- Research article
- Open Access
- Open Peer Review
Ultrasonic reflection coefficient and surface roughness index of OA articular cartilage: relation to pathological assessment
© Niu et al; licensee BioMed Central Ltd. 2012
- Received: 13 October 2011
- Accepted: 10 March 2012
- Published: 10 March 2012
Early diagnosis of Osteoarthritis (OA) is essential for preventing further cartilage destruction and decreasing severe complications. The aims of this study are to explore the relationship between OA pathological grades and quantitative acoustic parameters and to provide more objective criteria for ultrasonic microscopic evaluation of the OA cartilage.
Articular cartilage samples were prepared from rabbit knees and scanned using ultrasound biomicroscopy (UBM). Three quantitative parameters, including the roughness index of the cartilage surface (URI), the reflection coefficients from the cartilage surface (R) and from the cartilage-bone interface (Rbone) were extracted. The osteoarthritis grades of these cartilage samples were qualitatively assessed by histology according to the grading standards of International Osteoarthritis Institute (OARSI). The relationship between these quantitative parameters and the osteoarthritis grades was explored.
The results showed that URI increased with the OA grade. URI of the normal cartilage samples was significantly lower than the one of the OA cartilage samples. There was no significant difference in URI between the grade 1 cartilage samples and the grade 2 cartilage samples. The reflection coefficient of the cartilage surface reduced significantly with the development of OA (p < 0.05), while the reflection coefficient of the cartilage-bone interface increased with the increase of grade.
High frequency ultrasound measurements can reflect the changes in the surface roughness index and the ultrasound reflection coefficients of the cartilage samples with different OA grades. This study may provide useful information for the quantitative ultrasonic diagnosis of early OA.
- High-frequency ultrasound
- Ultrasound roughness index
- Ultrasound reflection coefficient
Osteoarthritis (OA) is a chronic joint disease that occurs with the progressive degeneration of articular cartilage followed by secondary bone hyperplasia. Clinical symptoms of OA include joint pain, function loss and joint deformity [1, 2]. Early diagnosis of OA is essential for the timely treatment and prevention against advanced complications .
Techniques such as arthroscopy, X-ray computed tomography (CT) and magnetic resonance imaging (MRI) have been used in studies on articular cartilage and OA assessment [4–10]. Arthroscopy is usually used to qualitatively observe the visual changes in the superficial layer of articular cartilage. However, the degenerative changes in the deeper layers of the cartilage tissue are revealed by using this method with difficulty. Due to the principle of X-ray imaging, CT is effective in revealing bone degeneration but insensitive to soft tissue without contrast agents. MRI, which is considered as the most promising diagnostic technique, can noninvasively detect the cartilage surface contour, tissue compositions, and collagen orientation. However, clinical MRI imaging remains expensive and its relative long time of imaging and low resolution is limited to cartilage fine structure.
As a cheap, easy-to -use testing method and experimental means, ultrasonic measurement technique has been paid much attention in the research for the early diagnosis of OA in recent years. Many researchers measured the deformation of the cartilage tissue during the compression or swelling of articular cartilage using high-frequency ultrasound and input the experimental data to the single-phase, biphasic, and triphasic models to investigate the altered mechanical properties of the degraded cartilage [11–17]. Other researchers applied high-frequency ultrasound to explore the relation between the acoustic parameters (such as velocity, attenuation, etc.) and the composition changes in the progress of the cartilage degradation [18–20]. Previous studies have shown the potential of high frequency ultrasound to detect the cartilage degeneration.
Clinically, the severity of OA is usually graded in accordance with the morphological changes of the cartilage surface (such as degree of roughness and fibrillation) and the histopathological assessment of the tissue sections [15, 21]. This method could provide more accurate grading with the observation of the changes in the cartilage surface and the inside tissue. Therefore, histopathological score of OA has been accepted as a "gold standard" for assessment of cartilage lesions in OA and moreover for validation of other assessment methods such us arthroscopy, ultrasound and MRI.
