Biomechanical comparison of anterior lumbar interbody fusion: stand-alone interbody cage versus interbody cage with pedicle screw fixation - a finite element analysis
© Choi et al.; licensee BioMed Central Ltd. 2013
Received: 10 July 2012
Accepted: 28 June 2013
Published: 26 July 2013
Anterior lumbar interbody fusion (ALIF) followed by pedicle screw fixation (PSF) is used to restore the height of the intervertebral disc and provide stability. Recently, stand-alone interbody cage with anterior fixation has been introduced, which eliminates the need for posterior surgery. We compared the biomechanics of the stand-alone interbody cage to that of the interbody cage with additional PSF in ALIF.
A three-dimensional, non-linear finite element model (FEM) of the L2-5 segment was modified to simulate ALIF in L3-4. The models were tested under the following conditions: (1) intact spine, (2) destabilized spine, (3) with the interbody cage alone (type 1), (4) with the stand-alone cage with anterior fixation (SynFix-LR®; type 2), and (5) with type 1 in addition to PSF (type 3). Range of motion (ROM) and the stiffness of the operated level, ROM of the adjacent segments, load sharing distribution, facet load, and vertebral body stress were quantified with external loading.
The implanted models had decreased ROM and increased stiffness compared to those of the destabilized spine. The type 2 had differences in ROM limitation of 8%, 10%, 4%, and 6% in flexion, extension, axial rotation, and lateral bending, respectively, compared to those of type 3. Type 2 had decreased ROM of the upper and lower adjacent segments by 3-11% and 3-6%, respectively, compared to those of type 3. The greatest reduction in facet load at the operated level was observed in type 3 (71%), followed by type 2 (31%) and type 1 (23%). An increase in facet load at the adjacent level was highest in type 3, followed by type 2 and type 1. The distribution of load sharing in type 2 (anterior:posterior, 95:5) was similar to that of the intact spine (89:11), while type 3 migrated posterior (75:25) to the normal. Type 2 reduced about 15% of the stress on the lower vertebral endplate compared to that in type 1. The stress of type 2 increased two-fold compared to the stress of type 3, especially in extension.
The stand-alone interbody cage can provide sufficient stability, reduce stress in adjacent levels, and share the loading distribution in a manner similar to an intact spine.
KeywordsALIF Stand-alone cage Pedicle screw fixation Finite element analysis
Anterior lumbar interbody fusion (ALIF) using a stand-alone interbody cage has been actively performed since the beginning of the 1990s [1, 2]. However, when examining the research that has been performed to date, the results showed that ALIF has provided less than desirable stability in unstable spinal segments . Presently, it is common to clinically perform an additional posterior fixation to recover the stability of spinal segments, as well as to enhance the fusion. If sufficient stability can be provided in a single surgery, the problems that can arise from extensive anterior and posterior approaches can be reduced . This single surgical procedure could reduce the postoperative pain and number of days of hospitalization, and could lead to a quick return to one’s daily routine. Recently, a stand-alone interbody cage (SynFix-LR®; Synthes Gmbh, Oberdorf, Switzerland) with reinforced fixation from forward screws and a metal plate has been introduced; however, there is insufficient systematic biomechanical research to confirm the stability of an interbody fusion and increased fusion. SynFix-LR® consists of a polyetheretherketone (PEEK) body with an additional connected metal plate and a diverging locking metal screw; this design is meant to provide strong support and fixation to an unstable spine so that an additional posterior fusion would not be required.
Using finite element model (FEM) analysis, this study analyzed the range of motion (ROM) and facet joint load of the operated and adjacent segments, as well as changes in the anterior and posterior load in a normal model and in anterior lumbar interbody fixation performed using various methods. Through this analysis, the biomechanical effects of an ALIF for the stand-alone interbody cage were revealed.
Finite element model (FEM) of a normal lumbar spine
Material properties used in finite element model of lumbar spine
Shirazi et al. 
Sharma et al. 
Goel et al. 
Adapted from Goel et al. 
Realization of the implanted model
Material properties used in finite element model of spinal fixation systems
Load conditions and boundary conditions
A multi-segment spinal model from L2 to L5 was used to compare and analyze the ROM of the operated and adjacent segments. All nodal points of the lower end plate of the lowest segment were confined, while the upper end plate of the highest segment was subjected to a pure moment of 10 Nm of flexion/extension/axial rotation and a pure moment of 5 Nm of lateral bending. A compressive force of 400 N was added to the validated intact lumbar spinal model in the follower load path direction as suggested by Patwardhan et al. .
