Influence of bi- and tri-compartmental knee arthroplasty on the kinematics of the knee joint
© Wünschel et al; licensee BioMed Central Ltd. 2011
Received: 6 September 2010
Accepted: 27 January 2011
Published: 27 January 2011
The cruciate ligaments are important stabilizers of the knee joint and determine joint kinematics in the natural knee and after cruciate retaining arthroplasty.
No in vitro data is available to biomechanically evaluate the ability of the anterior cruciate ligament (ACL) to maintain knee joint kinematics after bicruciate-retaining bi-compartmental knee arthroplasty (BKA).
Therefore, the objective of the current study was to investigate the kinematics of the natural knee joint, before and after installing bicruciate-retaining BKA and posterior cruciate retaining total knee arthroplasty. Specifically, we incorporated a dynamic knee simulator to simulate weight-bearing flexions on cadaveric knee specimen before and after surgical manipulations.
In this cadaveric study we investigated rotational and translational tibiofemoral kinematics during simulated weight-bearing flexions of the intact knee, after bi-compartmental knee arthroplasty (BKA+), after resecting the ACL in BKA (BKA-), and after posterior cruciate retaining total knee arthroplasty (TKA).
Rotation of BKA+ is closest to the intact knee joint, whereas TKA shows significant differences from 30 to 90 degree of flexion. Within the tested flexion range (15 to 90 degree of flexion), there was no significant difference in the anterior-posterior translation among intact, BKA+, and TKA knees. Resecting the ACL in BKA leads to a significant anterior tibial translation.
BKA with intact cruciate ligaments resembles rotation and translation of the natural knee during a simulated weight-bearing flexion. It is a suitable treatment option for medial and patellofemoral osteoarthritis with advantages in rotational characteristics compared to TKA.
Unicondylar knee Arthroplasty (UKA) is a well established treatment option for osteoarthritis of either medial or less often lateral knee joint compartment. Advantages include the minimal surgical exposure and the intact cruciate ligaments. It also provides the possibility to switch to a total knee arthroplasty (TKA) later, even though UKA conversion to TKA has been shown to be associated with poorer clinical outcome compared to primary TKA . Shortcomings include: that it is unable to correct severe deviations of the mechanical axis; may result in restricted joint range of motion; requires intact ligamentous structures. UKA has been reported with excellent results comparable to those of TKA in longitudinal studies [2, 3]. One of the main reasons for revision surgery after UKA is ongoing degeneration in the other compartments particularly in the patellofemoral compartment [4, 5].
Bi-compartmental knee arthroplasty (BKA), replacing the medial and patellofemoral compartments, attempts to satisfy the fact that these compartments are most often affected by osteoarthritis [6, 7]. The surgical approach for BKA is either to combine UKA and patellofemoral arthroplasty (PFA) in a modular design [8, 9], or to use a recently developed non-modular femoral design [10–13]. In BKA, the anterior and posterior cruciate ligaments (PCL) can be preserved, and the reasons for retaining the cruciate ligaments in knee arthroplasty (KA) design include enhanced stability, decreased shear force between implant-bone interface, more physiological tibiofemoral kinematics, and maintenance of proprioception [14, 15]. Therefore, theoretically, BKA may have same advantages of the UKA in terms of joint stability.
However, considering the articular surface in the tibiofemoral joint is altered after KA, the effect of the ACL on knee joint kinematics after bicruciate-retaining BKA may be different from that in the native knee. Although clinical data do exist [8, 9, 12], no in vitro data is available to biomechanically evaluate the ability of the ACL to maintain knee joint kinematics after bicruciate-retaining BKA.
Therefore, the objective of the current study was to investigate the kinematics of the natural knee joint, with bicruciate-retaining BKA and with posterior cruciate retaining TKA. Specifically, we incorporated a dynamic knee simulator to simulate weight-bearing flexions on cadaveric knee specimen before and after surgical manipulations.
We hypothesized that, (1) comparing to the native knee, the kinematics in internal-external rotation and anterior-posterior translation are not affected after a bicruciate-retaining BKA and (2) resecting the ACL in BKA as well as an ACL- sacrificing (PCL-retaining) TKA lead to different kinematic patterns.