Ultrasound biomicroscopy (UBM) is a high-resolution ultrasound imaging system that can be used to observe the changes of the internal fine structure of the soft tissue. Previous results showed that there was a significant correlation between the UBM image and the pathological image of arthritis cartilage with different grades . Unfortunately, no quantitative indicators were given in that study. As mentioned above, the morphological changes on the cartilage surface and the composition changes of articular cartilage are important bases for clinical grading of arthritis. Meanwhile, the roughness index can describe the morphological changes of the cartilage surface and the alterations in acoustic characteristics follow the composition changes of articular cartilage. One of the most intuitive alterations in acoustic characteristics is the changes in the reflection coefficient. Therefore, this study employed the UBM technique and calculated the roughness index of cartilage surface and the reflection coefficient of the cartilage tissue. The aims of this study are to explore the relationship between OA pathological grades and quantitative acoustic parameters and to provide more objective criteria for ultrasonic microscopic evaluation of the OA cartilage.
Eighteen normal adult New Zealand white female rabbits weighing 3.5 to 4.5 kg (mean, 4.1 ± 0.3 kg) were used in this study. Radiographs of both femorotibial joints were taken and evaluated by two orthopedists to exclude animals with joint pathology. Six rabbits were treated as control. To induce OA, surgical transection of the anterior cruciate ligament (ACL) in the left femorotibial joint was performed under general anesthesia in twelve rabbits. Routine skin incision closure was performed. Antibiotics (penicillin 20,000 IU) were injected intramuscularly twice a day preoperatively and for 2 days postoperatively in the operated animals. Following surgery, free movement was allowed in separate cages for the duration of the experimental period. Experiments on the rabbits were approved by our institutional animal care and use committee and performed under the guidelines of the National Institutes of Health for the care of laboratory animals.
At 2, 4, and 6 weeks after surgery, six animals were euthanized, including two control rabbits and four rabbits with ACL transection. Each of the left knees was dissected and sectioned with a band saw to obtain the articular cartilage samples with pathological characteristics. 72 cartilage specimens were collected, including 18 medial femoralcondyles, 18 lateral femoral condyles, 18 medial tibial plateaus, and 18 lateral tibial plateaus. All the specimens were wrapped in wet gauze soaked with saline buffer and stored at -20°C until the ultrasound examination.
UBM imaging was performed on these cartilage surfaces using a Vevo770 (VisualSonics, Toronto, ON, Canada) high-frequency UBM system. A transducer (RMV708) with a nominal central frequency of 55 MHz was used. This transducer has a focal length of 4.5 mm and an axial resolution of 30 μm.
According to the grading standards of International Osteoarthritis Institute (OARSI), OA is graded as follows: grade 1 = uneven surface that can demonstrate superficial fibrillation; grade 2 = surface discontinuity that may be accompanied by cell proliferation, increased or decreased matrix staining in the mid zone; grade 3 = vertical fissures extending into the mid zone or erosion; grade 4 = denudation (the unmineralized hyaline cartilage is completely eroded); grade 5 = deformation. OA develops with three stages: early stage (grade 1-2), intermediate stage (grade 3) and advanced stage (grade 4-5) [15, 21].
Cartilage samples used for pathological observation were fixed in 10% neutral buffered formalin, decalcified with 14% ethylenediamine tetra-acetic acid, dehydrated through graded alcohols, cleared with toluene, and embedded in paraffin. Careful attention was paid to make the histologically assessed site consistent with the ultrasound-scanned site. Six-micrometer sections of articular cartilage were cut from the proximal part to the ultrasound-scanned site. The sections were stained with toluidine blue. Then the sections were observed and graded by three board-certified pathologists blinded with the purpose of the whole study.