ROM and stiffness of the models
Changes in the ROM in the operated and adjacent segments (L2–3 and L4–5) with regard to exterior load were measured before and after implanting the spinal fixation device. The hybrid test protocol [11, 12] was used to assess ROM at the operated and adjacent levels. The hybrid protocol has the following 2 steps: (1) application of a pure moment to the models and determination of total ROM and (2) application of the pure moment to the postoperative model in a displacement control mode until its ROM equals that of the intact models. At this time, pure moment applied to the postoperative model was defined as the resulting moment. The resulting moment was measured to confirm the effects of each variable element on stiffness of the spine. Subsequently the changes in ROM characteristics were investigated.
Load sharing distribution
To investigate the effect on the anterior-posterior load-sharing ratio of the vertebral body, the intact spine and each implanted model were placed under 400 N of compressed loading, and the resulting values were compared and analyzed. We measured forces at the node on the x, y, and z axes of the anterior and posterior elements. The magnitude of the forces was calculated by adding each measured nodal force and confirmed load-sharing ratio. We did not consider the influence of the ligaments.
Verification of the FEM
ROM and stiffness of the models
Adjacent segment motion
Load sharing distribution and facet load
The normal shared load was 89% anteriorly and 11% posteriorly. The ratios of anterior to posterior load sharing in types 1, 2, and 3 were 94:6, 95:5, and 75:25, respectively. The facet load in the implanted models decreased the most in type 3 (71%), followed by type 2 (31%) and type 1 (23%). The facet loads in the adjacent segments (L2–3) had similar pressures in type 1 and type 2, while type 3 had higher pressures than the other types.
Peak von Mises stress in the lower vertebral body
Lumbar interbody fusion has an exceptionally high fusion rate, and correction and restoration of the spine is relatively simple compared to that in posterolateral fusion [19, 20]. An ALIF with a cage is widely used to maintain the height of the intervertebral discs and to restore the stability of the spinal segments. Generally, posterior pedicle screw fixation is used in parallel to heighten the stability of the operated segment and fusion rate in ALIF . However, there are several potential issues, including biomechanical changes from the anterior-posterior fusion, surgical risks and complications, increased cost of the surgery, and increased operative time due to the fixation of an additional PSF [1, 22]. Additional posterior fixation after cage implantation provided superior biomechanical stability but increased the risk of neurological damage or damage to the muscle and ligament surrounding the spine from the additional posterior surgery. Therefore, questions have been raised as to whether this anterior-posterior fixation achieves better clinical results . In the case of anterior fusion using stand-alone interbody cage, in several studies it has shown stability and resistance to flexion or lateral bending but was vulnerable to applied extension or axial rotation . To overcome these problems, there have been studies in which an anterolateral plate is attached to the cage  or screws are cross-fixed between the femoral ring allograft and cancellous bone . When the results of research using anterolateral or lateral metal plates and screws after cage implantation were examined, a vastly improved stability was noted. In a study using cadavers that compared implanting a femoral ring allograft between the vertebral body to cross-fixing screws attached to the femoral ring allograft to the upper and lower vertebral body, the experimental group with the screws holding the femoral ring had increased stiffness under all given loads compared to that in the group with only the femoral ring allograft during extension (52.9% vs. 16.9%) and axial rotation (40.2% vs. 18.3%) . Using additional anterior fixation of the anterior cage, these studies were able to heighten the stability by supplementing the vulnerability regarding extension from the removal of the anterior longitudinal ligament, which had been the most vulnerable area of the stand-alone cage. In another study using cadaver segments between L5 and S1, a model using the anterior cage fixed with 3 screws and a triangular shaped metal plate, and a model using posterior fixation with pedicle screws were compared. Both models showed similar limitations in movement and stiffness in the operated segment. However, the anterior plate fixation model exhibited vulnerable stability to lateral bending compared to that exhibited by the model with the pedicle screws; this difference in the experimental results was presumed to arise from the difference in the number of fixed points . In our study, the SynFix-LR® (in which the anterior fixation provided superior stability compared to that provided by just using the vertebral interbody cage [SynCage-LR®]), had a similar stiffness in the operated segment compared to that in the model with the pedicle screws, and a difference in limitation in ROM of less than 10%. When other research results regarding SynFix-LR® were examined, it was found that the SynFix-LR® had better stability in lateral bending, extension, and axial rotation when compared to those for a semicircle wedge-shaped stand-alone cage equipped with screws facing the center of the vertebral body (STALIF; Surgicraft Ltd, Redditch, UK). These differences in the results are presumed to be due to the variation in the direction of the screws (divergence vs. convergence), the characteristics of the screws (locking screw), and the cage shape . Cain et al.  used a cadaver to study the biomechanics of SynFix-LR® similar to the experimental method used by the authors in this study. When comparing the results of the experimental group with the SynFix-LR® and additional PSF after cage implant, there were no significant differences when flexion, lateral bending, or extension were applied. However, SynFix-LR® was reported to exhibit a narrower elastic zone and stronger stiffness in the axial rotation. This is probably due to the convex cage body and strong adhesion with the vertebral body through the anterior fixation screws and metal plates with the 4 fixation points. This supplemented the stability during extension, axial rotation, and lateral bending, which had been a vulnerability of fixation procedures using stand-alone cages and anterior plates.