Twelve fresh-frozen human cadaveric knee specimens with an age at the time of death of 75 ± 13 years (mean ± standard deviation) were studied. The femur and tibia were cut 15 cm from the joint line, and, while keeping the joint capsule and the collateral ligaments intact and the five aforementioned muscle tendons exposed, all other skin and soft tissues were removed. The fibula was secured to the tibia with cortical screws to prevent its motion during the test. Each of the femur and tibia was mounted onto a thick-walled steel cylinder using a bone cement compound (PMMA: Technovit 2060, Heraeus Kulzer, Hanau, Germany) and multiple accurately-positioned set screws. To achieve a secure grip of the tendon, a custom-manufactured metal tendon clamp was used to connect the tendon to the muscle actuator with a cable whose line of action is parallel to the femoral shaft.
To generate the weight-bearing knee flexions, the main actuator produced a continuous, descending motion of the hip assembly from 15 to 90 degrees of knee flexion with a constant rate of 1 deg/s. During this movement, the control system dynamically adjusted the muscle cable tension by varying their lengths so as to maintain a constant resulting ground reaction force on the ankle joint. The reaction force on the ankle joint, which holds equilibrium with the applied muscle forces quasi-statically, was assumed to characterize the amount of body weight. In order to prevent the tendon ruptures caused by excessive muscle forces , we selected a conservative ankle force of 50 Newton to simulate a portion of the body weight, which required the quadriceps-actuators to pull with a linearly rising force and a maximum of approximately 600N at 90 degrees of flexion. During knee flexion, the three quadriceps forces were always maintained identical to one another, while the hamstrings forces were kept constant at 10 N. A constant hamstring force was used to simplify the control algorithm and to reveal the effects of other factors of research interests. This knee flexion with a 50 N simulated partial body weight was repeated twice for each of the four different parameters.
To study the kinematics of the joint, the movements of the femur and the tibia were measured with a marker-based ultrasonic measuring system for 3D motion analysis (ZEBRIS® CMS100, Isny, Germany) at a sampling rate of 1 Hz and spatial resolution < 1 mm . A triad of ultrasound markers was attached at each of the femur and tibia fixation cylinders. To define the two body-fixed coordinate systems for the femur and tibia, we first recorded the positions of the medial and lateral prominences of the tibia plateau as reference points using a ZEBRIS® stylus pointer when the knee was fully extended. Both of tibial and femoral coordinate systems were assumed to be identical at full extension, and their common origins at full extension were defined as the midpoint of the two digitized reference points. For each of the tibial and femoral coordinate systems, the flexion axis (z-axis) was defined along the line between the two reference points. The y-axis was defined as a vector normal to a plane constructed by the z-axis and the longitudinal axis of the respective segment shafts, which were recorded by the ultrasound sensor triads at the corresponding segments. The x-axis was then defined by the cross product of the y and z axes (figure 1). During the flexion/extension of the knee, the segment fixed coordinate systems at each instant were calculated according to the positions of tibial and femoral ZEBRIS® marker triads. The tibial translation with respect to the femur was defined as the position difference between the centers of the two moving coordinates, and the relative orientation of the tibia with respect to the femur was calculated in terms of Euler angles (rotation sequence: flexion-extension, abduction-adduction, internal-external rotation).
To evaluate the effect of the KA design on the kinematics, at each of the tested knee flexion angles, we conducted a one-way repeated-measure analysis of variance (ANOVA) with knee states as the independent factors. A post hoc test using Tukey-Kramer method was also conducted to investigate the individual effect. A p-value less than 0.05 was considered to be statistically significant. The statistical analysis was conducted with a statistics computer software, SAS® (SAS Institute Inc., Cary, NC).
The research presented in this work conforms to the Helsinki Declaration and to local legislation. It has been approved by the ethical committee of the medical faculty of the University of Tübingen.
After explanting BKA and performing TKA, we found significantly different rotational data between the intact knee and TKA trials from 30 to 90 degrees of flexion. The TKA design resulted in external tibial rotation from 15 to 30 degree of flexion and then changed into internal rotation from 30 to 90 degree of flexion. The rotation was approximately 8 and 5 degree less than those of the intact knee and BKA+ trials, respectively.