In this study, three parameters are extracted to quantitatively describe the acoustic properties of articular cartilage, i.e. ultrasound roughness index (URI) of the cartilage surface, reflection coefficient (R) of the cartilage surface, reflection coefficient (Rbone) of the cartilage-bone interface. URI is used to describe the microtopography of the cartilage surface. R and Rbone are used to describe characteristics of cartilage tissue .
where m is the number of the scan lines in the 4 mm sampling length. d i is the distance between the transducer and the solution- cartilage interface in the scan line i, and 〈d〉 is the average distance between the transducer and the surface.
where A si and A bi are the peak-to-peak amplitude of the ultrasound RF signals reflected from the cartilage surface and the cartilage-bone interface, respectively, in the scan line i. m is the number of scan lines. A iref is the reference peak-to-peak amplitude of the echoes reflected from the solution-air interface with the same distance as A si . In the parameter calculation, the ultrasound speed in the solution is 1500 m/s, and the average ultrasound speed in the cartilage tissue equals 1600 m/s . The analysis of ultrasound echo signals and the extraction of the ultrasound parameters were carried out by a custom-designed Matlab program (MATLAB, Version 2009, The Math-Works, USA).
Statistical analyses were conducted with SPSS software (Version 17, SPSS Inc., USA). All values in the text are presented as mean ± standard deviation (SD). All specimens were evaluated to the normal, grade 1, grade 2, grade 3, grade 4, grade 5 groups. The analysis of variance in URI, R, Rbone between grading groups were performed using One-way ANOVA and LSD post-hoc tests. P < 0.05 was taken as statistically significant.
OARSI grades of articular cartilage samples and ultrasound parameters
Normal (N = 29)
10.36 ± 5.82
9.05 ± 1.78
19.314 ± 5.46
grade 1 (N = 14)
69.55 ± 27.43
5.22 ± 2.13
19.81 ± 7.33
grade 2 (N = 23)
76.86 ± 39.20
3.90 ± 2.07
21.87 ± 8.82
grade 3 (N = 6)
193.27 ± 75.06.
1.08 ± 0.85
30.33 ± 6.19
Figure 4b shows the reflection coefficient of the cartilage surface of normal and OA cartilage samples, indicating that the reflection coefficient reduced significantly with the development of OA (p < 0.05). There were significant differences between normal and OA cartilage (p < 0.05), and significant intergroup differences among the samples with different grades (p < 0.05).
However, the reflection coefficient of cartilage-bone interfaces showed a gradual increase with the increase of grade (Figure 4c). Rbone of the grade 1 cartilage samples is close to that of normal cartilage. There were no significant difference between the grade 1 cartilage and the grade 2 cartilage (p > 0.05). Both groups of the grade 2 cartilage samples and the grade 3 cartilage samples had a significant increase in Rbone in comparison with normal cartilage (p < 0.05).
In this study, the reflection coefficients of the cartilage tissue and the roughness index of the cartilage surface were quantitatively described using the ultrasound microscopic imaging, and the relationship between the quantitative acoustic parameters and the severity of OA cartilage qualitatively assessed by histology was explored.
The results of our histological assessment indicate that the pathological symptoms in OA cartilage become more obvious and serious with the increase of OA grade. The cartilage surface became more uneven, the fibrillation of the superficial tissue more serious, and toluidine blue staining faded or totally disappeared from the surface to the deep layers as OA grade increased. The cartilage thickness decreased in the grade 3 cartilage samples. Previous studies reported consistent results that the proteoglycan content of articular cartilage changed first in the early OA, decreasing gradually from the surface to the deeper layers with the degeneration [22, 23].
It has been well known that the earliest signs of OA include the loss of proteoglycans in the superficial layer and the disruption of the superficial collagen network, leading to fibrillation in the surface and softening in the superficial tissue [24–26]. High-frequency UBM has been demonstrated its useful application in detecting the early damage in the articular cartilage tissue due to the ability of ultrasound to penetrate into the tissue . Responding to the changes in the surface roughness and the internal compositions of the OA cartilage, the ultrasound reflection and scattering in the tissue increased and thus the enhanced echoes reflected from the internal tissue were demonstrated in the UBM image . The finding is consistent with the results of other studies [12, 27]. These studies, however, did not provide any quantitative criteria for the ultrasonic diagnosis of OA cartilage.
