- Research article
- Open Access
- Open Peer Review
Fixation strength of biocomposite wedge interference screw in ACL reconstruction: effect of screw length and tunnel/screw ratio. A controlled laboratory study
© Herrera et al; licensee BioMed Central Ltd. 2010
- Received: 3 March 2010
- Accepted: 30 June 2010
- Published: 30 June 2010
Primary stability of the graft is essential in anterior cruciate ligament surgery. An optimal method of fixation should be easy to insert and provide great resistance against pull-out forces.
A controlled laboratory study was designed to test the primary stability of ACL tendinous grafts in the tibial tunnel. The correlation between resistance to traction forces and the cross-section and length of the screw was studied.
The tibial phase of ACL reconstruction was performed in forty porcine tibias using digital flexor tendons of the same animal. An 8 mm tunnel was drilled in each specimen and two looped tendons placed as graft. Specimens were divided in five groups according to the diameter and length of the screw used for fixation. Wedge interference screws were used. Longitudinal traction was applied to the graft with a Servohydraulic Fatigue System. Load and displacement were controlled and analyzed.
The mean loads to failure for each group were 295,44 N (Group 1; 9 × 23 screw), 564,05 N (Group 2; 9 × 28), 614,95 N (Group 3; 9 × 35), 651,14 N (Group 4; 10 × 28) and 664,99 (Group 5; 10 × 35). No slippage of the graft was observed in groups 3, 4 and 5. There were significant differences in the load to failure among groups (ANOVA/P < 0.001).
Longer and wider interference screws provide better fixation in tibial ACL graft fixation. Short screws (23 mm) do not achieve optimal fixation and should be implanted only with special requirements.
- Anterior Cruciate Ligament
- Anterior Cruciate Ligament Reconstruction
- Tibial Tunnel
- Interference Screw
- Hamstring Tendon
Anterior cruciate ligament (ACL) reconstruction has become one of the most frequent procedures in arthroscopic surgery of the knee . For many years, arthroscopic bone-tendon-bone reconstruction has been considered the gold standard operation in ACL surgery. In the last decade, the use of double looped hamstrings has been growing in popularity among arthroscopists. Excellent results have been reported with this technique [2, 3], without significant differences between the outcomes of both procedures .
However, there is some degree of controversy about the increase of knee laxity during the first stages of the postoperative period after the use of hamstring tendons. Four bundle hamstrings grafts have more resistance than normal ACL and BTB grafts [5, 6], but the primary fixation obtained by the use of an interference screw between bone blocks and bone tunnels is supposed to be better than fixation obtained with hamstrings and any other device.
Looking for a device that allows the use of hamstrings with a solid primary fixation has been the challenge for both, orthopaedic surgeons and bioengineers. This ideal device should be biocompatible, respectful with the graft and easy to insert. In addition, it should allow the placement as near as possible from the joint space to avoid/decrease the windshield-wiper effect and should be able to allow motion, weight-bearing and close kinetic chain exercises (CKCE) from the very early stages of rehabilitation . During the last years, new methods of fixation have been developed on this basis.
Bio-absorbable implants would add the advantage of avoiding hardware permanence, intolerance and need of removal. In addition, these implants allow magnetic resonance explorations without misrepresentation of the obtained images .
The purpose of this experimental study was to test the primary fixation obtained with a specific type of screw and to determine the effect of the screw/tunnel ratio and implant length in the resistance of the graft to traction forces.
The primary hypothesis of the study was: "For a determined tibial tunnel diameter, longer and wider screws provide better fixation of the graft"
Forty rear limbs from young female or castrated male pigs were used in this study. Pigs were sacrificed at an industrial slaughterhouse after being stunned with CO2. Rear limbs were vacuum packed and stored refrigerated (4°C) for 24 hours and then manipulated at room temperature. Tibias and flexor digitorum profundus tendons were dissected from the rest of the tissue.
