- Case report
- Open Access
- Open Peer Review
Metal hypersensitivity in patient with posterior lumbar spine fusion: a case report and its literature review
© Shang et al.; licensee BioMed Central Ltd. 2014
- Received: 21 April 2014
- Accepted: 17 September 2014
- Published: 26 September 2014
Metal hypersensitivity, mostly documented in prosthesis implantation, is a rare complication after arthroplasty. Such cases become rarer and more difficult to diagnose when it comes to lumbar surgery.
We present the case of a 52-year-old female patient with reoccured low back pain and sciatica after posterior lumbar decompression and fusion (PLDF) for her lumbar disc herniation. The initial clinical and radiological examinations showed no pathologies. Further imaging and histopathological studies in later period revealed an aseptic loosening of the hardware and an aseptic inflammatory response which was diagnosed to be metal hypersensitivity. To our knowledge, few allergic cases in the matter of spinal fusion were reported so far.
Metal hypersensitivity after spinal fusion should be considered in patients with representation of postoperative back pain. And elaborate history taking would conduce a lot to it’s diagnose.
- Lumbar spine
- Decompression with fusion
- Aseptic loosening
- Prostheses and implants/adverse effects
Patients can be sensitive to metal debris released from the hardware for orthopedics treatment, and presenting with pain, swelling, inflammatory skin reactions, implant loosening, and fistula formation. Metal hypersensitivity, most documented among total joint arhtroplasty, is one of the rare complications after orthopedic procedures, with small number of cases after spine arthrodesis. The typical clinical presentation towards the high level of metal ions after surgery is persistent unexplained pain, or a development of unexplained pain after an initial pain-free interval, usually within the first 6 months after implantation. The level of metal ion in serum decrease rapidly with time after revision but still remained above normal levels for 4 years after surgery.
In patients representing late-onset persistent pain and trouble walking, after spinal fusion, an infection or incomplete surgery must be considered, but also the possibility of metal hypersensitivity. Once an infection and imcomplete surgery have been excluded, metal hypersensitivity is about to come into notice. As we know, all metals which are in contact with biologic systems are subject to corrosion. The released ions could activate the immune system by forming metal-protein complexes, considered as candidate antigens to elicite hypersensitivity responses.
In this report, we presented a case of 52-year-old female with metal hypersensitivity in a few months after Posterior Lumbar Decompression and Fusion (PLDF) due to lumbar intervertebral disc herniation, aiming to highlight the uncertainty in the diagnostic process and the significance of a complete history taking.
This report was made under the ethical approval of The Medical Ethic Committee of the First Affiliated Hospital of Dalian Medical University.
She gave a clear history of skin sensitivity to metal for many years before receiving PLDF and was unable to wear a metal watch or ring. After PLDF, she did not notice rashes or irritation over the low back area, or any other skin reaction, and other clinical evidence of infection.
Metal hypersensitivity as a complication after arthroplasty is rare, however, it is likely that cases involving implant-related metal sensitivity have been underreported because of the difficulty of diagnosis[5, 6]. Even so, prospective studies have shown a higher incidence of metal hypersensitivity in patients with implant failure.
Metal hypersensitivity has been associated with arthroplasty recipients mainly with metal on metal bearing surfaces. This could be attributed to the immunologic effects and /or cell toxicity mediated by exposure to wear debris[8–10]. It is hypothesized that metal particles are slowly released from the prosthetic bearing surfaces as a by-product of normal wear. And these particles are subject to corrosion resulting in producing high levels of ions potentially causing cell death. Wear debris from metal prosthetics are demonstrable in adjacent periprosthetic soft tissue as well in distant sites such as lymph nodes, liver, and spleen[11–13]. Adjacent tissue reactions to the debris have been given a variety of names: aseptic lymphocytic-vasculitis-associated lesions (ALVAL), pseudotumor, necrosis, adverse reaction to metal debris, and adverse local tissue response, with or without a clear underlying cause. But it’s generally believed that these wear particles, in conjunction with native proteins, form haptens that elicit a late onset type IV hypersensitivity in the local tissue[2, 3].
Most alloys used for orthopaedic implants rely on an oxidative layer to protect them from corrosion. Mechanical factors such as applied stress, fretting, and even micromotion will cause this protective layer to break down. Besides crevices formed during the manipulation also will defect the protective layer. The corrosive process accelerated, once the alloy was exposed to the chemical factors from body fluid through the imperfection of the oxidative layer. With corrosion, metal ions are released into the body. Spine fixators are static load-bearing devices subjected to micromotion and fretting at least until a successful fusion would have been achieved. However, even after a successful arthrodesis, there is continued load sharing between the implants and the fusion mass and this may lead to continued stress within the implants, potentially resulting in fretting corrosion. Therefore, spine implants can cause metal ion and debris release from fretting corrosion with elevating levels in body fluids, especially in the periprosthetic tissue. This has been reported by several studies about metal ion levels in spinal implants[14–16]. Moreover, titanium particulate debris at the level of a spinal arthrodesis could elicit a cytokine-mediated particulate-induced response favoring pro-inflammatory infiltrates and increased expression of intracellular tumor necrosis factor-alpha, increased osteoclastic activity, and cellular apoptosis in an animal model. In the clinical setting, the presence of titanium particulate, secondary to corrosion of spinal implants could serve as the impetus for both late-onset inflammatory-infectious complications and long-term osteolysis in the established posterolateral fusion mass.
