- Research article
- Open Access
- Open Peer Review
Biomechanical comparison of different combinations of hook and screw in one spine motion unit - an experiment in porcine model
© Tai et al.; licensee BioMed Central Ltd. 2014
- Received: 29 October 2013
- Accepted: 30 May 2014
- Published: 9 June 2014
The biomechanical performance of the hooks and screws in spinal posterior instrumentation is not well-characterized. Screw-bone interface failure at the uppermost and lowermost vertebrae is not uncommon. Some have advocated for the use of supplement hooks to prevent screw loosening. However, studies describing methods for combined hook and screw systems that fully address the benefits of these systems are lacking. Thus, the choice of which implant to use in a given case is often based solely on a surgeon’s experience instead of on the biomechanical features and advantages of each device.
We conducted a biomechanical comparison of devices instrumented with different combinations of hooks and screws. Thirty-six fresh low thoracic porcine spines were assigned to three groups (12 per group) according to the configuration used for of fixation: (1) pedicle screw; (2) lamina hook and (3) combination of pedicle screw and lamina hook. Axial pullout tests backward on transverse plane in the direction normal to the rods were performed using a material testing machine and a specially designed grip with self-aligned function.
The pullout force for the pedicle screws group was significantly greater than for the hooks and the combination (p < 0.05). However, no significant difference was found between the hooks and the combination (p > 0.05).
Pedicle screws achieve the maximal pullout strength for spinal posterior instrumentation.
- Pedicle screw
- Lamina hook
- Biomechanical study
- Porcine model
The implantation of bone screws into the pedicle was reported in the early 1960s. Pedicle screw fixation is regarded as a significant improvement over conventional methods for stabilizing the spine, which use wires, hooks or both. Pedicle screws provide three-column fixation, allowing shorter segment of instrumentation to be used to restore sagittal alignment. Despite these advantages, pedicle screws have drawbacks. For example, pedicle screws can bend or break particularly in the caudal segments[2, 3]. There have been no clear biomechanical comparisons between hooks and screws. Although pedicle screw anchors offer three-dimensional correction, which is beneficial compared with hooks, pullout tests have shown that the purchase power of a single pedicle screw is greater than a single hook in healthy bone, but less powerful than a single hook in osteoporotic bone. In osteoporotic vertebrae, the bone quality of the lamina is better than the bone quality in the vertebral body. To provide better curve correction and maintain a balanced spine, the anchoring of the implant to the bone should be very strong and resistant to loading.
Harrington’s rod for the treatment of scoliosis was implemented in 1962 as the first implant for thoracic spine surgery[5, 6]. This device is a non-segmental hook and rod system. The first attempt to provide segmental fixation was Luque’s sublaminar wire[7, 8]. Sublaminar wires provide stable fixation for each vertebral segment but do not reduce coronal spinal deformities. The development of the multiple segmental hook and rod system by Cotrel and Dubousset was a milestone in scoliosis surgery[9, 10]. There are two types of hooks that used in this instrumentation: pedicle and lamina. Pedicle hooks provide stronger anchoring power and do not enter the neural foramen. However, they can only translate the correction force upward (cranially) in the thoracic spine. Lamina hooks can provide upward and downward (caudal) forces in the thoracic and lumbar spine. However, they must enter the neural canal to anchor to the lamina. Suk began using thoracic pedicle screws for scoliosis correction in 1991 and has since reported the highest curve correctivity of any system. In spite of this encouraging result, the use of thoracic screws for scoliosis has not been widely accepted. Screw-bone interface failure at the uppermost and lowermost vertebrae is not uncommon. Some have advocated for the use of supplement hooks to prevent screw loosening[11–13]. However, the benefits of combined hook and screw methods are not clear. Thus, the choice of which implant to use in a given case is often been based solely on a surgeon’s experience rater than on the biomechanical features and advantages of each device. The aim of this study was to measure the pullout strength of pedicle screws versus lamina hooks and combinations of screws and lamina hooks in porcine thoracic spine to simulate the surgical correction techniques used to treat scoliosis.
Because of the relatively early application of hooks for treating scoliosis, most surgeons are familiar with this technique. There are standard size hooks available for adolescent patients and miniature hooks available for small children (body weight < 30 kg). However, there are potential disadvantages associated with hook systems[12–14]. First, the tip of the lamina hook has to enter the neural canal, increasing the potential for injury to the spinal cord. Second, hooks engage only the posterior elements instead of engaging both the anterior and medial columns. Pedicle screws are more commonly used in the lumbar spine than in the thoracic spine. Placement of thoracic pedicle screws is an exact and demanding technique[15, 16]. Surrounding the thoracic spine, there are many vital structures, including the trachea, aorta, segmental vessels, anterior vena cava, lung and spinal cord. Dr. Suk reported the use of thoracic pedicle screws to treat thoracic scoliosis in 1991. That report demonstrated improved radiographic results in patients treated with screw versus hook.