In engineering, the roughness index is mainly used to describe the small valley and pitch conditions on the material surface, as known as micro-roughness. Recently, it has been used to describe the roughness of the cartilage surface in the evaluation of cartilage degeneration. Our results of the surface roughness calculation show that the roughness of the cartilage surface increased with the OA grade, indicating that the cartilage surface became rough and uneven with the development of OA. Similar results have been obtained in previous studies of cartilage degradation . The statistical results in this study showed significant differences in URI between the grade 3 cartilage and the cartilage with grade 1 and grade 2 (p < 0.05), but an insignificant change between the grade 1 cartilage and the grade 2 cartilage (p > 0.05), showing that cartilage degradation progressively developed from quantitative change to qualitative change. It was found that the surfaces of the cartilage samples with grade 1 and 2 started to become rough with a significant increase in URI compared with normal cartilage, but no significant change between groups of grade 1 and grade 2. However, a sharp deterioration occurred in the grade 3 cartilage samples with a great increase in URI to 193.27 ± 75.06 μm. The results show that the surface roughness index URI could be used to distinguish early OA and mid OA, but has no ability to accurately distinguish the grades of early OA (i.e. grade 1 and grade 2).
Two possible reasons may explain the decrease in the reflection coefficient of the cartilage surface with the OA grade. First, the increase of surface roughness resulted in diffuse reflection and consequently decreased the amplitude of the echoes. Secondly, while OA occurred, the cartilage surface was softened. The compositions and structure of articular cartilage gradually changed from the surface to the deep layer. As more transmitted ultrasonic energies were absorbed, the reflection coefficient decreased. Contrary to the reflection coefficient of the surface, the reflection coefficient of the cartilage-bone interface increased with OA grades (Figure 4c). It was revealed that the bottom reflection coefficient of the grade 1 cartilage changed insignificantly compared with normal cartilage while that of OA cartilage with grade 2 and 3 increased significantly. The OA cartilage with grade 3 had a significant increase compared with normal cartilage and early OA cartilage (grade 1 and 2).
Therefore, it might be speculated that the reflection coefficient of the cartilage surface could be a more sensitive index to distinguish the early OA grades than the surface roughness index and the reflection coefficient of the cartilage-bone interface. The integrated analysis of these three parameters in diagnosis of cartilage degeneration not only evaluates the surface morphology (surface roughness), but also assesses the impact of the composition changes by measurement of the reflection coefficients of both the cartilage surface and the cartilage-bone interface. Thus, more accurate diagnostic results may be obtained.
Two limitations of this study require further investigations. First, the OA grades were only evaluated subjectively by three pathologists in this study. The quantitative analysis of changes in compositions and structural parameters such as PG content and cartilage thickness and the sequent study in their relationships to ultrasound parameters are needed. Secondly, the numbers of the samples with different OA grades were uneven, especially the number of OA cartilage with grade 3 was small (N = 6). The small number of samples may have some impacts on the statistical results. Thus further studies with large number of samples are needed.
The UBM measurements reflect the changes in the surface roughness index and the ultrasound reflection coefficients of the cartilage samples with different OA grades. This study suggests that these two ultrasound acoustic parameters have potential to become objective criteria in OA grading.
This work was supported by the National Natural Science Foundation of China (31170896) and China National 973 Program for High-tech Research (2010CB732505).