Levels of metal ions measured in blood after spinal arthrodesis is comparable to that seen after total joint arthroplasty. Metal concentrations in serum was not seemed as a useful indicator of hardware loosening or implant failure, but may be associate with the systematic reaction, including urticaria, eczema, and pruritus. As to the topo-reaction around the periprosthetic tissue, metal deposition in the surrounding tissue seems to be attributable and can be a local destructive response leading to pain, eczematous reaction, ostoslysis, and loosening of the implants.
Patients presenting a late-onset postoperative pain with no clinical evidence of infection stand a good chance of metal hypersensitivity and a further evaluation should be performed. Basic physical examination, blood tests including ESR and CRP, radiographs, preoperative and intraoperative biopsy may be helpful to make a definite diagnosis.
There is no practical guide in the literature on how to differentiate between metal hypersensitivity and infection in a painful spinal fusion. Consequently, the evaluation largely depends on the process of the pain elimination. Routine laboratory data including the ESR and CRP are noted to have a high diagnostic accuracy towards infection. Studies also give confirmable sensitivities and specificities of various tests to diagnose infection. For example the combination of white cell sulphur colloid scan and Technetium Tc99m bone scan with a high accuracy in diagnosing infection was used in diagnosing metal hypersensitivity related to implants failure has also been reported.
Although culture of intraoperative specimens, including tissue samples and swabs is an effective method to identify infection, however, this could present false positives or false negatives due to cross contamination or previous antimicrobial therapy.
Biopsy study is reported to be the most accurate predictor of infection, featuring a predominant infiltration of neutrophils in periprosthetic tissues, while the characteristic histologic appearance of hypersensitivity reactions to metal prostheses is dominated by lymphoplasmacytic infiltration and droplike inclusions in the cytoplasm of macrophages. The appearance in a biopsy specimen may help the surgeon to reach a definite diagnosis. It has been recommended that in cases of suspected metal hypersensitivity, an arthroscopic biopsy followed by histological analysis should be performed.
Assessment of hypersensitivity has historically been conducted in vivo by skin patch testing and in vitro by leukocyte migration inhibition testing (LMIT). Patch testing has been used for many years with a greater frequency of positivity in patients with metal implants, but recommendations for routine testing are still controversial[3, 18, 22, 23]. The short duration of dermal contact in skin patch testing is different to the long term closed environment of the orthopaedic implant, and the relationship between dermal and deep implant sensitivity is yet unknown. Concerns also exist that patch testing could possibly induce hypersensitivity in a previously insensitive patient. In vitro tests like lymphpcyte transformation testing, LMIT appear more promising but are expensive and labor intensive. Besides, these tests are unavailable in many hospitals and have not been proven in the clinical setting.
The primary symptom in our case is delayed aggravating postoperative back pain, that is, there was a pain free/alleviation interval, which was consistent with most metal hypersensitivity cases. We note that the back pain before removing the hardware was severe and persistent without postural relief even when she stayed in bed. However after removal of the hardware, the pain alleviated greatly when stay in bed, not that much when she got out of bed. That is due to the instability of lumbar spine, resulted from endplates destruction and laminectomy performed during primary surgery. We also believe that the length of this pain free/ alleviation interval is negatively related to the integriy of protective layer on the hardware surface. Unlike severe cases, her pain was limited to the low back region, no other cutaneous or vascular symptoms were found over the whole body. This seems optimistic because a small amount of metal debris was produced in the short segment applied hardware and only localized in the surrounding area. Another prominent symptom, herein, is urinary retention occurred 8 months after the primary surgery. The urinary retention is consistent with Cauda Equina Syndrome (CES), which is a well-known complication for lumbar disc herniation. But the episode of CES after a complete decompressive procedure is quite unusual. In total disc replacement cases, Guyer RD et al. found the matellosis soft-tissue were responsible for CES. However, no abnormal tissue was found close enough to cause CES in our case.