Tibias were placed in a holder and a 8 mm tunnel was performed with the Stryker® ACL instrumentation. The direction of the tunnel was from the medial cortex of the proximal tibia to the lateral side of the anterior half of the medial tibial spine.
For each case, two flexor digitorum profundus tendons were selected and prepared to compose a double-looped graft with a cross-section optimal for the diameter of the tunnel (8 mm). The Stryker® ACL Workstation was used for pre-tensing the graft and a 17 lbs (75,62 N) tension was applied.
The looped tendons were passed through the tunnel forming a four strand graft and then fixed with Biosteon® Wedge interference screws (Stryker®) of different lengths and diameters. Screws were inserted until their rear end was at the level of the entrance of the tunnel (at the antero-medial cortex of the tibial shaft), a 1 mm nitinol guide wire was passed among the tendon bundles for this purpose. There was no protrusion of the tip of the screws at the articular surface of the tibia.
Five groups of testing were conformed, with 8 different specimens in each group. In all of them, the tunnel diameter was 8 mms but the screw selection was done according to the following distribution:
Group 1: Screws of 9 mm diameter and 23 mm length.
Group 2: Screws of 9 mm diameter and 28 mm length.
Group 3: Screws of 9 mm diameter and 35 mm length.
Group 4: Screws of 10 mm diameter and 28 mm length.
Group 5: Screws of 10 mm diameter and 35 mm length.
Tests were performed with a Servohydraulic Fatigue System (Universal Testing Machine Instron® 8800). A special fixture was designed which allowed accurate positioning of the tibia for applying longitudinal tension to the graft. A metal pin was passed through the loop formed by the graft and a second fixture was placed. Load and displacement were controlled and analyzed by the specific Instron® software.
Figure 1. Test arrangement
Specimens were peak loaded to failure at a rate of 2 mm/min. This speed causes a slow load increase, which according to ASTM (American Section of the International Association for Testing Materials) standards is appropriate for testing static friction, as it sets the lowest limit for pull-out resistance, since dynamic friction is higher.
Tests were finished when failure of the fixation (pull-out of the graft) or failure of the graft (elongation) were evident according to the resulting data (shown in real time in the computer monitor).
Descriptive statistical analysis was done with the determination of mean loads, standard deviation, median, minimum and maximum loads for each group. Confidence intervals (C.I.) for the mean value of the load at 95% were obtained applying the Student's t-test for a normal distribution of the sample.
Analysis of variance (ANOVA) was performed for testing the differences among the five groups. A post hoc pairwise comparison was done with the Fisher's protected least square difference test (PLSD). Differences with a significance level of P < 0.05 were considered as statistically significant.
Type of Failure
Confidence Int. (95%)
In Group 1 there was evident fixation failure (graft pull-out) of all the specimens. when a load of 295.4 N +/- 7.92 N was reached. C.I. (286.94, 303.93).
In Group 2, the born load was 565.3 N +/- 90.93N, with 4 cases of evident fixation failure, and 4 cases of graft failure. C.I. (496.45, 631.65).
In Group 3 there wasn't any evident fixation failure (late slippage in 3 cases, but with loads higher than 600 N and after elongation of the graft was evident). The obtained load was 617.45 N +/- 11.99. C.I. (602.16, 627.74).
In Group 4 there wasn't any cases of fixation failure, with graft failure in all the specimens. The reached load was 650.13 N +/- 22.8 N. C.I. (619.06, 683.22).
In Group 5 there weren't any cases of fixation failure, with elongation of the graft in all the specimens. The reached load was 665.6 N +/- 38.47N. C.I. (630.84, 699.13).
These differences in the load to failure among the groups were significant (ANOVA/p < 0.0001). The pairwise comparison (Fisher's PLSD) showed that, when comparing group to group, differences in the mean load to failure were significant except when comparing group 3 with group 4 and group 4 with group 5.