In this study, we conducted simulated weight-bearing knee flexions to investigate the effect of bicruciate-retaining BKA (BKA+), ACL-resected BKA design (BKA-), and posterior-cruciate retaining TKA on translational and rotational knee joint kinematics. Since the investigations were performed with simulated muscle forces, the tibiofemoral kinematics of different KA designs with different ACL-states can be analyzed in a more physiologic fashion compared to previous studies . The experimental design allows comparisons of multiple knee states on the same knee specimen and thus minimizes the effects of inter-specimen variations.
Concerning tibial rotation, TKA could not maintain physiologic kinematics from 30 to 90 degrees of flexion. Our results only show a slight internal rotation during flexion which was significantly less than that in the natural knee. This finding corroborates with other studies  and may result from the concave shape of the inlay, which limits rotational abilities. The amount of internal rotation in the intact knee in our setup was less than those reported in in-vivo MRI-studies . It has been hypothesized  that the hamstring muscles have a significant influence on tibial rotation in vivo thus explaining lower degrees of rotation in vitro, where this muscle group usually is not addressed at all. We used a constant hamstring force of 10N bilaterally and did not apply any rotational torque. Further investigation on this topic is clearly warranted.
The BKA+ and BKA- trials did not show a significant difference in tibial rotation, which leads to the assumption that the ACL does not have an influence on tibial rotation during a simulated weight-bearing flexion. This is in contrast to several other studies [19, 22] and might be caused by high quadriceps-forces used in this study to simulate partial body weight. Another difference is that we consistently increased quadriceps-force during flexion as it is in vivo, taking into account the changing lever-arms during flexion, while other studies usually apply the same amount of force to the quadriceps-muscle thus producing a less physiological condition [20, 23].
Figure 3 shows that the intact knee and BKA+ have almost parallel rotational curves. The fact that BKA+ rotates less may be explained by the influence of the surgery leading to changes of the soft tissue.
Our findings show that bicruciate-retaining BKA as well as TKA preserve the natural-knee anterior-posterior translation with a minimally more anteriorly positioned tibia. This may result from a slight ligamentous laxity caused by the surgical procedure. This is in contrast to the findings in previous in vivo clinical investigations  that the patients with the ACL-retaining TKA had more natural tibiofemoral kinematics than those with PCL-retaining TKA did during treadmill gait and stair stepping.
Our findings also suggest the geometrical congruity of the TKA used in this study was able to compensate for the missing ACL. However, after cutting the ACL in BKA (BKA-), the tibia significantly moved anterior compared to all other states. This is because the tibial plateau in BKA is flat and thus not able to compensate for the missing ACL. The designs of the implanted femoral components in TKA and BKA (except for the missing lateral condyle in BKA) are alike and thus do not cause differences in kinematics. Other studies also emphasize the importance of the articular geometry after KA .
Several limitations exist in the current study. First of all, we simulated only a partial body weight. This was because the quadriceps tendons in some of the specimens could not sustain the corresponding muscle forces resulting from higher body weight simulation. In order to avoid the risks of tendon rupture under high simulated muscle loading, we compromised to simulate only a portion of the body weight . Moreover, previous investigation  has shown that the change in the knee kinematics profile is not sensitive to the increasing simulated body weight. Therefore, we believe that qualitative clinical insights can still be elucidated with the partially loaded knee. However, although the results are in concordance with in vivo studies [26–28] they can not definitely be extrapolated to predict the knee kinematics under full body weight in vivo. A second limitation is that the tested specimens did not display the indications, such as medial compartment degeneration and ligamentous/capsular changes typical in patients eligible for KA. Therefore, extra care was taken by the surgeon to ensure that all the ligaments were properly tensioned and knee axes were well aligned after implantation of the components. A third limitation in this study is that the knee flexion only started from 15 degree, because the joint could not be flexed from fully-extended position by only applying a partial body weight and quadriceps force. This is due to the setup of the knee kinemator control: Since we drive the kinemator fully force-controlled the specimen might hyperextend and thus be destroyed if the extension angles reach below 15 degree. The information within first 15 degree of flexion is thus missing. A fourth limitation is the hamstring forces were kept constant during the simulated flexion. This is mainly because the multiple agonist and antagonist muscle forces composed a mechanically-indeterminate system, and it is not likely to determine their individual contributions. We therefore believe the use of constant hamstring forces can isolate the effects of other factors without sacrificing the fidelity of the model.