- Brandt K, Doherty M, Lohmander S: Osteoarthritis Cartilage. 1998, London: Oxford University PressGoogle Scholar
- Arden N, Nevitt MC: Osteoarthritis: epidemiology. Best Pract Res Clin Rheumatol. 2006, 20: 3-25. 10.1016/j.berh.2005.09.007.View ArticlePubMedGoogle Scholar
- Mow VC, Gu WY, Chen FH: Structure and function of articular cartilage and meniscus. Basic orthopaedic biomechanics and mechano-biology. Edited by: Mow VC, Huiskes R. 2005, Philadelphia: Lippincott Williams & Wilkins, 181-258. 3Google Scholar
- Ihara H: Double-contrast CT arthrography of the cartilage of the patellofemoral joint. Clin Orthop. 1985, 198: 50-55.PubMedGoogle Scholar
- Gagliardi JA, Chung EM, Chandnani VP, Keslin KL, Christensen KP, Null RN, Radvany MG, Hansen MF: Detection and staging of chondromalacia patellae: relative efficacies of conventional MR imaging, MR arthrography, and CT arthrography. AJR Am J Roentgenol. 1994, 163: 629-636.View ArticlePubMedGoogle Scholar
- Nieminen MT, Toyras J, Rieppo J, Hakumaki JM, Silvennoinen J, Helminen HJ, Jurvelin JS: Quantitative MR microscopy of enzymatically degraded articular cartilage. Magn Reson Med. 2000, 43: 676-681. 10.1002/(SICI)1522-2594(200005)43:5<676::AID-MRM9>3.0.CO;2-X.View ArticlePubMedGoogle Scholar
- Burstein D, Bashir A, Gray ML: MRI techniques in early stages of cartilage disease. Invest Radiol. 2000, 35: 622-638. 10.1097/00004424-200010000-00008.View ArticlePubMedGoogle Scholar
- Hardy PA, Ridler AC, Chiarot CB, Plewes DB, Henkelman RM: Imaging articular cartilage under compression-cartilage elastography. Magn Reson Med. 2005, 53: 1065-1073. 10.1002/mrm.20439.View ArticlePubMedGoogle Scholar
- Lopez O, Amrami KK, Manduca A, Rossman PJ, Ehman RL: Developments in dynamic MR elastography for in vitro biomechanical assessment of hyaline cartilage under high-frequency cyclical shear. J Magn Reson Imaging. 2007, 25: 310-320. 10.1002/jmri.20857.View ArticlePubMedGoogle Scholar
- Roemer FW, Mohr A, Lynch JA, Meta MD, Guermazi A, Genant HK: Micro-CT arthrography: a pilot study for the ex vivo visualization of the rat knee joint. AJR Am J Roentgenol. 2005, 184: 1215-1219.View ArticlePubMedGoogle Scholar
- Saarakkala S, Töyräs J, Hirvonen J, Laasanen MS, Lappalainen R, Jurvelin JS: Ultrasonic quantitation of superficial degradation of articular cartilage. Ultrasound Med Biol. 2004, 30: 783-792. 10.1016/j.ultrasmedbio.2004.03.005.View ArticlePubMedGoogle Scholar
- Saied A, Cherin E, Gaucher H, Laugier P, Gillet P, Floquet J, Netter P, Berger G: Assessment of articular cartilage and subchondral bone: subtle and progressive changes in experimental osteoarthritis using 50 MHz echography in vitro. J Bone Miner Res. 1997, 12: 1378-1386. 10.1359/jbmr.1922.214.171.1248.View ArticlePubMedGoogle Scholar
- Suh JKF, Youn I, Fu FH: An in situ calibration of an ultrasound transducer: a potential application for an ultrasonic indentation test of articular cartilage. J Biomech. 2001, 34: 1347-1353. 10.1016/S0021-9290(01)00088-4.View ArticlePubMedGoogle Scholar
- Wang Q: Ultrasonic characterization of transient and inhomogeneous swelling behavior and progressive degeneration of articular cartilage. PhD Thesis. 2007, The HongKong Polytechnic University, Department of Health Technology and InformaticsGoogle Scholar
