational alignment of the femoral components. Further studies using CT to evaluate the rotational alignment are necessary to determine whether CN-TKA merits alternative use in conventional TKA.
CN-TKA is theoretically useful for arthritic knees with Ranawat type II valgus deformity where accurate restoration of joint line, proper alignment of limb and component positioning may be challenging because of associated osseous deformities and contracture of soft tissue. Our results demonstrated that CN-TKA can provide proper restoration of the joint line level, however there were no differences in clinical function, limb and component alignment, as well as survival of the prosthesis between the CN-TKA and conventional TKA groups after a mean follow-up of 6.2 years.
The authors wish to thank Miss Yu-Shuan Lin for recording radiographic and functional data and assistance in statistical analyses.
- Insall JN, Binazzi R, Soudry M, Mestriner LA: Total knee arthroplasty. Clin Orthop Relat Res. 1985, 192: 13-22.PubMedGoogle Scholar
- Taylor M, Barrett DS: Explicit finite element simulation of eccentric loading in total knee replacement. Clin Orthop Relat Res. 2003, 414: 162-View ArticlePubMedGoogle Scholar
- Ritter MA: The anatomical graduated component total knee replacement: a longterm evaluation with 20-year survival analysis. J Bone Joint Surg Br. 2009, 91-B: 745-10.1302/0301-620X.91B6.21854.View ArticleGoogle Scholar
- Rodricks DJ, Patil S, Pulido P, Colwell CW: Press-fit condylar design total knee arthroplasty: fourteen to seventeen-year follow-up. J Bone Joint Surg Am. 2007, 89: 89-95. 10.2106/JBJS.E.00492.View ArticlePubMedGoogle Scholar
- Kim YH, Park JW, Kim JS, Park SD: The relationship between the survival of total knee arthroplasty and postoperative coronal, sagittal and rotational alignment of knee prosthesis. Int Orthop. 2014, 38: 379-385. 10.1007/s00264-013-2097-9.View ArticlePubMedGoogle Scholar
- Mizuuchi H, Matsuda S, Miura H, Okazaki K, Akasaki Y, Iwamoto Y: The evaluation of post-operative alignment in total knee replacement using a CT-based navigation system. J Bone Joint Surg Br. 2008, 90: 1025-1031.View ArticleGoogle Scholar
- Decking R, Markmann Y, Fuchs J, Puhl W, Scharf HP: Leg axis after computer-navigated total knee arthroplasty: a prospective randomized trial comparing computer-navigated and manual implantation. J Arthroplasty. 2005, 20: 282-288. 10.1016/j.arth.2004.09.047.View ArticlePubMedGoogle Scholar
- Mullaji A, Kanna R, Marawar S, Kohli A, Sharma A: Comparison of limb and component alignment using computer-assisted navigation versus image intensifier-guided conventional total knee arthroplasty: a prospective, randomized, single-surgeon study of 467 knees. J Arthroplasty. 2007, 22: 953-959. 10.1016/j.arth.2007.04.030.View ArticlePubMedGoogle Scholar
- Weng YJ, Hsu RW, Hsu WH: Comparison of computer-assisted navigation and conventional instrumentation for bilateral total knee arthroplasty. J Arthroplasty. 2009, 24: 668-673. 10.1016/j.arth.2008.03.006.View ArticlePubMedGoogle Scholar
- Hsu WH, Hsu RW, Weng YJ: Effect of preoperative deformity on postoperative leg axis in total knee arthroplasty: a prospective randomized study. Knee Surg Sports Traumatol Arthrosc. 2010, 18: 1323-1327. 10.1007/s00167-010-1146-9.View ArticlePubMedGoogle Scholar
- Huang TW, Hsu WH, Peng KT, Hsu RW, Weng YJ, Shen WJ: Total Knee Arthroplasty with Use of Computer-Assisted Navigation Compared with Conventional Guiding Systems in the Same Patient: Radiographic Results in Asian Patients. J Bone Joint Surg Am. 2011, 93: 1197-1202.View ArticlePubMedGoogle Scholar