Metal hypersensitivity, usually requiring complete removal or revision of the metal instruments, could be an unpleasant incident to both patients and surgeons, and is often neglected due to its low incidence and unpredictable occurance. Moreover, it also gives a diagnostic challenge because of the lack of practical diagnosis guide. Since it’s a well-recognized causative factor for reoccurrence of persistent preoperative pain after spinal fusion, precaution and diagnosis of this entity should always keep in mind. Although preoperative history-taking alone appears to be insufficient for identifying patients with metal sensitivity, it is still strongly recommended to be taken thoroughly, because this could provide a strong clue towards the delayed postoperative pain in time, and therewith, a timely customized examination and treatment could be conducted to benefit the patient tremendously. In this case, the patient sustained low back pain for another 5 months because of the difficulty in diagnosis, but the related signs and symptoms resolved shortly after removal of the metal instruments like other cases reported[25–27].
Written informed consent was obtained from the patient for publication of this Case report and any accompanying images. A copy of the written consent is available for review by the Editor of this journal.
We are thankful to Bin Wang for editing the manuscript. This study was funded by National Natural Science Foundation, the People’s Republic of China (81271719). The device (s)/drug (s) is/are FDA-approved or approved by corresponding national agency for this indication. No benefits in any form have been or will be received from a commercial party related directly or indirectly to the subject of this manuscript.
- Willert HG, Buchhorn GH, Fayyazi A, Flury R, Windler M, Koster G, Lohmann CH: Metal-on-metal bearings and hypersensitivity in patients with artificial hip joints. A clinical and histomorphological study. J Bone Joint Surg Am. 2005, 87 (1): 28-36. 10.2106/JBJS.A.02039pp.View ArticlePubMedGoogle Scholar
- Jacobs JJ, Hallab NJ, Skipor AK, Urban RM: Metal degradation products: a cause for concern in metal-metal bearings?. Clin Orthop Relat Res. 2003, 417: 139-147.PubMedGoogle Scholar
- Hallab N, Merritt K, Jacobs JJ: Metal sensitivity in patients with Orthopaedic implants. J Bone Joint Surg Am. 2001, 83-A (3): 428-436.PubMedGoogle Scholar
- Thyssen JP, Jakobsen SS, Engkilde K, Johansen JD, Soballe K, Menne T: The association between metal allergy, total hip arthroplasty, and revision. Acta Orthop. 2009, 80 (6): 646-652. 10.3109/17453670903487008.View ArticlePubMedPubMed CentralGoogle Scholar
- Schuh A, Thomas P, Kachler W, Goske J, Wagner L, Holzwarth U, Forst R: [Allergic potential of titanium implants]. Orthopade. 2005, 34 (4): 327-328. 10.1007/s00132-005-0764-2. 330–323View ArticlePubMedGoogle Scholar
- Watts RA, Lane SE, Bentham G, Scott DG: Epidemiology of systemic vasculitis: a ten-year study in the United Kingdom. Arthritis Rheum. 2000, 43 (2): 414-419. 10.1002/1529-0131(200002)43:2<414::AID-ANR23>3.0.CO;2-0.View ArticlePubMedGoogle Scholar
- Frigerio E, Pigatto PD, Guzzi G, Altomare G: Metal sensitivity in patients with orthopaedic implants: a prospective study. Contact Dermatitis. 2011, 64 (5): 273-279. 10.1111/j.1600-0536.2011.01886.x.View ArticlePubMedGoogle Scholar
- Hallab NJ, Anderson S, Caicedo M, Skipor A, Campbell P, Jacobs JJ: Immune responses correlate with serum-metal in metal-on-metal hip arthroplasty. J Arthroplasty. 2004, 19 (8 Suppl 3): 88-93.View ArticlePubMedGoogle Scholar
- Kwon YM, Thomas P, Summer B, Pandit H, Taylor A, Beard D, Murray DW, Gill HS: Lymphocyte proliferation responses in patients with pseudotumors following metal-on-metal hip resurfacing arthroplasty. J Orthop Res. 2010, 28 (4): 444-450.PubMedGoogle Scholar
- Savarino L, Padovani G, Ferretti M, Greco M, Cenni E, Perrone G, Greco F, Baldini N, Giunti A: Serum ion levels after ceramic-on-ceramic and metal-on-metal total hip arthroplasty: 8-year minimum follow-up. J Orthop Res. 2008, 26 (12): 1569-1576. 10.1002/jor.20701.View ArticlePubMedGoogle Scholar