Degeneration of adjacent segments after fusion is a potential long-term complication. Limitation in motion occurs in the lumbar fusion region as a compensatory mechanism, and excessive movement occurs in the segments above and below the fusion region. Results from many studies showed that PSF increased the degeneration of the adjacent segments [27, 28]. In posterior lumbar fusion, when the pedicle screws were removed after fusion, the movement of the adjacent segments for flexion, extension, axial rotation, and lateral bending increased relatively less than when the pedicle screws were retained, and the stress of the intervertebral disc of the adjacent segments also decreased more than 50%. When the pedicle screws were retained, a marked increase in the movement of the segments above the operated area was observed when flexion was applied . This study confirmed that there were changes in the anterior-posterior load-sharing of the lumbar vertebrae after fusion [29, 30]. When PSF was performed, the load was focused more posteriorly compared with a normal lumbar segment, and to compensate, the burden on the facet joints of the adjacent areas increased. On the other hand, the stand-alone cage model had a similar load sharing distribution to a normal lumbar segment, and had relatively less motion in the adjacent segments compared to that for the PSF. This was considered result from the stand-alone cage with the reinforced anterior fixation moving the load that had been applied to the posterior of the spine to the anterior, which relieved the heavy load that had been applied to the facet joints of the adjacent segments.
The SynFix-LR® showed an approximate decrease of 15% in applied stress to the lower endplate compared to that using only the SynCage-LR® while in flexion. However, when extension was applied, the additional PSF model received half the PVMS of the stand-alone cage. When pedicle screws were used, the stress on the vertebral body moved anteriorly, while when only the SynCage-LR® was used, the stress was focused on the posterior part of the vertebral body. Through anterior fixation, the stand-alone cage can reduce the stress focused on the posterior of the vertebral body, as well as disperse the stress from surrounding vertebral body at the same time.
The limitation of this study is that the material properties of this simulation, such as the non-linear behavior of the spinal ligaments, the viscoelasticity of the disc, and the grade of degenerative disc, were simplified and idealized based on the properties of a cadaver specimen. This is due to the differences in the geometry and material properties, where the intervertebral discs within the human body are extremely flexible structures; however, the finite element simulated the intervertebral disc as a solid element. Thus, the non-linear characteristics were difficult to reenact. In addition, the state of the bone, which is an important factor to consider in fusion procedures, was not considered. Finite element research is similar to studies that are conducted in vitro, and as such, muscle contraction around the spine could not be considered.
In conclusion, the stand-alone cage with reinforced anterior fixation provided sufficient stability and stiffness necessary to carry out lumbar fusion and at the same time, reduced the excessive motion of the adjacent segments and the stress on the adjacent segment joints by exhibiting similar load sharing characteristics to a normal lumbar. Therefore, the stand-alone cage can supplement with the shortcomings of the anterior lumbar interbody cage with no fixation device and have the advantage of parallel use with PSF.
This study was supported by a grant from the Wooridul Spine Foundation.