In ACL surgery, primary stability of the graft is essential for allowing early rehabilitation during the post-operative period and a solid fixation of the graft in the tibial tunnel is required for this purpose [13–15]. In this study, experimental testing of a screw tibial fixation for hamstrings tendons has been performed. This device is designed for four-strand grafts; this surgical technique has being gaining ground on bone-tendon-bone procedures because of its low morbidity and good results [2–4], although two-bundle grafts have demonstrated to achieve enough stability . Bio-absorbable screws have evident advantages : they provide similar or even better fixation than metallic screws [17–20], causing less damage to the graft  and without interfering with the graft incorporation process . Zantop et al  reported less laceration of the graft with poly-D,L-lactide (PLDLA) screws than with titanium screws, the results were better when a composite of PLDLA and Tricalcium phosphate was used.
Post-operative stability of the graft is essential during the first stages, after four weeks collagen Sharpey-like fibers begin to develop in the bone-graft interface  and biological fixation will be established. Synthesis of these fibers from postoperative formation of granulation tissue is easier in tendinous grafts than in grafts including bone blocks . Thus, key for success and knee stability depends on the capability of the fixation device to allow weight bearing and early mobilization of the knee.
An optimal study design should include human young fresh cadaveric tibias, but this kind of specimens are difficult to obtain and are destined to be used as allograft. The use of cadaveric bones from elderly specimens is related with low bone mineral density (BMD) and poor fixation strength [24–26]. For these reasons, the use of an animal model is an obvious solution for a controlled laboratory study. Porcine knee models have demonstrated to be a good alternative to cadaveric bone [27–29]; whereas bovine models are correlated with higher traction resistance than human [29–31] or porcine [28, 30–33] models. In the tests performed by Nagarkatti et al. , the average density of porcine bone was similar to that of young human bone, and significantly higher than that of elderly human cadaveric bone.
Biosteon® is a biocomposite made out of Poly L-Lactic Acid and Hydroxyapatite which increases the biocompatibility of the implant and provides osteoconductive potential . This wedge-tip interference screw design (Figure 6) facilitates its insertion even with screws of diameters 1 or 2 mm larger than the tunnel diameter. On the other hand, the wedge tip design makes the diameter of the distal third of the screw smaller than the diameter of rest of the screw. There is less contact between the screw and the bone walls in the proximal portion of the tibial tunnel. During the experimental testing it was obvious that the shortest screws (23 mm length) were the ones that obtained the worst pull-out values (slightly less than 300 N). This difference was significant (p < 0,0001) when comparing group 1 with any of the other groups (Table 2). These values wouldn't guarantee proper fixation during the postoperative usual exercises [36, 37].
9 × 23
9 × 28
9 × 23
9 × 35
9 × 23
10 × 28
9 × 23
10 × 35
9 × 28
9 × 35
9 × 28
10 × 28
9 × 28
10 × 35
9 × 35
10 × 28
9 × 35
10 × 35
10 × 28
10 × 35
With 28 mm length screws and diameter 1 mm wider than the tunnel (9 mm) there was a mean bore load of 564,05 N. When increasing the length to the screw to 35 mm a significant increase (p = 0,0293) in the mean peak load was found. Even better results were obtained using screws with diameter 2 mm wider than the tunnel, with no significant differences between 28 and 35 mm implants.
These results are slightly lower than those reported by Kousa el al.  in a similar porcine study, but in their paper there were two main differences: they used a bone tendon-bone type graft, and they performed their peak load tests at a rate of 50 mm/min; which measures mainly dynamic friction. In our tests, the rate of 2 mm/min determined the static friction, which is the lowest limit for pull-out resistance. Main differences in the design of biomechanical studies (bovine, porcine or human specimens; bone blocks or tendon fixation, metal or absorbable screws) make difficult to compare data obtained in different studies.
These loads are higher than the ones reported during the activities of the initial stages of rehabilitation after ACL surgery like isokinetic/isometric extension of the knee  or quadriceps muscle pull against gravity ; in fact, the graft is loaded less than 500 N during daily living activities [6, 38, 39].