Our data show that the translational and rotational knee joint kinematics after bicruciate-retaining BKA resembles that of the native knee. On the other hand PCL-retaining TKA results in less rotation and similar translation during a partially weight-bearing flexion. Our findings suggests that, provided functional ligamentous structures, bicruciate-retaining BKA is a suitable treatment option for medial and patellofemoral osteoarthritis of the knee joint with advantages in rotational characteristics compared to TKA.
This study was partly funded by Smith & Nephew.
- Jarvenpaa J, Kettunen J, Miettinen H, Kroger H: The clinical outcome of revision knee replacement after unicompartmental knee arthroplasty versus primary total knee arthroplasty: 8-17 years follow-up study of 49 patients. Int Orthop. 2010, 34: 649-653. 10.1007/s00264-009-0811-4.View ArticlePubMedGoogle Scholar
- Berger RA, Meneghini RM, Jacobs JJ, Sheinkop MB, la Valle CJ, Rosenberg AG, Galante JO: Results of unicompartmental knee arthroplasty at a minimum of ten years of follow-up. J Bone Joint Surg Am. 2005, 87: 999-1006. 10.2106/JBJS.C.00568.View ArticlePubMedGoogle Scholar
- Saccomanni B: Unicompartmental knee arthroplasty: a review of literature. Clin Rheumatol. 2010, 29: 339-346. 10.1007/s10067-009-1354-1.View ArticlePubMedGoogle Scholar
- Callahan CM, Drake BG, Heck DA, Dittus RS: Patient outcomes following unicompartmental or bicompartmental knee arthroplasty. A meta-analysis. J Arthroplasty. 1995, 10: 141-150. 10.1016/S0883-5403(05)80120-2.View ArticlePubMedGoogle Scholar
- Ackroyd CE: Medial compartment arthroplasty of the knee. J Bone Joint Surg Br. 2003, 85: 937-942. 10.1302/0301-620X.85B7.14650.View ArticlePubMedGoogle Scholar
- Davies AP, Vince AS, Shepstone L, Donell ST, Glasgow MM: The radiologic prevalence of patellofemoral osteoarthritis. Clin Orthop Relat Res. 2002, 206-212. 10.1097/00003086-200209000-00020.Google Scholar
- McAlindon TE, Snow S, Cooper C, Dieppe PA: Radiographic patterns of osteoarthritis of the knee joint in the community: the importance of the patellofemoral joint. Ann Rheum Dis. 1992, 51: 844-849. 10.1136/ard.51.7.844.View ArticlePubMedPubMed CentralGoogle Scholar
- Heyse TJ, Khefacha A, Cartier P: UKA in combination with PFR at average 12-year follow-up. Arch Orthop Trauma Surg. 2009Google Scholar
- arratte S, Pauly V, Aubaniac JM, Argenson JN: Survival of bicompartmental knee arthroplasty at 5 to 23 years. Clin Orthop Relat Res. 2010, 468: 64-72. 10.1007/s11999-009-1018-0.View ArticleGoogle Scholar
- Engh GA: A bi-compartmental solution: what the Deuce?. Orthopedics. 2007, 30: 770-771.PubMedGoogle Scholar
- Rolston L, Bresch J, Engh G, Franz A, Kreuzer S, Nadaud M, Puri L, Wood D: Bicompartmental knee arthroplasty: a bone-sparing, ligament-sparing, and minimally invasive alternative for active patients. Orthopedics. 2007, 30: 70-73.PubMedGoogle Scholar
- Rolston L, Siewert K: Assessment of knee alignment after bicompartmental knee arthroplasty. J Arthroplasty. 2009, 24: 1111-1114. 10.1016/j.arth.2008.07.006.View ArticlePubMedGoogle Scholar
- Wang H, Dugan E, Frame J, Rolston L: Gait analysis after bi-compartmental knee replacement. Clin Biomech (Bristol , Avon ). 2009, 24: 751-754. 10.1016/j.clinbiomech.2009.07.014.View ArticleGoogle Scholar
- Andriacchi TP, Galante JO: Retention of the posterior cruciate in total knee arthroplasty. J Arthroplasty. 1988, 3 Suppl: S13-S19. 10.1016/S0883-5403(88)80003-2.View ArticlePubMedGoogle Scholar