- Gerber M, Crawford NR, Chamberlain RH, Fifield MS, LeHuec JC, Dickman CA: Biomechanical assessment of anterior lumbar interbody fusion with an anterior lumbosacral fixation screw-plate: comparison to stand-alone anterior lumbar interbody fusion and anterior lumbar interbody fusion with pedicle screws in an unstable human cadaver model. Spine (Phila Pa 1976). 2006, 31: 762-768. 10.1097/01.brs.0000206360.83728.d2.View ArticleGoogle Scholar
- Lubbers T, Bentlage C, Sandvoss G: Anterior lumbar interbody fusion as a treatment for chronic refractory lower back pain in disc degeneration and spondylolisthesis using carbon cages - stand alone. Zentralbl Neurochir. 2002, 63: 12-17. 10.1055/s-2002-31580.View ArticlePubMedGoogle Scholar
- Oxland TR, Lund T: Biomechanics of stand-alone cages and cages in combination with posterior fixation: a literature review. Eur Spine J. 2000, 9 (Suppl 1): S95-S101.View ArticlePubMedGoogle Scholar
- Fritzell P, Hagg O, Wessberg P, Nordwall A: Chronic low back pain and fusion: a comparison of three surgical techniques: a prospective multicenter randomized study from the Swedish lumbar spine study group. Spine (Phila Pa 1976). 2002, 27: 1131-1141. 10.1097/00007632-200206010-00002.View ArticleGoogle Scholar
- Smit TH, Odgaard A, Schneider E: Structure and function of vertebral trabecular bone. Spine (Phila Pa 1976). 1997, 22: 2823-2833. 10.1097/00007632-199712150-00005.View ArticleGoogle Scholar
- Goel VK, Monroe BT, Gilbertson LG, Brinckmann P: Interlaminar shear stresses and laminae separation in a disc. Finite element analysis of the L3-L4 motion segment subjected to axial compressive loads. Spine (Phila Pa 1976). 1995, 20: 689-698. 10.1097/00007632-199503150-00010.View ArticleGoogle Scholar
- Shirazi-Adl SA, Shrivastava SC, Ahmed AM: Stress analysis of the lumbar disc-body unit in compression. A three-dimensional nonlinear finite element study. Spine (Phila Pa 1976). 1984, 9: 120-134. 10.1097/00007632-198403000-00003.View ArticleGoogle Scholar
- Sharma M, Langrana NA, Rodriguez J: Role of ligaments and facets in lumbar spinal stability. Spine (Phila Pa 1976). 1995, 20: 887-900. 10.1097/00007632-199504150-00003.View ArticleGoogle Scholar
- Polikeit A, Ferguson SJ, Nolte LP, Orr TE: Factors influencing stresses in the lumbar spine after the insertion of intervertebral cages: finite element analysis. Eur Spine J. 2003, 12: 413-420. 10.1007/s00586-002-0505-8.View ArticlePubMedGoogle Scholar
- Patwardhan AG, Havey RM, Meade KP, Lee B, Dunlap B: A follower load increases the load-carrying capacity of the lumbar spine in compression. Spine (Phila Pa 1976). 1999, 24: 1003-1009. 10.1097/00007632-199905150-00014.View ArticleGoogle Scholar
- Panjabi MM: Hybrid multidirectional test method to evaluate spinal adjacent-level effects. Clin Biomech (Bristol, Avon). 2007, 22: 257-265. 10.1016/j.clinbiomech.2006.08.006.View ArticleGoogle Scholar
- Rohlmann A, Neller S, Claes L, Bergmann G, Wilke HJ: Influence of a follower load on intradiscal pressure and intersegmental rotation of the lumbar spine. Spine (Phila Pa 1976). 2001, 15: E557-E561.View ArticleGoogle Scholar
- Yamamoto I, Panjabi MM, Crisco T, Oxland T: Three-dimensional movements of the whole lumbar spine and lumbosacral joint. Spine (Phila Pa 1976). 1989, 14: 1256-1260. 10.1097/00007632-198911000-00020.View ArticleGoogle Scholar
- Brown T, Hansen RJ, Yorra AJ: Some mechanical tests on the lumbosacral spine with particular reference to intervertebral discs; a preliminary report. J Bone Joint Surg Am. 1957, 39: 1135-1164.PubMedGoogle Scholar
- Tencer AF, Ahmed AM, Burke DL: Some static mechanical properties of the lumbar intervertebral joint, intact and injured. J Biomech Eng. 1982, 104: 193-201. 10.1115/1.3138348.View ArticlePubMedGoogle Scholar
- Nachemson AL: Lumbar intradical pressure: Experimental studies on Post-mortem material. Acta Orthop Scand Suppl. 1960, 43: 1-104.View ArticlePubMedGoogle Scholar