These results confirm the importance of the length of the interference screw. With longer screws it is easier to place the insertion torque near of the joint line [11, 13, 40]. Some studies, however, showed little effect of the screw length [9, 11]. The effect of the screw diameter is another point of discrepancy. In this experimental study, there was an evident improvement with wider screws, which correlates with some other studies .
Morris et al. , in a porcine model, observed serious damage in the graft due to cut-out with wider screws. In our testing, we found no differences and there were no evident damage to the graft with any of the screws, perhaps because the shape of the screw is crucial in the insertion and facilitates the implantation of wider screws without graft damage. Weiler et al.  in a bovine model concluded that increasing the length of the screw improved tibial fixation more than oversizing the diameter of the screw, but this study was done with 23 and 28 mm screws. In our model, for screws longer than 23 mm, the improvement was more evident when oversizing the screw (Table 1: Group 2 to Group 4 = + 87.09 N) than when using a longer screw (Table 1: Group 2 to Group 3 = + 50.90 N).
As Buelow et al reported  the insertion of an interference screw not only compresses the graft in the tunnel but also leads to an enlargement of the tunnel itself. The use of wider screws determine the presence of wider tibial tunnels in the postoperative period (in X-rays, CT or MRI) but with no further enlargement and without any effect in the outcome of the procedure.
According to this experimental testing, the 23 mm screw should only be used in exceptional situations when there is a very short tibial tunnel. With 28 mm screws and an oversized diameter of +1 mm, traction forces higher than 500 N are necessary to obtain some degree of slippage of the graft. With an increase in the length or the diameter of the screw there isn't any kind of slippage, even with loads higher than 600 N. These results are similar or better than the ones obtained in previous studies [10, 12, 41, 43–45].
With this type of screw (Biosteon® Wedge Interference Screw) and a minimal length of 28 mm, primary stability of the graft is achieved and early mobilization of the knee can be allowed. The resistance to traction forces achieved is 200 N higher than the biomechanical requirements for walking with monopodal weight bearing  and could be enough for jumping. The use of 2 mm oversized screws and the maximal length of the implant is recommended to increase primary fixation.
Screws 1 mm wider than the tunnel with a minimal of 28 mm length resist loads higher than 500 N.
Longer and wider interference screws provide better fixation in tibial ACL graft fixation.
Short screws (23 mm) do not achieve optimal fixation and should be implanted only under special requirements.
The authors would like to acknowledge the contribution of José Alberto Villuendas in the logistic management of the specimens during the development of this study.
- Weiler A, ScheZer S, Hoher J: Transplant selection for primary replacement of the anterior cruciate ligament. Orthopade. 2002, 8: 731-740. 10.1007/s00132-002-0331-z.View ArticleGoogle Scholar
- Anderson AF, Snyder RB, Lipscomb AB: Anterior cruciate ligament reconstruction using the semitendinosus and gracilis tendons augmented by the Losee iliotibial band tenodesis. A long-term study. Am J Sports Med. 1994, 22: 620-626. 10.1177/036354659402200509.View ArticlePubMedGoogle Scholar