- Wang Y, Guo Y, Zhang L, Niu H, Xu M, Zhao B, Wan W: Ultrasound biomicroscopy for the detection of early osteoarthritis in an animal model. Acad Radiol. 2011, 18: 167-173. 10.1016/j.acra.2010.09.011.View ArticlePubMedGoogle Scholar
- Niu HJ, Wang Q, Zheng YP, Fan YB: A new method for computing the uniaxial modulus of articular cartilages using modified inhomogeneous triphasic model. Acta Mechanica Sinica. 2010, 26: 121-126. 10.1007/s10409-009-0287-x.View ArticleGoogle Scholar
- Zheng YP, Niu HJ, Mak AFT, Huang YP: Ultrasonic measurement of depth-dependent transient behaviors of articular cartilage under compression. J Biomech. 2005, 38: 1830-1837. 10.1016/j.jbiomech.2004.08.020.View ArticlePubMedGoogle Scholar
- Hattori K, Ikeuchi K, Morita Y, Takakura Y: Quantitative ultrasonic assessment for detecting microscopic cartilage damage in osteoarthritis. Arthritis Res Ther. 2005, 7: 38-46. 10.1186/ar1497.View ArticleGoogle Scholar
- Nieminen HJ, Toyras J, Rieppo J, Nieminen MT, Hirvonen J, Korhonen R, Jurvelin JS: Real-time ultrasound analysis of articular cartilage degradation in vitro. Ultrasound Med Biol. 2002, 28: 519-525. 10.1016/S0301-5629(02)00480-5.View ArticlePubMedGoogle Scholar
- Pellaumail B, Watrin A, Loeuille D, Netter P, Berger G, Laugier P, Saïed A: Effect of articular cartilage proteoglycan depletion on high frequency ultrasound backscatter. Osteoarthr Cartilage. 2002, 10: 535-541. 10.1053/joca.2002.0790.View ArticleGoogle Scholar
- Pritzker KP, Gay S, Jimenez SA, Ostergaard K, Pelletier JP, Revell PA, Salter D, van den Berg WB: Osteoarthritis cartilage histopathology: grading and staging. Osteoarthr Cartilage. 2006, 14: 13-29. 10.1016/j.joca.2005.07.014.View ArticleGoogle Scholar
- Arokoski JP, Jurvelin JS, Väätäinen U, Väätäinen U, Helminen HJ: Normal and pathological adaptations of articular cartilage to joint loading. Scand J Med Sci Sports. 2000, 10: 186-198. 10.1034/j.1600-0838.2000.010004186.x.View ArticlePubMedGoogle Scholar
- Freeman MA: Is collagen fatigue failure a cause of osteoarthrosis and prosthetic component migration? A hypothesis. J Orthop Res. 1999, 17: 3-8. 10.1002/jor.1100170103.View ArticlePubMedGoogle Scholar
- Bland YS, Ashhurst DE: Development and ageing of the articular cartilage of the rabbit knee joint: distribution of the fibrillar collagens. Anat Embryol. 1996, 194: 607-619.View ArticlePubMedGoogle Scholar
- Joiner GA, Bogoch ER, Pritzker KP, Buschmann MD, Chevrier A, Foster FS: High frequency acoustic parameters of human and bovine articular cartilage following experimentally-induced matrix degradation. Ultrasonic Imaging. 2001, 23: 106-116.View ArticlePubMedGoogle Scholar
- Kim HK, Babyn PS, Harasiewicz KA, Gahunia HK, Pritzker KP, Foster FS: Imaging of immature articular cartilage using ultrasound backscatter microscopy at 50 MHz. J Orthop Res. 1995, 13: 963-970. 10.1002/jor.1100130622.View ArticlePubMedGoogle Scholar
- Julkunen P, Iivarinen J, Brama PA, Arokoski J, Jurvelin JS, Helminen HJ: Maturation of collagen fibril network structure in tibial and femoral cartilage of rabbits. Osteoarthr Cartilage. 2010, 18: 406-415. 10.1016/j.joca.2009.11.007.View ArticleGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/13/34/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.