- Ranu HS, Denton RA, King AI: Pressure distribution under an intervertebral disc–an experimental study. J Biomech. 1979, 12: 807-812. 10.1016/0021-9290(79)90166-0.View ArticlePubMedGoogle Scholar
- Rolander SD: Motion of the lumbar spine with special reference to the stabilizing effect of posterior fusion. An experimental study on autopsy specimens. Acta Orthop Scand. 1966, Suppl 90: 1-144.View ArticleGoogle Scholar
- Musluman AM, Yilmaz A, Cansever T, Cavusoglu H, Colak I, Genc HA, Aydin Y: Posterior lumbar interbody fusion versus posterolateral fusion with instrumentation in the treatment of low-grade isthmic spondylolisthesis: midterm clinical outcomes. J Neurosurg Spine. 2011, 14: 488-496. 10.3171/2010.11.SPINE10281.View ArticlePubMedGoogle Scholar
- Zhou ZJ, Zhao FD, Fang XQ, Zhao X, Fan SW: Meta-analysis of instrumented posterior interbody fusion versus instrumented posterolateral fusion in the lumbar spine. J Neurosurg Spine. 2011, 15: 295-310. 10.3171/2011.4.SPINE10330.View ArticlePubMedGoogle Scholar
- Resnick DK, Choudhri TF, Dailey AT, Groff MW, Khoo L, Matz PG, Mummaneni P, Watters WC, Wang J, Walters BC, Hadley MN: Guidelines for the performance of fusion procedures for degenerative disease of the lumbar spine. Part 17: bone growth stimulators and lumbar fusion. J Neurosurg Spine. 2005, 2: 737-740. 10.3171/spi.2005.2.6.0737.View ArticlePubMedGoogle Scholar
- O'Brien JP, Holte DC: Simultaneous combined anterior and posterior fusion. A review of its concept and 10 years of refinement of the technique: a solution for the patient with severe back and leg pain. Eur Spine J. 1992, 1: 2-6. 10.1007/BF00302134.View ArticlePubMedGoogle Scholar
- Bozkus H, Chamberlain RH, Perez Garza LE, Crawford NR, Dickman CA: Biomechanical comparison of anterolateral plate, lateral plate, and pedicle screws-rods for enhancing anterolateral lumbar interbody cage stabilization. Spine (Phila Pa 1976). 2004, 29: 635-641. 10.1097/01.BRS.0000115126.13081.7D.View ArticleGoogle Scholar
- Kuzhupilly RR, Lieberman IH, McLain RF, Valdevit A, Kambic H, Richmond BJ: In vitro stability of FRA spacers with integrated crossed screws for anterior lumbar interbody fusion. Spine (Phila Pa 1976). 2002, 27: 923-928. 10.1097/00007632-200205010-00007.View ArticleGoogle Scholar
- Schleicher P, Gerlach R, Schär B, Cain CM, Achatz W, Pflugmacher R, Haas NP, Kandziora F: Biomechanical comparison of two different concepts for stand alone anterior lumbar interbody fusion. Eur Spine J. 2008, 17: 1757-1765. 10.1007/s00586-008-0797-4.View ArticlePubMedPubMed CentralGoogle Scholar
- Cain CM, Schleicher P, Gerlach R, Pflugmacher R, Scholz M, Kandziora F: A new stand-alone anterior lumbar interbody fusion device: biomechanical comparison with established fixation techniques. Spine (Phila Pa 1976). 2005, 30: 2631-2636. 10.1097/01.brs.0000187897.25889.54.View ArticleGoogle Scholar
- Park P, Garton HJ, Gala VC, Hoff JT, McGillicuddy JE: Adjacent segment disease after lumbar or lumbosacral fusion: review of the literature. Spine (Phila Pa 1976). 2004, 29: 1938-1944. 10.1097/01.brs.0000137069.88904.03.View ArticleGoogle Scholar
- Kim HJ, Chun HJ, Moon SH, Kang KT, Kim HS, Park JO, Moon ES, Sohn JS, Lee HM: Analysis of biomechanical changes after removal of instrumentation in lumbar arthrodesis by finite element analysis. Med Biol Eng Comput. 2010, 48: 703-709. 10.1007/s11517-010-0621-2.View ArticlePubMedGoogle Scholar
- Cripton PA, Jain GM, Wittenberg RH, Nolte LP: Load-sharing characteristics of stabilized lumbar spine segments. Spine (Phila Pa 1976). 2000, 25: 170-179. 10.1097/00007632-200001150-00006.View ArticleGoogle Scholar
- Ahn YH, Chen WM, Lee KY, Park KW, Lee SJ: Comparison of the load-sharing characteristics between pedicle-based dynamic and rigid rod devices. Biomed Mater. 2008, 3: 044101-10.1088/1748-6041/3/4/044101.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2474/14/220/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.