- Yasuda K, Kondo E, Ichiyama H, Kitamura N, Tanabe Y, Tohyama H, Minami A: Anatomic reconstruction of the anteromedial and posterolateral bundles of the anterior cruciate ligament using hamstring tendon grafts. Arthroscopy. 2004, 20: 1015-1025.View ArticlePubMedGoogle Scholar
- Maletis GB, Cameron SL, Tengan JJ, Burchette RJ: A prospective randomized study of anterior cruciate ligament reconstruction: a comparison of patellar tendon and quadruple-strand semitendinosus-gracilis tendons fixed with bioabsorbable interference screws. Am J Sport Med. 2007, 35: 384-394. 10.1177/0363546506294361.View ArticleGoogle Scholar
- Hamner DL, Brown CH, Steiner ME, Hecker AT, Hayes WC: Hamstring tendon grafts for reconstruction of the anterior cruciate ligament: Biomechanical evaluation of the use of multiple strands and tensioning techniques. J Bone J Surg 81A. 1999, 4: 549-557.Google Scholar
- Noyes FR, Butler DL, Grood ES: et al., Biomechanical analysis of human ligament grafts used in knee-ligament repairs and reconstructions. J Bone Joint Surg 66A. 1984, 344-352.Google Scholar
- Shelbourne KD, Klotz C: What I have learned about ACL: utilizing a progressive rehabilitation scheme to achieve total knee symmetry alter anterior cruciate ligament reconstruction. J Orthop Science. 2006, 11: 318-325. 10.1007/s00776-006-1007-z.View ArticleGoogle Scholar
- Kousa P, Järvinen TLN, Kannus P, Järvinen M: Initial fixation strength of bioabsorbable and titanium interference screws in anterior cruciate ligament reconstruction. Biomechanical evaluation by single cycle and cyclic loading. Am J Sports Med. 2001, 29: 420-425.PubMedGoogle Scholar
- Harvey AR, Thomas NP, Amis AA: The effect of screw length and position on fixation of four-stranded hamstring grafts for anterior cruciate ligament reconstruction. Knee. 2003, 10 (1): 97-102. 10.1016/S0968-0160(02)00092-3.View ArticlePubMedGoogle Scholar
- Morris MW, Williams JL, Thake AJ, Lang Y, Brown JN: Optimal screw diameter for interference fixation in a bone tunnel: a porcine model. Knee Surg Sports Traumatol Arthrosc. 2004, 12 (5): 486-9. 10.1007/s00167-003-0466-4.View ArticlePubMedGoogle Scholar
- Selby JB, Johnson DL, Hester P, Caborn DN: Effect of screw length on bioabsorbable interference screw fixation in a tibial bone tunnel. Am J Sport Med. 2001, 29 (5): 614-619.Google Scholar
- Weiler A, Hoffman RFG, Siepe CJ, Kolbeck SF, Sudkamp NP: The influence of screw geometry on hamstring tendon interference fit fixation. Am J Sports Med. 2000, 28: 356-9.PubMedGoogle Scholar
- Phillips BB, Cain EL, Jeffrey A, Dlabach JA, Azar FM: Correlation of interference screw insertion torque with depth of placement in the tibial tunnel using a quadrupled semitendinosus-gracilis graft in anterior cruciate ligament reconstruction. Arthroscopy. 2004, 20: 1026-1029.View ArticlePubMedGoogle Scholar
- Steiner ME, Hecker AT, Brown CH, Hayes WC: Anterior cruciate ligament graft fixation. Comparison of hamstring and patellar tendon grafts. Am J Sports Med. 1994, 22 (2): 240-246. 10.1177/036354659402200215.View ArticlePubMedGoogle Scholar
- Woo SL, Zeminski J, Knaub MA, Kanamori A, Fu FH: Mechanical testing of anterior cruciate ligaments grafts. Tech Orthop. 1999, 14: 2-13. 10.1097/00013611-199903000-00002.View ArticleGoogle Scholar
- Markolf KL, Parck S, Jackson SR, McAlister DR: Simulated pivot-shift testing with single and double-bundle anterior cruciate ligament reconstructions. J Bone J Surg Am. 2008, 90: 1681-1689. 10.2106/JBJS.G.01272.View ArticleGoogle Scholar