- Conditt MA, Noble PC, Bertolusso R, Woody J, Parsley BS: The PCL significantly affects the functional outcome of total knee arthroplasty. J Arthroplasty. 2004, 19: 107-112. 10.1016/j.arth.2004.06.006.View ArticlePubMedGoogle Scholar
- Müller O, Lo J, Wünschel M, Obloh C, Wülker N: Simulation of force loaded knee movement in a newly developed in vitro knee simulator. Biomed Tech (Berl). 2009, 54: 142-149.View ArticleGoogle Scholar
- Zavatsky AB: A kinematic-freedom analysis of a flexed-knee-stance testing rig. J Biomech. 1997, 30: 277-280. 10.1016/S0021-9290(96)00142-X.View ArticlePubMedGoogle Scholar
- Stukenborg-Colsman C, Ostermeier S, Wenger KH, Wirth CJ: Relative motion of a mobile bearing inlay after total knee arthroplasty--dynamic in vitro study. Clin Biomech (Bristol , Avon ). 2002, 17: 49-55. 10.1016/S0268-0033(01)00103-6.View ArticleGoogle Scholar
- Markolf KL, Park S, Jackson SR, McAllister DR: Anterior-posterior and rotatory stability of single and double-bundle anterior cruciate ligament reconstructions. J Bone Joint Surg Am. 2009, 91: 107-118. 10.2106/JBJS.G.01215.View ArticlePubMedGoogle Scholar
- Varadarajan KM, Harry RE, Johnson T, Li G: Can in vitro systems capture the characteristic differences between the flexion-extension kinematics of the healthy and TKA knee?. Med Eng Phys. 2009, 31: 899-906. 10.1016/j.medengphy.2009.06.005.View ArticlePubMedGoogle Scholar
- Hill PF, Vedi V, Williams A, Iwaki H, Pinskerova V, Freeman MA: Tibiofemoral movement 2: the loaded and unloaded living knee studied by MRI. J Bone Joint Surg Br. 2000, 82: 1196-1198. 10.1302/0301-620X.82B8.10716.View ArticlePubMedGoogle Scholar
- Andersen HN, Dyhre-Poulsen P: The anterior cruciate ligament does play a role in controlling axial rotation in the knee. Knee Surg Sports Traumatol Arthrosc. 1997, 5: 145-149. 10.1007/s001670050042.View ArticlePubMedGoogle Scholar
- Most E, Li G, Sultan PG, Park SE, Rubash HE: Kinematic analysis of conventional and high-flexion cruciate-retaining total knee arthroplasties: an in vitro investigation. J Arthroplasty. 2005, 20: 529-535. 10.1016/j.arth.2004.09.034.View ArticlePubMedGoogle Scholar
- Moro-oka TA, Muenchinger M, Canciani JP, Banks SA: Comparing in vivo kinematics of anterior cruciate-retaining and posterior cruciate-retaining total knee arthroplasty. Knee Surg Sports Traumatol Arthrosc. 2007, 15: 93-99. 10.1007/s00167-006-0134-6.View ArticlePubMedGoogle Scholar
- Bull AM, Kessler O, Alam M, Amis AA: Changes in knee kinematics reflect the articular geometry after arthroplasty. Clin Orthop Relat Res. 2008, 466: 2491-2499. 10.1007/s11999-008-0440-z.View ArticlePubMedPubMed CentralGoogle Scholar
- Li G, Papannagari R, Nha KW, Defrate LE, Gill TJ, Rubash HE: The coupled motion of the femur and patella during in vivo weightbearing knee flexion. J Biomech Eng. 2007, 129: 937-943. 10.1115/1.2803267.View ArticlePubMedGoogle Scholar
- Varadarajan KM, Gill TJ, Freiberg AA, Rubash HE, Li G: Gender differences in trochlear groove orientation and rotational kinematics of human knees. J Orthop Res. 2009, 27: 871-878. 10.1002/jor.20844.View ArticlePubMedPubMed CentralGoogle Scholar
- Defrate LE, Sun H, Gill TJ, Rubash HE, Li G: In vivo tibiofemoral contact analysis using 3D MRI-based knee models. J Biomech. 2004, 37: 1499-1504. 10.1016/j.jbiomech.2004.01.012.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/12/29/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.