- Kuo LT, Huang TW, Peng KT, Hsu RW: Computer-assisted navigation for cruciate-retaining total knee arthroplasty in patients with advanced valgus arthritic knees. Formosan J Musculoskeletal Dis. 2013, 4: 99-104. 10.1016/j.fjmd.2013.05.003.View ArticleGoogle Scholar
- Kim YH, Park JW, Kim JS: Computer-navigated versus conventional total knee arthroplasty a prospective randomized trial. J Bone Joint Surg Am. 2012, 94: 2017-2024. 10.2106/JBJS.L.00142.View ArticlePubMedGoogle Scholar
- Elkus M, Ranawat CS, Rasquinha VJ, Babhulkar S, Rossi R, Ranawat AS: Total knee arthroplasty for severe valgus deformity. Five to fourteen-year follow-up. J Bone Joint Surg Am. 2004, 86: 2671-2676.PubMedGoogle Scholar
- Ranawat AS, Ranawat CS, Elkus M, Rasquinha VJ, Rossi R, Babhulkar S: Total knee arthroplasty for severe valgus deformity. J Bone Joint Surg Am. 2005, 87 (Suppl 1): 271-284.PubMedGoogle Scholar
- Favorito PJ, Mihalko WM, Krackow KA: Total knee arthroplasty in the valgus knee. J Am Acad Orthop Surg. 2002, 10: 16-24.View ArticlePubMedGoogle Scholar
- Koskinen E, Remes V, Paavolainen P, Harilainen A, Sandelin J, Tallroth K, Kettunen J, Ylinen P: Results of total knee replacement with a cruciate-retaining model for severe valgus deformity–a study of 48 patients followed for an average of 9 years. Knee. 2011, 18: 145-150. 10.1016/j.knee.2010.04.001.View ArticlePubMedGoogle Scholar
- Easley ME, Insall JN, Scuderi GR, Bullek DD: Primary constrained condylar knee arthroplasty for the arthritic valgus knee. Clin Orthop Relat Res. 2000, 380: 58-64.View ArticlePubMedGoogle Scholar
- Rossi R, Rosso F, Cottino U, Dettoni F, Bonasia DE, Bruzzone M: Total knee arthroplasty in the valgus knee. Int Orthop. 2014, 38: 273-83. 10.1007/s00264-013-2227-4.View ArticlePubMedGoogle Scholar
- Pearle AD, Kendoff D, Musahl V: Perspectives on computer-assisted orthopaedic surgery: movement toward quantitative orthopaedic surgery. J Bone Joint Surg Am. 2009, 91 (Suppl 1): 7-12.View ArticlePubMedGoogle Scholar
- Hadjicostas PT, Soucacos PN, Thielemann FW: Computer-assisted osteotomy of the lateral femoral condyle with non-constrained total knee replacement in severe valgus knees. J Bone Joint Surg Br. 2008, 90: 1441-1445.View ArticlePubMedGoogle Scholar
- Chou WY, Siu KK, Ko JY, Chen JM, Wang CJ, Wang FS, Wong T: Preoperative templating and computer-assisted total knee arthroplasty for arthritic valgus knee. J Arthroplasty. 2013, 28: 1781-1787. 10.1016/j.arth.2012.09.020.View ArticlePubMedGoogle Scholar
- Yau WP, Chiu KY, Tang WM, Ng TP: Coronal bowing of the femur and tibia in Chinese: its incidence and effects on total knee arthroplasty planning. J Orthop Surg (Hong Kong). 2007, 15: 32-36.Google Scholar
- Whiteside LA: Correction of ligament and bone defects in total arthroplasty of the severe valgus knee. Clin Orthop Relat Res. 1993, 288: 234-245.PubMedGoogle Scholar
- Scott RD, Chmell MJ: Balancing the posterior cruciate ligament during cruciate-retaining fixed and mobile-bearing total knee arthroplasty: description of the pull-out lift-off and slide-back tests. J Arthroplasty. 2008, 23: 605-608. 10.1016/j.arth.2007.11.018.View ArticlePubMedGoogle Scholar
- Hsu RWW, Himeno S, Coventry MB, Chao EYS: Normal axial alignment of the lower extremity and load-bearing distribution at the knee. Clin Orthop Relat Res. 1990, 255: 215-227.PubMedGoogle Scholar