Spinal biomechanics have been studied in several ways. Using a long segment construct to study the biomechanics of implants is time consuming and impractical. To conquer this problem, it is reasonable to use a motion unit to study spinal biomechanics. The previous literature has examined the pullout behavior of pedicles, which were loaded by tension during flexion[17, 18]. This situation can be exacerbated in long instrumentation constructs, where large axial pullout forces are generated due to cantilever bending at the proximal levels during forward flexion[17, 18]. Our results indicated that the pullout force for pedicle screws was significantly greater than for hook and combinations of screws and hooks (p < 0.05). No significant difference was observed between the hook and combination groups (p > 0.05). Several studies have described the mechanical roles of pedicle screws and lamina hooks in spinal posterior instrumentation[19–21]. However, the anchoring strength for combinations of screws and hooks was not considered. Liljenqvist et al. investigated the axial pullout strength of pedicle screws versus laminar hooks in the thoracic spine with respect to the surgical correction techniques used to treat scoliosis. Their results indicated that the average pullout strength of pedicle screws was significant higher than for laminar hooks. These observations were consistent with our results. Murakami et al. compared fixation in the thoracolumbar spine using pedicle screws combined with infra-lamina hooks (at both the cranial and caudal ends of the same vertebrae) with fixation using pedicle screws alone. They concluded that a pedicle screw combined with an infra-lamina hook offers significantly greater strength compared to a pedicle screw alone. In osteoporotic vertebrae, the bone quality of the lamina is better than the bone quality in the vertebral body. Therefore, the bone mineral density might be an important consideration in the selection of screws or hooks. Hackenberg et. al studied the fixation strength of pedicle screws versus pedicle and laminar hooks in the thoracic spine in relation to the bone mineral density. They concluded that pedicle screws may be beneficial for thoracic spine instrumentation because they offer significantly more resistant to axial and tangential loads than pedicle and lamina hooks on the condition that bone mineral density is higher than 100 mg hydroxyapatite/mL. However, the difference is not significant if bone mineral density is lower than 100 mg hydroxyapatite/mL.
Our study is an in vitro analysis of specimens prepared in a laboratory environment, which does not necessarily represent clinical circumstances. There are several limitations to our study. First, we used porcine spines instead of human cadaveric spines. Although the physiological structures are somewhat different from those of human cadaveric spines, animal spines are the most convenient choice for performing the experiments when human cadaveric spines cannot be accessed. Numerous studies[22, 23] have been performed to evaluate the biomechanical behavior of spinal columns. These studies typically use porcine spines as models for the human spine. Smit’s report showed similarities in the mechanics of quadruped and human spines. He concluded that a quadruped can serve as a valuable model for studying the human spine in spite of its horizontal position. Second, our work is limited to static loading (pullout) and does not consider other physiological loads. In actual physiological situations, the screw/bone and hook/bone interfaces are subjected to complex dynamic multi-directional loads. Consequently, the choice between a pedicle screw and a hook should not be determined solely based on the pullout strength. In practice, fixation devices with the ability to correct deformity, restore sagittal balance, prevent deterioration and influence adjacent segments should all be considered. Although the loading mode for pullout does not necessarily represent the actual physiological loading condition and may have great impact on the clinical relevance, the specimens used in this study were all prepared and tested in a uniform and reproducible manner. Thus, we believe that these results will provide useful information for orthopedic surgeons to assess spinal posterior instrumentation systems.
The pedicle screw instrumentation achieves the maximal pullout strength for spinal posterior instrumentation. Based on this comparative study, surgeons should be better informed for selecting the optimal fixation method for the thoraco-lumbar spine before surgery.
The authors would like to thank National Science Council of Taiwan and Chang Gung Memorial Hospital for financial support (CMRPD280021, CMRPD280022).