- Caborn DNM, Cohen M, Neef R, Hamilton D, Nylan J, Johnson DL: Quadrupled semitendinosus-gracilis autograft fixation in the femoral tunnel: A comparison between a metal and a bioabsorbable interference screw. Arthroscopy. 1998, 14 (3): 241-245.View ArticlePubMedGoogle Scholar
- Kousa P, Järvinen TLN, Pohjonen T, Kannus P, Kotikiski M, Järvinen M: Fixation strength of a biodegradable screw in anterior cruciate ligament reconstruction. J Bone J Surg Br 77 B. 1995, 901-905.Google Scholar
- Walton M: Absorbable and metal interference screws: Comparison of graft security during healing. Arthroscopy. 1999, 15 (8): 818-826.View ArticlePubMedGoogle Scholar
- Woo SL, Wu C, Dede O, Vercillo F, Noorani S: Biomechanics and anterior cruciate ligament reconstruction. J Orthop Surg Res. 2006, 25 (1): 2-11. 10.1186/1749-799X-1-2.View ArticleGoogle Scholar
- Shapiro JD, Jackson DW, Aberman HM, Lee TQ, Simon TM: Comparison of pullout strength for seven- and nine-millimetre diameter interference screw size as used in anterior cruciate ligament reconstruction arthroscopy. Arthroscopy. 1995, 11: 596-599.View ArticlePubMedGoogle Scholar
- Zantop T, Weimann A, Schmidtko R, Herbort M, Raschke MJ, Petersen WD: Graft Laceration and Pullout Strength of Soft-Tissue AnteriorCruciate Ligament Reconstruction: In Vitro Study ComparingTitanium, Poly-D,L-Lactide, and Poly-D,L-Lactide-Tricalcium Phosphate Screws. Arthroscopy. 2006, 22: 1204-1210.View ArticlePubMedGoogle Scholar
- Rodeo SA, Arnoczky SP, Torzilli PA, Hidaka C, Warren RF: Tendon-healing in a bone tunnel. A biomechanical and histological study in the dog. J Bone Joint Surg Am. 1993, 75: 1795-1803.PubMedGoogle Scholar
- Adam F, Pape D, Schiel K, Steimer O, Kohn D, Rupp S: Biomechanical properties of patellar and hamstring graft tibial fixation techniques in anterior cruciate ligament reconstruction. Am J Sports Med. 2004, 32 (1): 71-78. 10.1177/0095399703258608.View ArticlePubMedGoogle Scholar
- Brand JC, Pienkowski D, Steenlage E, Hamilton D, Johnson DL, Caborn DNM: Interference screw fixation strength of a quadrupled hamstring tendon graft is directly related to bone mineral density and insertion torque. Am J Sports Med. 2000, 28: 705-710.PubMedGoogle Scholar
- Miller CM, Tibone JE, Hewitt M, Kharrazi DF, El Attrache NS: Interference screw divergence in femoral túnel fixation during endoscopic anterior cruciate ligament reconstruction using hamstring grafts. Arthroscopy. 2002, 18 (5): 510-514.View ArticlePubMedGoogle Scholar
- Nurmi JT, Jarvinen TLN, Kannus P, Sievanen H, Toukosalo J, Jarvinen M: Compaction versus extraction drilling for fixation of the hamstring tendon graft in anterior cruciate ligament reconstruction. Am J Sports Med. 2002, 30: 167-173.PubMedGoogle Scholar
- Paschal SO, Seemann MD, Ashman RB, Allard RN, Montgomery JB: Interference fixation versus post fixation of bone-patellar tendon-bone grafts for anterior cruciate ligament reconstruction. A biomechanical comparative study in porcine knees. Clin Orthop. 1994, 300: 281-287.PubMedGoogle Scholar
- Matthews LS, Lawrence SJ, Yahiro MA, Sinclair MR: Fixation strengths of patellar tendon-bone grafts. Arthroscopy. 1993, 9: 76-81.View ArticlePubMedGoogle Scholar
- Brown CH, Hecker AT, Hipp JA, Myers ER, Hayes WC: The biomechanics of interference screw fixation of patellar tendon anterior cruciate ligament grafts. Am J Sports Med. 1993, 21: 880-886. 10.1177/036354659302100622.View ArticlePubMedGoogle Scholar
- Kohn D, Rose C: Primary stability of interference screw fixation. Influence of screw diameter and insertion torque. Am J Sports Med. 1994, 22: 334-338. 10.1177/036354659402200307.View ArticlePubMedGoogle Scholar