- Laurin CA, Levesque HP, Dussault R, Labelle H, Peides JP: The abnormal lateral patellofemoral angle. J Bone Joint Surg Am. 1978, 60: 55-60.PubMedGoogle Scholar
- Laurin CA, Dussault R, Levesque HP: The tangential X-ray investigation of the patellofemoral joint: X-ray technique, diagnostic criteria and their interpretation. Clin Orthop Relat Res. 1979, 144: 16-26.PubMedGoogle Scholar
- Kawahara S, Okazaki K, Matsuda S, Nakahara H, Okamoto S, Iwamoto Y: Internal Rotation of Femoral Component Affects Functional Activities After TKA-Survey With The 2011 Knee Society Score. J Arthroplasty. In pressGoogle Scholar
- Ewald FC: The Knee Society total knee arthroplasty roentgenographic evaluation and scoring system. Clin Orthop Relat Res. 1989, 248: 9-12.PubMedGoogle Scholar
- Figgie HE, Goldberg VM, Heiple KG, Moller HS, Gordon NH: The influence of tibial-patellofemoral location on function of the knee in patients with the posterior stabilized condylar knee prosthesis. J Bone Joint Surg Am. 1986, 68: 1035-1040.PubMedGoogle Scholar
- Insall JN, Dorr LD, Scott RD, Scott WN: Rationale of the Knee Society clinical rating system. Clin Orthop Relat Res. 1989, 248: 13-14.PubMedGoogle Scholar
- Feller JA, Bartlett RJ, Lang DM: Patellar resurfacing versus retention in total knee arthroplasty. J Bone Joint Surg Br. 1996, 78: 226-228.PubMedGoogle Scholar
- Konig C, Sharenkov A, Matziolis G, Taylor WR, Perka C, Duda GN, Heller MO: Joint line elevation in revision TKA leads to increased patellofemoral contact forces. J Orthop Res. 2010, 28: 1-5.PubMedGoogle Scholar
- Ensini A, Catani F, Biasca N, Belvedere C, Giannini S, Leardini A: Joint line is well restored when navigation surgery is performed for total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2012, 20: 495-502. 10.1007/s00167-011-1558-1.View ArticlePubMedGoogle Scholar
- Anderson JA, Baldini A, MacDonald JH, Pellicci PM, Sculco TP: Primary constrained condylar knee arthroplasty without stem extensions for the valgus knee. Clin Orthop Relat Res. 2006, 442: 199-203. 10.1097/01.blo.0000185675.99696.29.View ArticlePubMedGoogle Scholar
- Verlinden C, Uvin P, Labey L, Luyckx JP, Bellemans J, Vandenneucker H: The influence of malrotation of the femoral component in total knee replacement on the mechanics of patellofemoral contact during gait: an in vitro biomechanical study. J Bone Joint Surg Br. 2010, 92: 737-742.View ArticlePubMedGoogle Scholar
- Chauhan SK, Scott RG, Breidahl W, Beaver RJ: Computer-assisted knee arthroplasty versus a conventional jig-based technique. A randomised, prospective trial. J Bone Joint Surg Br. 2004, 86: 372-10.1302/0301-620X.86B3.14643.View ArticlePubMedGoogle Scholar
- Stöckl B, Nogler M, Rosiek R, Fischer M, Krismer M, Kessler O: Navigation improves accuracy of rotational alignment in total knee arthroplasty. Clin Orthop Relat Res. 2004, 426: 180-186.View ArticlePubMedGoogle Scholar
- Siston RA, Cromie MJ, Gold GE, Goodman SB, Delp SL, Maloney WJ, Giori NJ: Averaging different alignment axes improves femoral rotational alignment in computer-navigated total knee arthroplasty. J Bone Joint Surg Am. 2008, 90: 2098-2104. 10.2106/JBJS.G.00996.View ArticlePubMedGoogle Scholar
- Matziolis G, Krocker D, Weiss U, Tohtz S, Perka C: A prospective, randomized study of computer-assisted and conventional total knee arthroplasty. Three-dimensional evaluation of implant alignment and rotation. J Bone Joint Surg Am. 2007, 89: 236-243. 10.2106/JBJS.F.00386.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/15/390/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.