- Hicks DG, Judkins AR, Sickel JZ, Rosier RN, Puzas JE, O’Keefe RJ: Granular histiocytosis of pelvic lymph nodes following total hip arthroplasty. The presence of wear debris, cytokine production, and immunologically activated macrophages. J Bone Joint Surg Am. 1996, 78 (4): 482-496.PubMedGoogle Scholar
- Munichor M, Cohen H, Volpin G, Kerner H, Iancu TC: Chromium-induced lymph node histiocytic proliferation after hip replacement. A case report. Acta Cytol. 2003, 47 (2): 270-274. 10.1159/000326515.View ArticlePubMedGoogle Scholar
- Urban RM, Jacobs JJ, Tomlinson MJ, Gavrilovic J, Black J, Peoc'h M: Dissemination of wear particles to the liver, spleen, and abdominal lymph nodes of patients with hip or knee replacement. J Bone Joint Surg Am. 2000, 82 (4): 457-476.PubMedGoogle Scholar
- Kim YJ, Kassab F, Berven SH, Zurakowski D, Hresko MT, Emans JB, Kasser JR: Serum levels of nickel and chromium after instrumented posterior spinal arthrodesis. Spine. 2005, 30 (8): 923-926. 10.1097/01.brs.0000158872.42802.be.View ArticlePubMedGoogle Scholar
- del Rio J, Beguiristain J, Duart J: Metal levels in corrosion of spinal implants. Eur Spine J. 2007, 16 (7): 1055-1061. 10.1007/s00586-007-0311-4.View ArticlePubMedPubMed CentralGoogle Scholar
- Kasai Y, Iida R, Uchida A: Metal concentrations in the serum and hair of patients with titanium alloy spinal implants. Spine. 2003, 28 (12): 1320-1326.PubMedGoogle Scholar
- Cunningham BW, Orbegoso CM, Dmitriev AE, Hallab NJ, Sefter JC, McAfee PC: The effect of titanium particulate on development and maintenance of a posterolateral spinal arthrodesis: an in vivo rabbit model. Spine. 2002, 27 (18): 1971-1981. 10.1097/00007632-200209150-00004.View ArticlePubMedGoogle Scholar
- Spangehl MJ, Masri BA, O'Connell JX, Duncan CP: Prospective analysis of preoperative and intraoperative investigations for the diagnosis of infection at the sites of two hundred and two revision total hip arthroplasties. J Bone Joint Surg Am. 1999, 81 (5): 672-683.PubMedGoogle Scholar
- Biant LC, Bruce WJ, van der Wall H, Walsh WR: Infection or allergy in the painful metal-on-metal total hip arthroplasty?. J Arthroplasty. 2010, 25 (2): 334-e311-336View ArticlePubMedGoogle Scholar
- Davies AP, Willert HG, Campbell PA, Learmonth ID, Case CP: An unusual lymphocytic perivascular infiltration in tissues around contemporary metal-on-metal joint replacements. J Bone Joint Surg Am. 2005, 87 (1): 18-27. 10.2106/JBJS.C.00949.View ArticlePubMedGoogle Scholar
- Thomas P, Summer B, Sander CA, Przybilla B, Thomas M, Naumann T: Intolerance of osteosynthesis material: evidence of dichromate contact allergy with concomitant oligoclonal T-cell infiltrate and TH1-type cytokine expression in the peri-implantar tissue. Allergy. 2000, 55 (10): 969-972. 10.1034/j.1398-9995.2000.00619.x.View ArticlePubMedGoogle Scholar
- Granchi D, Cenni E, Tigani D, Trisolino G, Baldini N, Giunti A: Sensitivity to implant materials in patients with total knee arthroplasties. Biomaterials. 2008, 29 (10): 1494-1500. 10.1016/j.biomaterials.2007.11.038.View ArticlePubMedGoogle Scholar
- Lohmann CH, Nuechtern JV, Willert HG, Junk-Jantsch S, Ruether W, Pflueger G: Hypersensitivity reactions in total hip arthroplasty. Orthopedics. 2007, 30 (9): 760-761.PubMedGoogle Scholar
- Guyer RD, Shellock J, MacLennan B, Hanscom D, Knight RQ, McCombe P, Jacobs JJ, Urban RM, Bradford D, Ohnmeiss DD: Early failure of metal-on-metal artificial disc prostheses associated with lymphocytic reaction: diagnosis and treatment experience in four cases. Spine. 2011, 36 (7): E492-497. 10.1097/BRS.0b013e31820ea9a2.View ArticlePubMedGoogle Scholar
- Lim HP, Lee KM, Koh YI, Park SW: Allergic contact stomatitis caused by a titanium nitride-coated implant abutment: a clinical report. J Prosthet Dent. 2012, 108 (4): 209-213. 10.1016/S0022-3913(12)60163-2.View ArticlePubMedGoogle Scholar
- Post ZD, Orozco FR, Ong AC: Metal sensitivity after TKA presenting with systemic dermatitis and hair loss. Orthopedics. 2013, 36 (4): e525-528. 10.3928/01477447-20130327-35.View ArticlePubMedGoogle Scholar
- Zairi F, Remacle JM, Allaoui M, Assaker R: Delayed hypersensitivity reaction caused by metal-on-metal total disc replacement. J Neurosurg Spine. 2013, 19 (3): 389-391. 10.3171/2013.6.SPINE121010.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/15/314/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.