- Hulstyn M, Fadale PD, Abate J, Walsh WR: Biomechanical evaluation of interference screw fixation in a bovine patellar bone-tendon-bone autograft complex for anterior cruciate ligament reconstruction. Arthroscopy. 1993, 9: 417-424.View ArticlePubMedGoogle Scholar
- Jomha NM, Raso VJ, Leung P: Effect of varying angles on the pulloutstrength of interference screw fixation. Arthroscopy. 1993, 9: 580-583.View ArticlePubMedGoogle Scholar
- Nagarkatti DG, McKeon BP, Donahue BS, Fulkerson JP: Mechanical evaluation of a soft tissue interference screw in free tendon anterior cruciate ligament graft fixation. Am J Sport Med. 2001, 29 (1): 67-71.Google Scholar
- Klein C, Patka P, Den Hollander V: Macroporous calcium phosphate bio-ceramic in dog femora: a histological study of interface and biodegradation. Biomaterials. 1989, 10: 59-62. 10.1016/0142-9612(89)90011-2.View ArticlePubMedGoogle Scholar
- Rupp S, Krauss PW, Fritsch EW: Fixation strength of a biodegradable interference screw and a press-fit technique in anterior cruciate ligament reconstruction with a BPTB graft. Arthroscopy. 1997, 13: 61-65.View ArticlePubMedGoogle Scholar
- Toutoungi DE, Lu TW, Leardini A: et al., Cruciate ligament forces in the human knee during rehabilitation exercises. Clin Biomech. 2000, 15: 176-187. 10.1016/S0268-0033(99)00063-7.View ArticleGoogle Scholar
- Holden JP, Grood ES, Korvick DL: et al., In vivo forces in the anterior cruciate ligament: Direct measurements during walking and trotting in a quadruped. J Biomech. 1994, 27: 517-526. 10.1016/0021-9290(94)90063-9.View ArticlePubMedGoogle Scholar
- Markolf KL, Burchfield DM, Shapiro MM: et al., Biomechanical consequences of replacement of the anterior cruciate ligament with a patellar ligament allograft. Part II: Forces in the graft compared with forces in the intact ligament. J Bone Joint Surg. 1996, 78A: 1728-1734.Google Scholar
- Brown GA, Pena F, Grontvedt T, Labadie D, Engebretsen L: Fixation strength of interference screw fixation in bovine, young human, and elderly human cadaver knees: Influence of insertion torque, tunnel-bone block gap, and interference. Knee Surg Sports Traumatol Arthros. 1996, 3: 238-244. 10.1007/BF01466626.View ArticleGoogle Scholar
- Namkoong S, Heywood CS, Bravman JT, Ieyasa K, Kummer FJ, Meislin RJ: The effect of interference screw diameter on soft tissue graft fixation. Bull Hosp Jt Dis. 2006, 63 (3-4): 153-155.PubMedGoogle Scholar
- Buelow JU, Siebold R, Ellermann A: A prospective evaluation of tunnel enlargement in anterior cruciate ligament reconstruction with hamstrings: extracortical versus anatomical fixation. Knee Surg, Sports Traumatol, Arthrosc. 2002, 10: 80-85. 10.1007/s00167-001-0267-6.View ArticleGoogle Scholar
- Brand JC, Nyland J, Caborn DN, Johnson DL: Soft-tissue interference fixation: bioabsorbable screw versus metal screw. Arthroscopy. 2005, 21 (8): 911-916.View ArticlePubMedGoogle Scholar
- Nylan J, Kocabey Y, Caborn DN: Insertion torque pullout strength relationship of soft tissue tendon graft tibia tunnel fixation with a bioabsorbable interference screw. Arthroscopy. 2004, 20 (4): 379-384.View ArticleGoogle Scholar
- Simonian PT, Sussmann PS, Baldini TH, Crockett HC, Wickiewicz TL: Interference screw position and hamstring graft location for anterior cruciate ligament reconstruction. Arthroscopy. 1998, 14 (5): 459-464.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/11/139/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.