- Roy-Camille R, Saillant G, Mazel C: Internal fixation of the lumbar spine with pedicle screw plating. Clin Orthop Relat Res. 1986, 203: 7-17.PubMedGoogle Scholar
- Hicks JM, Singla A, Shen FH, Arlet V: Complications of pedicle screw fixation in scoliosis surgery: a systematic review. Spine. 2010, 35 (11): E465-70. 10.1097/BRS.0b013e3181d1021a.View ArticlePubMedGoogle Scholar
- McLain RF, Sparling E, Benson DR: Early failure of short-segment pedicle instrumentation for thoracolumbar fractures. A preliminary report. J Bone Joint Surg Am. 1993, 75 (2): 162-7.PubMedGoogle Scholar
- Hasegawa K, Takahashi HE, Uchiyama S, Hirano T, Hara T, Washio T, Sugiura T, Youkaichiya M, Ikeda M: An experimental study of a combination method using a pedicle screw and laminar hook for the osteoporotic spine. Spine. 1997, 22 (9): 958-62. 10.1097/00007632-199705010-00004.View ArticlePubMedGoogle Scholar
- Harrington PR: Correction and internal fixation by spine instrumentation. J Bone Joint Surg Am. 1962, 44: 591-610.PubMedGoogle Scholar
- Hitchon PW: Harrington distraction rods for thoracic and lumbar fractures. Techniques in Spinal Fusion and Stabilization. Edited by: Hitchon PW, Traynelis VC, Rengachary SS. 1995, New York: Thieme, 204-208.Google Scholar
- Luque ER: Segmental spine instrumentation: A method of rigid internal fixation of the spine to induce arthrodesis. Orthop Trans. 1980, 4: 391-Google Scholar
- Hitchon PW, Follett KA: Luque instrumentation for the thoracic and lumbar spine. Techniques in Spinal Fusion and Stabilization. Edited by: Hitchon PW, Traynelis VC, Rengachary SS. 1995, New York: Thieme, 198-203.Google Scholar
- Cotrel Y: What new Cotrel-Dubousset instrumentation is used in the treatment of idiopathic scoliosis. Acta Orthop Belg. 1992, 58 (Suppl 1): 111-4.PubMedGoogle Scholar
- Bennett GJ: Cotrel-Dubousset instrumentation for thoracolumbar instability. Techniques in Spinal Fusion and Stabilization. Edited by: Hitchon PW, Traynelis VC, Rengachary SS. 1995, New York: Thieme, 209-217.Google Scholar
- Suk SI, Lee CK, Kim WJ, Chung YJ, Park YB: Segmental pedicle screw fixation in the treatment of thoracic idiopathic scoliosis. Spine. 1995, 20 (12): 1399-1405. 10.1097/00007632-199506020-00012.View ArticlePubMedGoogle Scholar
- Benzel EC, Baldwin NG, Ball PA: Texas Scottish Rite Hospital Hook-Rod spinal fixation. Techniques in Spinal Fusion and Stabilization. Edited by: Hitchon PW, Traynelis VC, Rengachary SS. 1995, New York: Thieme, 229-239.Google Scholar
- Margulies JY, Casar RS, Caruso SA: The mechanical role of laminar hook protection of pedicle screws at the caudal end vertebra. Eur Spine J. 1997, 6 (4): 245-248. 10.1007/BF01322446.View ArticlePubMedPubMed CentralGoogle Scholar
- Grisafi FN, Emery SE: Migration of laminar hook causing thoracic myelopathy eight years post scoliosis surgery: a case report. Spine. 2010, 35 (6): E228-30. 10.1097/BRS.0b013e3181bf20a4.View ArticlePubMedGoogle Scholar
- Ebelke DK, Asher MA, Neff JR, Kraker DP: Survivorship analysis of VSP spine instrumentation in the treatment of thoracolumbar and lumbar burst fractures. Spine. 1991, 16 (8 Suppl): S428-32.PubMedGoogle Scholar
- Zindrick MR, Wiltse LL, Widell EH, Thomas JC, Holland WR, Field BT, Spencer CW: A biomechanical study of interpeduncular screw fixation in the lumbosacral spine. Clin Orthop. 1986, 203: 99-112.PubMedGoogle Scholar
- Ashman RB, Galpin RD, Corin JD, Johnston CE: Biomechanical analysis of pedicle screw instrumentation systems in a corpectomy model. Spine. 1989, 14 (12): 1398-405. 10.1097/00007632-198912000-00019.View ArticlePubMedGoogle Scholar
- Law M, Tencer AF, Anderson PA: Caudo-cephalad loading of pedicle screws: mechanisms of loosening and methods of augmentation. Spine. 1993, 18 (16): 2438-43. 10.1097/00007632-199312000-00012.View ArticlePubMedGoogle Scholar
- Liljenqvist U, Hackenberg L, Link T, Halm H: Pullout strength of pedicle screws versus pedicle and laminar hooks in the thoracic spine. Acta Orthop Belg. 2001, 67 (2): 157-63.PubMedGoogle Scholar
- Murakami H, Tsai KJ, Attallah-Wasif ES, Yamazaki K, Shimamura T, Hutton WC: A biomechanical assessment of infra-laminar hooks as an alternative to supra-laminar hooks in thoracolumbar fixation. Spine. 2006, 31 (9): 967-71. 10.1097/01.brs.0000214932.49027.6a.View ArticlePubMedGoogle Scholar
- Hackenberg L, Link T, Liljenqvist U: Axial and tangential fixation strength of pedicle screws versus hooks in the thoracic spine in relation to bone mineral density. Spine. 2002, 27 (9): 937-42. 10.1097/00007632-200205010-00010.View ArticlePubMedGoogle Scholar
- Kaigle A, Ekstrom L, Holm S, Rostedt M, Hansson T: In vivo dynamic stiffness of the porcine lumbar spine exposed to cyclic loading: influence of load and degeneration. J Spinal Disord. 1998, 11 (1): 65-70.View ArticlePubMedGoogle Scholar
- Smit TH: The use of a quadruped as an in vivo model for the study of the spine – biomechanical considerations. Eur Spine J. 2002, 11 (2): 137-144. 10.1007/s005860100346.View ArticlePubMedPubMed CentralGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/15/197/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.